1 English Español 10 series Blood Pressure Monitor Model BP786 Instruction Manual B i
2 English Product includes: El producto incluye: AC Adapter Adaptador de CA Monitor Monitor ComFit Cuff Brazalete ComFit 10 series Blood Pressure Monitor Model BP786 Instruction Manual Eespañol Quick Start Guide Guía resumida ii Instruction Manual Manual de Instrucciones
3 Contents Thank you for purchasing the OMRON BP786 Blood Pressure Monitor. Your new blood pressure monitor uses the oscillometric method of blood pressure measurement. This means the monitor detects your blood movement through your brachial artery and converts the movements into a digital reading. An oscillometric monitor does not need a stethoscope so the monitor is simple to use. Intended Use This device is a digital monitor intended for use in measuring blood pressure and pulse rate in adult patient population. The device detects the appearance of irregular heartbeats during measurement and gives a warning signal with the measurement result. Important Safety Information Know Your Device Display Symbols Before Taking a Measurement Preparation Battery Installation Using the AC adapter Setting the TruReadTM Mode Setting the Date and Time Using the Device Applying the Arm Cuff How to Sit Correctly Taking a Measurement Using the Memory Function Error Messages and Troubleshooting Error Messages Troubleshooting Maintenance and Storage Maintenance Storage Optional Medical Accessories Specifications FCC/IC Statement and Trademarks Limited Warranty Guidance and Manufacturer s Declaration Please read this instruction manual thoroughly before using the device. Please keep for future reference. For specific information about your own blood pressure, CONSULT YOUR PHYSICIAN.
4 Important Safety Information Warning: Indicates a potentially hazardous situation which, if not avoided, could result in death or serious injury. (General Usage) DO NOT adjust medication based on measurement results from this blood pressure monitor. Take medication as prescribed by your physician. Only a physician is qualified to diagnose and treat High Blood Pressure. The monitor is not intended to be a diagnostic device. Consult your physician before using the device for any of the following conditions: common arrhythmias such as atrial or ventricular premature beats or atrial fibrillation, arterial sclerosis, poor perfusion, diabetes, age, pregnancy, pre-eclampsia, renal diseases. Note that PATIENT motion, trembling, shivering may affect the measurement reading. Do not use the device on the injured arm or the arm under medical treatment. Do not apply the arm cuff on the arm while on an intravenous drip or blood transfusion. Consult your physician before using the device on the arm with an arterio-venous (A-V) shunt. Do not use the device with other medical electrical (ME) equipment simultaneously. Do not use the device in the area of HF surgical equipment, MRI, or CT scanner, or in an oxygen rich environment. The air tube or the AC adapter cable may cause accidental strangulation in infants. Contains small parts that may cause a choking hazard if swallowed by infants. (Data Transmission) Do not use this product on aircraft or in hospitals. Please remove the battery and AC adapter from the unit. This product emits radio frequencies (RF) in the 2.4 GHz band, use of this product in locations where RF is restricted is not recommended. The use of RF in this product is licensed for use by the FCC, for further information on any potential restrictions refer to documentation on Bluetooth usage by the FCC. (AC Adapter Usage) Do not use the AC adapter if the device or the power cord is damaged. Turn off the power and unplug the power cord immediately. Plug the AC adapter into the appropriate voltage outlet. Do not use in a multi-outlet plug. Never plug in or unplug the power cord from the electric outlet with wet hands. EN 1
5 Important Safety Information 2 Caution: Indicates a potentially hazardous situation which, if not avoided, may result in minor or moderate injury to the user or patient or damage to the equipment or other property. (General Usage) Always consult your physician. Self-diagnosis of measurement results and self-treatment are dangerous. Consult your physician before using the device for any of the following conditions: If you have had a mastectomy. Do not take measurements more than necessary. It may cause bruising due to blood flow interference. People with severe blood flow problems or blood disorders as cuff inflation can cause bruising. Remove the arm cuff if it does not start deflating during the measurement. Do not use this device on infants or persons who cannot express their intentions. Do not use the device for any purpose other than measuring blood pressure. Use only the approved arm cuff for this device. Use of other arm cuffs may result in incorrect measurement results. Do not use a mobile phone or other devices that emit electromagnetic fields near the device except when in use for wireless communications. This may result in incorrect operation of the device. Do not disassemble the monitor or arm cuff. This may cause an inaccurate reading. Do not use in a location with moisture, or a location where water may splash on the device. This may damage the device. Do not use the device in a moving vehicle (car, airplane). Read "If your systolic pressure is more than 210 mmhg" (page14) of this instruction manual, if your systolic pressure is known to be more than 210 mmhg. Inflating to a higher pressure than necessary may result in bruising where the cuff is applied. (AC Adapter Usage) Fully insert the power plug into the outlet. When disconnecting the power plug from the outlet, do not pull the power cord. Be sure to pull from the power plug safely. When handling the power cord, take care not to do the following: Do not damage. Do not break it. Do not tamper with it. Do not forcibly bend or pull. Do not twist. Do not bundle during use. Do not pinch. Do not place under heavy objects.
6 Important Safety Information Wipe the dust off from the power plug. Unplug monitor when not in use. Disconnect the power plug before cleaning. Use only an OMRON AC adapter designed for this device. Use of unsupported adapters may damage and/or may be hazardous to the device. (Battery Usage) Do not insert the batteries with their polarities incorrectly aligned. Use only 4 AA alkaline or manganese batteries with this device. Do not use other types of batteries. Do not use new and used batteries together. Remove the batteries if the device will not be used for three months or more. EN General Precautions Do not forcibly crease the arm cuff or the air tube excessively. Do not press the air tube while taking a measurement. To unplug the air plug, pull on the air plug at the connection with the monitor, not the tube itself. Do not drop the monitor or subject device to strong shocks or vibrations. Do not inflate the arm cuff when it is not wrapped around your arm. Do not use the device outside the specified environment. It may cause an inaccurate reading. Dispose of the device, components and optional accessories according to applicable local regulations. Unlawful disposal may cause environmental pollution. 3
7 1. Know Your Device 4 Open the rear cover page to read the following: The letter identifiers on the rear cover page correspond to those in the body of this page. Display Monitor A Air jack B Blood pressure color indicator C Date/Time setting button D USER ID selection switch E Display F Memory button G START/STOP button H Up/Down buttons I Battery compartment J AC adapter jack Arm cuff K Arm cuff (Arm circumference 9-17 (22-42 cm)) L Air plug M Air tube N O P Q R S T U V W X Y Z AA AB AC AD Memory symbol USER ID symbol Average value symbol Systolic blood pressure Diastolic blood pressure Transfer indicator Memory full symbol OK symbol Heartbeat symbol (Flashes during measurement) Date/Time display TruRead TM symbol Movement error symbol Irregular heartbeat symbol Blood pressure level indicator Low battery symbol Deflation symbol Pulse display / Memory number
8 1. Know Your Device 1.1 Display Symbols Irregular Heartbeat Symbol ( ) When the monitor detects an irregular rhythm two or more times during the measurement, the irregular heartbeat symbol will appear on the display with the measurement values. An irregular heartbeat rhythm is defined as a rhythm that is 25% less or 25% more than the average rhythm detected while the monitor is measuring the systolic and diastolic blood pressure. If the irregular heartbeat symbol displays with your measurement results, we recommend you consult your physician. Follow the directions of your physician. Pulse Blood pressure Blood pressure Normal Heartbeat Irregular Heartbeat Short Long Pulse EN Movement Error Symbol ( ) The movement error symbol is displayed if you move your body during the measurement. Please remove the arm cuff, and wait 2-3 minutes. Take another measurement, remain still during measurement. Average Value Symbol ( ) The average value symbol is displayed when you press and hold the memory button for more than 3 seconds. The most recent average value appears on the display screen. 5
9 1. Know Your Device Blood Pressure Color Indicator If your Systolic Blood Pressure is 135 mmhg or above and/or the Diastolic Blood Pressure is 85 mmhg or above, the blood pressure color indicator will light in orange when the measurement result is displayed. If the measurements are within the standard range, the blood pressure color indicator will light in green ESH/ESC Guidelines for the management of arterial hypertension Definitions of hypertension by office and home blood pressure levels Office Home Systolic Blood Pressure 140 mmhg 135 mmhg Diastolic Blood Pressure 90 mmhg 85 mmhg These are from statistical values for blood pressure. 1.2 Before Taking a Measurement To help ensure an accurate reading, follow these directions: 1. Avoid bathing, drinking alcohol or caffeine, smoking, exercising and eating for 30 minutes before taking a measurement. Rest for at least 15 minutes before taking the measurement. 2. Stress raises blood pressure. Avoid taking measurements during stressful times. 3. Measurements should be taken in a quiet place. 4. Remove tight-fitting clothing from your arm. 5. Sit on a chair with your feet flat on the floor. Rest your arm on a table so that the arm cuff is at the same level as your heart. 6. Remain still and do not talk during the measurement. 7. Keep a record of your blood pressure and pulse readings for your physician. A single measurement does not provide an accurate indication of your true blood pressure. You need to take and record several readings over a period of time. Try to measure your blood pressure at the same time each day for consistency. 6
10 2. Preparation 2.1 Battery Installation 1. Remove the battery cover. 2. Insert 4 AA batteries as indicated in the battery compartment. EN 3. Replace the battery cover. Notes: When the low battery symbol ( ) appears on the display, turn the monitor off, then replace all batteries at the same time. Long life alkaline batteries are recommended. The measurement values continue to be stored in memory even after the batteries are replaced. Dispose of the device, components and optional accessories according to applicable local regulations. Unlawful disposal may cause environmental pollution. 7
11 2. Preparation 2.2 Using the AC adapter IMPORTANT SAFETY INSTRUCTIONS SAVE THESE INSTRUCTIONS - DANGER - TO REDUCE THE RISK OF FIRE OR ELECTRIC SHOCK, CAREFULLY FOLLOW THESE INSTRUCTIONS. The power unit (AC Adapter) is intended to be correctly oriented in a vertical or floor mount position. NOTES: Use only the authorized OMRON AC Adapter that came with this monitor. We recommend you install the batteries even if the AC Adapter is used. If no batteries are installed, you may need to reset the date and time if the AC Adapter is disconnected. The measurement results will not be deleted. 8 Use only OMRON authorized parts and accessories. Parts and accessories not approved for use with the device may damage the device.
12 2. Preparation 1. Insert the AC adapter plug into the AC adapter jack on the rear side of the monitor. 2. Plug the AC adapter into an electrical outlet. EN To disconnect the AC adapter, unplug the AC adapter from the electrical outlet first and then remove the AC adapter plug from the monitor. 9
13 2. Preparation 2.3 Setting the TruRead TM Mode The TruRead TM Mode takes 3 consecutive measurements. The monitor will inflate, take a measurement, and deflate - 3 times, separated by a short interval between each measurement. The TruRead TM Mode is set off by default. 1. Select your USER ID (1 or 2). 2. Press the button twice. The TruRead TM symbol ( ) appears on the display. 10 1) 2) 3) 4) Press or buttons to select on or off. Press the button to confirm. If on is selected the interval setting appears. Press or buttons to change the interval. It can be set 15, 30, 60, or 120 seconds. Press the button to confirm. The year flashes on the display. 3. To set the date and time, proceed to section 2.4 If the date and time are correct, press the START/STOP button to turn the monitor off.
14 2. Preparation Open the rear cover page to read the following: The letter identifiers on the rear cover page correspond to those in the body of this page. 2.4 Setting the Date and Time Set the monitor to the correct date and time before taking a measurement for the first time. 1. Press the C Date/Time setting ( ) button. Press button until the year flashes on the W Date/Time display. 2. Push or button to change the year. Push button to confirm the year and then the month flashes. Repeat the same steps to change the month, day, hour, and minute. Year Month Day Change Confirm : Forward : Back Change Confirm Change Confirm 3. Press the START/STOP button to turn the monitor off. Notes: If the batteries have been replaced, the date and time setting will need to be reset. If the date and time are not set, -/ - -:-- appears during or after measurement. Hour Change Confirm Minute Change Confirm EN 11
15 3. Using the Device 12 Open both the front and rear covers to read the following: The letter identifiers on the cover page correspond to those in the body of this page. 3.1 Applying the Arm Cuff Remove tight-fitting clothing or tight rolled up sleeve from your left upper arm. Do not place the arm cuff over thick clothes. 1. Insert L the air plug into A the air jack securely. 2. Apply the arm cuff to your left upper arm. The bottom edge of the arm cuff should be a 1/2 inch (1 to 2 cm) above the elbow. M Air tube is on the inside of your arm and aligned with your middle finger. 3. Secure closed with the fabric fastener. Notes: When you take a measurement on the right arm, the air tube will be at the side of your elbow. Be careful not to rest your arm on the air tube. --- b The blood pressure can differ between the right arm and the left arm, and the measured blood pressure values can be different. OMRON recommends to always use the same arm for measurement. If the values between both arms differ substantially, please check with your physician which arm to use for your measurements.
16 3. Using the Device 3.2 How to Sit Correctly To take a measurement, you need to be relaxed and comfortably seated, under comfortable room temperature. Avoid bathing, drinking alcohol or caffeine, smoking, exercising or eating 30 minutes before taking a measurement. Sit on a chair with your feet flat on the floor. Sit upright with your back straight. --- c Sit with your back and arm being supported. The arm cuff should be placed on your arm at the same level as your heart. --- d EN 13
17 3. Using the Device 3.3 Taking a Measurement Notes: To stop a measurement, press the START/STOP button once to deflate the arm cuff. Remain still and quiet while taking a measurement. The monitor is designed to take measurements and store the measurement values in the memory for 2 people using USER ID 1 and USER ID 2. Using the Single Mode 1. Select your USER ID (1 or 2). 2. Press the START/STOP button. The arm cuff starts to inflate automatically. START INFLATE DEFLATE COMPLETE If your systolic pressure is more than 210 mmhg After the arm cuff starts to inflate, press and hold the START/STOP button until the monitor inflates 30 to 40 mmhg higher than your expected systolic pressure. Notes: The monitor will not inflate above 299 mmhg. 14 Inflating to a higher pressure than necessary may result in bruising where the arm cuff is applied.
18 3. Using the Device 3. Remove the arm cuff. 4. Press the START/STOP button to turn the monitor off. The monitor automatically stores the measurement result in its memory. It will automatically turn off after 2 minutes. Note: Wait 2-3 minutes before taking another measurement. Waiting between measurements allows the arteries to return to the condition prior to taking a measurement. EN Using the TruRead TM Mode 1. Select your USER ID (1 or 2). 2. Press the START/STOP button. The arm cuff starts to inflate automatically. 15
19 3. Using the Device The first measurement is complete, waiting to start the next measurement. The interval will appear on the display. The second measurement is complete, waiting to start the next measurement. The interval will appear on the display. The third measurement is complete, the average for the 3 measurements appears on the display. 16 Press the button to display the individual measurement values.
20 3. Using the Device Using the Guest Mode The monitor stores measurement values for 2 users in the memory. The guest mode can be used to take a single measurement for another user. No measurement values are stored in the memory, and the TruRead TM mode is not available when the guest mode is selected. 1. Press and hold the START/STOP button for more than 3 seconds. The USER ID symbol and the Date/Time display will disappear. 2. Release the START/STOP button when the Date/Time display turns off. The arm cuff will start to inflate automatically. EN DO NOT adjust medication based on measurement results from this blood pressure monitor. Take medication as prescribed by your physician. Only a physician is qualified to diagnose and treat High Blood Pressure. This monitor is not intended to be a diagnostic device. Always consult your physician. Self-diagnosis of measurement results and self-treatment are dangerous. Read "If your systolic pressure is more than 210 mmhg" (page14) of this instruction manual, if your systolic pressure is known to be more than 210 mmhg. Inflating to a higher pressure than necessary may result in bruising where the cuff is applied. 17
21 3. Using the Device 3.4 Using the Memory Function The monitor automatically stores the results up to 100 sets for each user (1 and 2). It can also calculate an average value based on the last 3 measurement values taken within 10 minutes. Notes: If there are only 2 measurement values in the memory for that period, the average will be based on these 2 values. If there is 1 measurement value in the memory for that period, this is displayed as the average. If the memory is full, the monitor will delete the oldest value. When viewing the measurement value taken without setting the date and time, -/ - -:-- is displayed instead of the date and time. To View the Measurement Values Stored in Memory 1. Select your USER ID (1 or 2). 2. Press the button. The Memory number appears for a second before the pulse rate is displayed. The newest set is numbered 1. 18
22 3. Using the Device 3. Press the or button to view the values stored in the memory. : To view the older values : To view the more recent values Note: The value with the TruRead TM symbol is the average for the consecutive 3 measurements. To display the individual measurement values, press the button while the average is displayed. EN 19
23 3. Using the Device To View the Average Value 1. Select your USER ID (1 or 2). 2. Press and hold the button for more than 3 seconds. Notes: If the previous measurement was taken without setting the date and time, the average value is not calculated. If there are no measurement values stored in the memory, the screen to the right is displayed. 20
24 3. Using the Device To Pair This Device With a Smartphone To begin using the OMRON Wellness software for the first time, please visit for the initial set-up instructions. 1. Download and install onto your smartphone, the free "Omron Wellness" APP. EN 2. Open the App on your smartphone and follow set-up and pairing instructions. 3. Press the Date/Time setting ( ) button. The transfer indicator ( ) will appear on the display. 21
25 3. Using the Device 4. Open the App on your smartphone and follow the device pairing instructions shown on your smartphone, while the transfer indicator is flashing on the display. The OK symbol and will flash when pairing is completed. Any readings currently saved in the blood pressure monitor s memory will automatically be transfered to the Omron Wellness App after successful completion of the pairing process. Notes: If the pairing has failed, the screen to the right is displayed. You can pair up to 2 destination devices with the monitor. The monitor will automatically turn off after 10 seconds. 22
26 3. Using the Device To Unpair Your Device 1. Press the Date/Time setting ( ) button. The transfer indicator ( ) will appear on the display. 2. While holding the button down, press the START/STOP button for more than 2 seconds. EN The OK symbol and will flash if your device has been unpaired. Notes: Unpairing your device will not delete the information in the phone app. The monitor will automatically turn off after 10 seconds. If you re-pair your smartphone to your blood pressure monitor, all prior reading history stored on the mobile App will be retained and all readings currently stored on the blood pressure monitor will be added to the mobile App. 23
27 3. Using the Device To Delete All the Values Stored in Memory The values stored in the memory are deleted by USER ID. 1. Select your USER ID (1 or 2). 2. Press the Memory button, while the memory symbol ( ) appears. 3. While holding the button down, press the START/STOP button for more than 3 seconds. Note: You cannot partially delete the values stored in the memory. All values for the user you select will be deleted. 24
28 4. Error Messages and Troubleshooting 4.1 Error Messages Display Cause Solution Irregular heartbeats are detected. Movement during measurement. Remove the arm cuff. Wait 2-3 minutes and then take another measurement. Repeat the steps in section 3.3. If this error continues to appear, contact your physician. Carefully read and repeat the steps in section 3.3. EN The batteries are low. The batteries are exhausted. You should replace the batteries with new ones ahead of time. Refer to section 2.1. You should replace the batteries with new ones at once. Refer to section 2.1. Communication failed. Refer to "Data cannot be sent." in section
29 4. Error Messages and Troubleshooting Display Cause Solution 26 Air plug disconnected. Arm cuff is applied too loosely. Air is leaking from the arm cuff. Movement during measurement and the arm cuff has not been inflated sufficiently. The arm cuff was inflated exceeding the maximum allowable pressure, and then deflated automatically when inflating the arm cuff manually. Movement during measurement. Clothing is interfering with the arm cuff. Device error. Insert the air plug securely. Refer to section 3.1. Apply the arm cuff tighter. Refer to section 3.1. Replace the arm cuff with a new one. Refer to section 5.3. Repeat measurement. Remain still and do not talk during measurement. Refer to section 3.3. If E2 appears repeatedly, inflate the arm cuff manually until it is 30 to 40 mmhg above your previous measurement result. Refer to section 3.3. Do not touch the arm cuff and/or bend the air tube while taking a measurement. Do not inflate the arm cuff more than necessary. Refer to section 3.3. Repeat measurement. Remain still and do not talk during measurement. Refer to section 3.3. Remove any clothing interfering with the arm cuff. Refer to section 3.1. Contact Customer Service.
30 4. Error Messages and Troubleshooting 4.2 Troubleshooting Problem No power. No display appears on the monitor. Measurement values appear too high or too low. Data cannot be sent. Cause and Solution Replace all batteries with new ones. Check the battery installation for proper placement of the battery polarities. Refer to section 2.1. Blood pressure varies constantly. Many factors including stress, time of day, and how you wrap the cuff, may affect your blood pressure. Review the section 1.2 and section 3.3. The destination device is too far away from the monitor. After checking that there are no sources of interference nearby, move the device to a distance within 16 ft. (5 m) of the monitor. The Bluetooth function on the destination device is turned off. Turn the on the Bluetooth function and try sending the data again. Pairing (registering) has not been completed. Perform pairing (registration). Refer to "To Pair This Device With a Smartphone" in section 3.4. The application on the destination device is not ready. Check the application then try sending the data again. Refer to "To Pair This Device With a Smartphone" in section 3.4. If the Err symbol still lights after checking the application, contact customer service. EN 27
31 5. Maintenance and Storage 5.1 Maintenance To protect your device from damage, please observe the following: Store the device and the components in a clean, safe location. Do not use any abrasive or volatile cleaners. Do not wash the device and any components or immerse them in water. Do not use gasoline, thinners or similar solvents to clean the device. Use a soft and dry cloth, or a soft and moistened cloth and neutral soap to clean on the monitor and the arm cuff. Changes or modification not approved by the manufacturer will void the user warranty. Do not disassemble or attempt to repair the device or components. Consult Customer Service. 28
32 5. Maintenance and Storage 5.2 Storage 1. Unplug the air plug from the air jack. 2. Gently fold the air tube into the arm cuff. EN Note: Do not bend or crease the air tube excessively. Do not store the device in the following situations: If the device is wet. Locations exposed to extreme temperatures, humidity, direct sunlight, dust or corrosive vapors such as bleach. Locations exposed to vibrations, shocks or where it will be at an angle. 29
33 5. Maintenance and Storage 5.3 Optional Medical Accessories Arm Cuff Arm circumference 9-17 (22-42 cm) AC Adapter ComFit TM Cuff (CFX-WR17) (Model: HEM-FL31) HEM-ADPTW5 30
34 6. Specifications Model BP786 HEM-7321T-Z Display LCD digital display Measurement range Pressure: 0 to 299 mmhg Pulse: 40 to 180 beats / min. Accuracy Pressure: ±3 mmhg or 2% of reading Pulse: ± 5% of display reading Inflation Fuzzy-logic controlled by electric pump Deflation Automatic pressure release valve Measurement method Oscillometric method IP classification IP 20 Power source 4 AA batteries 1.5V or AC adapter (INPUT AC V 50/60Hz 0.12A) Battery life Approximately 1000 measurements (using new alkaline batteries) Operating temperature / humidity 50 F to 104 F (10 C to 40 C) / 15 to 90% RH Storage temperature / humidity / air pressure -4 F to 140 F (-20 C to 60 C) / 10 to 95% RH / 700 to 1060 hpa Weight Monitor : Approximately 14 1/8 oz. (400 g) not including batteries Arm cuff : Approximately 5 3/4 oz. (163 g) Dimensions Monitor : Approximately 4 7/8 (w) 3 1/2 (h) 6 3/8 (l) (124 mm 90 mm 161 mm) Arm cuff : Approximately 5 3/4 21 (air tube: 29 1/2 ) (145 mm 532 mm (air tube: 750 mm)) Cuff circumference 9 to 17 (220 to 420 mm) Memory Up to 100 per user Contents Monitor, arm cuff, AC adapter, instruction manual, quick start guide Applied part = Type BF Protection against electric shock Internally powered ME equipment (When using only the batteries) EN = Class II ME equipment (AC adapter) Notes: These specifications are subject to change without notice. In the clinical validation study, the 5th phase was used on 85 subjects for determination of diastolic blood pressure. This device has not been validated for use in pregnancy. 31
35 7. FCC/IC Statement and Trademarks FCC CAUTION Changes or modifications not expressly approved by the party responsible for compliance could void the user's authority to operate the equipment. Note: This equipment has been tested and found to comply with the limits for a Class B digital device, pursuant to part 15 of the FCC Rules. These limits are designed to provide reasonable protection against harmful interference in a residential installation. This equipment generates, uses and can radiate radio frequency energy and, if not installed and used in accordance with the instructions, may cause harmful interference to radio communications. However, there is no guarantee that interference will not occur in a particular installation. If this equipment does cause harmful interference to radio or television reception, which can be determined by turning the equipment off and on, the user is encouraged to try to correct the interference by one or more of the following measures: Reorient or relocate the receiving antenna. Increase the separation between the equipment and receiver. Connect the equipment into an outlet on a circuit different from that to which the receiver is connected. Consult the dealer or an experienced radio/tv technician for help. This transmitter must not be co-located or operated in conjunction with any other antenna or transmitter. This equipment complies with FCC/IC radiation exposure limits set forth for an uncontrolled environment and meets the FCC radio frequency (RF) Exposure Guidelines and RSS-102 of the IC radio frequency (RF) Exposure rules. This equipment has very low levels of RF energy that are deemed to comply without testing of specific absorption ratio (SAR). 32
36 7. FCC/IC Statement and Trademarks This Product operates in the unlicensed ISM band at 2.4GHz. In case this Product is used around the other wireless devices including microwave and wireless LAN, which operate same frequency band of this Product, there is a possibility that interference occurs between this Product and such other devices. If such interference occurs, please stop the operation of other devices or relocate this Product before using this Product or do not use this Product around the other wireless devices. The Bluetooth Smart word mark and logos are registered trademarks owned by Bluetooth SIG, Inc. and any use of such marks by OMRON HEALTHCARE Co.,Ltd. is under license. Other trademarks and trade names are those of their respective owners. EN Apple, the Apple logo, ipad, and iphone are trademarks of Apple Inc., registered in the U.S. and other countries. App Store is a service mark of Apple Inc. Android and Google Play are both trademarks of Google Inc. Samsung and Galaxy S are both registered trademarks of Samsung Electronics Co., Ltd. 33
37 8. Limited Warranty Your BP786 Automatic Blood Pressure Monitor, excluding the arm cuff and battery, is warranted to be free from defects in materials and workmanship appearing within 5 years from the date of purchase, when used in accordance with the instructions provided with the monitor. The arm cuff is warranted to be free from defects in materials and workmanship appearing within 1 year from the date of purchase when the monitor is used in accordance with the instructions provided with the monitor. The above warranty extends only to the original retail purchaser. We will, at our option, replace without charge any monitor or arm cuff covered by the above warranty. Replacement is our only responsibility and your only remedy under the above warranty. To obtain warranty service contact Customer Service by calling for the address of the inspection center and the return shipping and handling fee. Enclose the original printed receipt. Include a letter, with your name, address, phone number, and description of the specific problem. Pack the product carefully to prevent damage in transit. Because of possible loss in transit, we recommend insuring the product with return receipt requested. THE FOREGOING IS THE SOLE WARRANTY PROVIDED BY OMRON IN CONNECTION WITH THIS PRODUCT, AND OMRON HEREBY DISCLAIMS ANY OTHER WARRANTIES, EXPRESS OR IMPLIED, INCLUDING IMPLIED WARRANTIES OF MERCHANTABILITY AND FITNESS FOR A PARTICULAR PURPOSE. IMPLIED WARRANTIES AND OTHER TERMS THAT MAY BE IMPOSED BY LAW, IF ANY, ARE LIMITED IN DURATION TO THE PERIOD OF THE ABOVE EXPRESS WARRANTY. OMRON SHALL NOT BE LIABLE FOR LOSS OF USE OR ANY OTHER SPECIAL, INCIDENTAL, CONSEQUENTIAL OR INDIRECT COSTS, EXPENSES OR DAMAGES. This warranty provides you with specific legal rights, and you may have other rights that vary by jurisdiction. Because of special local requirements, some of the above limitations and exclusions may not apply to you. FOR CUSTOMER SERVICE Visit our web site at: Call toll free:
38 9. Guidance and Manufacturer s Declaration OMRON Automatic Blood Pressure Monitor Information for accompanying documents in the scope of IEC :2007 Model: BP786 including AC-adapter Important information regarding Electro Magnetic Compatibility (EMC) With the increased number of electronic devices such as PC s and mobile (cellular) telephones, medical devices in use may be susceptible to electromagnetic interference from other devices.electromagnetic interference may result in incorrect operation of the medical device and create a potentially unsafe situation. Medical devices should also not interfere with other devices. In order to regulate the requirements for EMC (Electro Magnetic Compatibility) with the aim to prevent unsafe product situations, the IEC standard has been implemented. This standard defines the levels of immunity to electromagnetic interferences as well as maximum levels of electromagnetic emissions for medical devices. Medical devices manufactured by OMRON Healthcare conform to this IEC :2007 standard for both immunity and emissions. Nevertheless, special precautions need to be observed: The use of accessories and cables other than those specified by OMRON, with the exception of cables sold by OMRON as replacement parts for internal components, may result in increased emission or decreased immunity of the device. The medical devices should not be used adjacent to or stacked with other equipment. In case adjacent or stacked use is necessary, the medical device should be observed to verify normal operation in the configuration in which it will be used. Refer to further guidance below regarding the EMC environment in which the device should be used. The MEDICAL ELECTRICAL EQUIPMENT BP786 including AC-adapter needs special precautions regarding EMC and needs to be installed and put into service according to the EMC information provided in this documentations. The Essential Performance of the BP786 including AC-adapter is to measure a blood pressure and a pulse rate and using the memory function. The BP786 including AC-adapter may be interfered with by other equipment, even if that other equipment complies with CISPR EMISSION requirements. EN Guidance and manufacturer s declaration - electromagnetic immunity OMRON BP786 including AC-adapter is intended for use in the electromagnetic environment specified below. The customer or the user of this OMRON BP786 including AC-adapter should assure that it is used in such environment. Emissions test Compliance Electromagnetic environment - guidance RF emissions CISPR 11 Group 1 The OMRON BP786 including AC-adapter uses RF energy only for its internal function. Therefore, its RF emissions are very low and are not likely to cause any interference in nearby electronic equipment. RF emissions CISPR 11 Class B Harmonic emissions IEC Voltage fluctuations/ flicker emissions IEC Class A Complies The OMRON BP786 including AC-adapter is suitable for use in all establishments, including domestic establishments and those directly connected to the public lowvoltage power supply network that supplies buildings used for domestic purposes. 35
39 9. Guidance and Manufacturer s Declaration Guidance and manufacturer s declaration - electromagnetic immunity OMRON BP786 including AC-adapter is intended for use in the electromagnetic environment specified below. The customer or the user of this OMRON BP786 including AC-adapter should assure that it is used in such environment. Immunity test IEC test level Compliance level Electromagnetic environment - guidance Electrostatic discharge (ESD) IEC Electrical fast transient/burst IEC Surge IEC Voltage dips, short interruptions and voltage variations on power supply inputlines IEC ±6 kv contact ±8 kv air ±2 kv for power supply lines ±1 kv for input/output lines ±1 kv line(s) to line(s) ±2 kv line(s) to earth <5 % UT (>95 % dip in UT) for 0.5 cycle 40 % UT (60 % dip in UT) for 5 cycles 70 % UT (30 % dip in UT) for 25 cycles <5 % UT (95 % dip in UT) for 5 sec. ±6 kv contact ±8 kv air Power frequency (50/60 Hz) magnetic field IEC A/m 3 A/m Note: UT is the A.C. mains voltage prior to application of the test level. ±2 kv for power supply lines ±1 kv for input/output lines ±1 kv line(s) to line(s) ±2 kv line(s) to earth <5 % UT (>95 % dip in UT) for 0.5 cycle 40 % UT (60 % dip in UT) for 5 cycles 70 % UT (30 % dip in UT) for 25 cycles <5 % UT (95 % dip in UT) for 5 sec. Floor should be wood, concrete, or ceramic tile. If floors are covered with synthetic material, the relative humidity should be at least 30 %. Mains power quality should be that of a typical commercial and/or hospital environment. Mains power quality should be that of a typical commercial and/or hospital environment. Mains power quality should be that of a typical commercial and/or hospital environment. If the user of the OMRON BP786 including ACadapter requires continued operation during power mains interruption, it is recommended that the OMRON BP786 including AC-adapter be powered from an uninterruptible power supply. Power frequency magnetic fields should be at levels characteristic of a typical location in a typical commercial or hospital environment. 36
40 9. Guidance and Manufacturer s Declaration Guidance and manufacturer s declaration - electromagnetic immunity OMRON BP786 including AC-adapter is intended for use in the electromagnetic environment specified below. The customer or the user of this OMRON BP786 including AC-adapter should assure that it is used in such environment. Immunity test IEC test level Compliance level Electromagnetic environment - guidance Portable and mobile RF communications equipment should be used no closer to any part of the OMRON BP786 including AC-adapter including cables, than the recommended separation distance calculated from the equation appropriate to the frequency of the transmitter. Conducted RF IEC V rms 150 khz to 80 MHz 3 V rms Recommend separation distance d = 1.2 P EN Radiated RF IEC V/m 80 MHz to 2.5 GHz 3 V/m d = 1.2 P 80 MHz to 800 MHz d = 2.3 P 800 MHz to 2.5 GHz where P is the maximum output power rating of the transmitter in watts (W) according to the transmitter manufacturer and d is the recommended separation distance in meters (m). Field strengths from fixed RF transmitters as determined by an electromagnetic site survey, a should be less than the compliance level in each frequency range. b Interference may occur in the vicinity of equipment marked with the following symbol: Note1: At 80 MHz and 800 MHz, the higher frequency range applies. Note2: These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection from structures, objects, and people. a Field strengths from fixed transmitters, such as base stations for radio (cellular/ cordless) telephones and land mobile radio, AM and FM radio broadcast, and TV broadcast cannot be predicted theoretically with accuracy. To assess the electromagnetic environment due to fixed RF transmitters, an electromagnetic site survey should be considered. If the measured field strength in the location in which the OMRON BP786 including AC-adapter is used exceeds the applicable RF compliance level above, the OMRON BP786 including AC-adapter should be observed to verify normal operation. If abnormal performance is observed, additional measures may be necessary, such as reorienting or relocating the OMRON BP786 including AC-adapter. b Over the frequency range 150 khz to 80 MHz, field strengths should be less than 3 V/m. 37
41 9. Guidance and Manufacturer s Declaration Recommended separation distance between portable and mobile RF communications equipment and the OMRON BP786 including AC-adapter OMRON BP786 including AC-adapter is intended for use in an electromagnetic environment in which radiated RF disturbances are controlled. The customer or the user of this OMRON BP786 including AC-adapter can help prevent electromagnetic interference by maintaining a minimum distance between portable and mobile RF communications equipment (transmitters) and the OMRON BP786 including AC-adapter as recommended below, according to the maximum output power of the communications equipment. Separation distance according to frequency of transmitter in meter Output Power of Transmitter in Watt 150 khz to 80 MHz 80 MHz to 800 MHz 800 MHz to 2.5GHz d = 1.2 P d = 1.2 P d = 2.3 P For transmitters rated at a maximum output power not listed above, the recommended separation distance d in meters (m) can be estimated using the equation applicable to the frequency of the transmitter, where P is the maximum output power rating of the transmitter in watts (W) according to the transmitter manufacturer. Note: At 80 MHz and 800 MHz, the separation distance for the higher frequency range applies. Note: These guidelines may not apply in all situations. Electromagnetic propagation is affected by absorption and reflection from structures, objects, and people. 38
42 E GI F A B C D K L G H N O P Q HJ W X Y Z R AA M S T U V AB AC AD
43 1 2 b A c L d a M
44 Manufactured for/fabricado para : OMRON HEALTHCARE Co., Ltd. 53, Kunotsubo, Terado-cho, Muko, Kyoto, JAPAN Distributed by/distribuido por : OMRON HEALTHCARE, INC West Field Court Lake Forest, IL U.S.A OMRON HEALTHCARE, INC. Made in China/Hecho en China
INSTRUCTION MANUAL 10 series Plus Blood Pressure Monitor with ComFit TM Cuff Model BP791IT ENGLISH ESPAÑOL TABLE OF CONTENTS Before Using the Monitor Introduction..................................................4
User Guide Microsoft Screen Sharing for Lumia Phones (HD-10) Issue 1.1 EN-US User Guide Microsoft Screen Sharing for Lumia Phones (HD-10) Contents Safety 3 About your accessory 4 Keys and parts 5 Connect
User Guide for the Wireless Headset (HS-11W) 9235346 Issue 1 DECLARATION OF CONFORMITY We, NOKIA CORPORATION declare under our sole responsibility that the product HS-11W is in conformity with the provisions
User Guide Nokia Portable Wireless Charging Plate DC-50 Issue 1.1 EN User Guide Nokia Portable Wireless Charging Plate DC-50 Contents For your safety 3 About your accessory 4 Keys and parts 5 Top up your
Upper Arm Blood Pressure Monitor SC 6800 Instructions for use Contents page Introduction... 3 Features of the SC 6800. 3 Safety instructions... 4 WHO-Interpretation of blood pressure readings... 4 Device
ITC-BTTN Cellular Bluetooth Gateway Owner s Manual 1 2 Table of Contents Introduction...3 Package Contents...3 XLink Connections Diagram...4 Setup...5 Pairing your Bluetooth Cell Phone to the XLink...6
HANDS-FREE BLUETOOTH CONTROLLER for SMART READY TABLETS AND COMPUTERS Operating Manual 2 CONTENTS Before Turning on Your PED... 4 Power... 5 ON... 5 OFF... 5 Modes... 6 Changing the MODE... 7 Connecting
1200093 User s Guide Weather Radio Alarm Clock Thank you for purchasing your Weather Radio Alarm Clock from RadioShack. Please read this user s guide before installing, setting up, and using your new weather
Alarm Clock USER GUIDE Jazwares, Inc. 2012 CONTENTS Please read the instructions along with the Alarm Clock carefully before you use it, so that you can operate it conveniently. WELCOME & Warnings Page
Business Audio System: Music & Messaging MP3 Player by Grace Digital Audio User Guide Model No. GDI-USBM10 User Guide Contents Introduction 2 Safety & General Use Information 2 Features 3 Set Up & Operation
TM 16-1003 User s Guide 7 High-Resolution Digital Photo Frame One demo photo included, as illustrated Please read this user s guide before using your new photo frame. Package contents Photo Frame AC Adapter
Model: 616-146v2 Quick Setup Guide DC: 071015 Atomic Projection Alarm with Indoor and Outdoor Temperature Snooze/Backlight BUTTONS Time, Alarm with Snooze, & Calendar Projection Arm Rotates 180 Indoor/Outdoor
BTH240 Wireless Bluetooth Headphone User Manual 2012 Kinivo LLC. All rights reserved. Kinivo is a trademark of Kinivo LLC. Bluetooth word mark and the Bluetooth logo are registered trademarks of Bluetooth
1 BLUETOOTH KEYBOARD for Tablets 410-2153-001B / AKB33US 1 YEAR LIMITED WARRANTY: We pride ourselves on the quality of our products. For complete warranty details and a list of our worldwide offices, please
TM Caller-ID on your TV! User s Manual and Installation Guide Copyright 1998 NetMedia Inc., All rights reserved. Rev. 061898 Thank you for your purchase of Caller-TV! You are now ready to enjoy the convenience
U.S. Cellular Home Phone 1 Contents Getting to Know Your Device...5 Appearance... 5 LED Indicators... 6 Device Installation...7 Before You Begin... 7 Installing the Battery... 7 Installing the Power Adapter...
PART NO. CMA113 MADE IN CHINA 1. Measuring CAT II 2. Max. voltage 250V ~ 3. Max. current 71 Amp Getting started with Electricity consumption monitoring single phase for homes and some smaller light commercial
THE NEW GENERATION IN TECHNOLOGY NI-3103A Alarm Clock w/fm Radio ipod & iphone Docking Operating Instructions Table of Contents Important Safeguards... 3 Parts... 6 Controls... 6 To Set Up... 7 ipod/iphone
Rechargeable Power Pack User s Manual 2013 VTech Printed in China 91-009656-000 US INTRODUCTION The Rechargeable Power Pack makes it easier than ever to keep the InnoTab 3 or InnoTab 3S charged and ready
INSTRUCTION MANUAL STRAPLESS HEART RATE MONITOR Model: HR-210 ENGLISH table TABLE of OF CONTENTS Before using the Monitor Introduction... 3 Important Safety Information...4 Operating the Device...4 Care
Targus Wireless RF Mouse Visit our Web site at: www.targus.com Features and specifications are subject to change without notice. 2004 Targus Group International and Targus, Inc.. 400-0111-001B USER S GUIDE
USER GUIDE The A+K Airgoo Digital Signage MicroPlayer is a totally new generation of product with wireless capabilities and innovative features. It is designed to operate with our Digital Signage Server
Home Automation, Inc. Model 70A00-1 GSM Cellular Communications Center Operation Manual Document Number 70I00-1 Rev A August, 2009 Contents Description... 1 Use with security systems...1 Everyday use...2
The Third Rail System User Manual System Contents: Slim Case and Smart Battery for iphone 4 Slim Case for iphone 4 Smart Battery micro-usb Cable USB Adapter Cable User Manual Smart Battery only Smart Battery
410-1681-001F / AMP13US 1 YEAR LIMITED WARRANTY: We pride ourselves on the quality of our products. For complete warranty details and a list of our worldwide offices, please visit www.targus.com. Features
Owner s Manual 1 Dear Customer, Thank you for purchasing this product. For optimum performance and safety, please read these instructions carefully. User Memo: Date of purchase: Dealer name: Dealer address:
ES-3305P V2 / ES-3308P V2 Quick Installation Guide 09-2013 / v1.0 1 COPYRIGHT Copyright Edimax Technology Co., Ltd. all rights reserved. No part of this publication may be reproduced, transmitted, transcribed,
Hands-free phone system features (for cellular phone) This system supports Bluetooth, which allows you to make or receive calls without using cables to connect a cellular phone and the system, and without
ReadyNet Easy Jack 2 Voice/Data and Data Only Owner s Manual PX-211d and PX-211v Phonex Broadband Corporation dba ReadyNet 6952 High Tech Drive Midvale, Utah 84047 801.566.0100 Phone 801.566.0880 Fax www.readynetsolutions.com
Panasonic Proprietary Telephone for Electronic Modular Switching System MODEL NO. KX-17030 Illustrated Model: White 1 KX-T7030 is compatible with all of the Panasonic Electronic Modular Switching Systems
Power Max Level 2 Charging Station en Installation and Operating Instructions This equipment has been tested and found to comply with the limits for a Class B digital device, pursuant to part 15 of the
33-1165.fm Page 1 Tuesday, August 10, 1999 10:50 AM Cat. No. 33-1165 OWNER S MANUAL Please read before using this equipment. 900 MHz Lightweight Wireless Stereo Headphones 33-1165.fm Page 2 Tuesday, August
ZTE WIRELESS HOME PHONE BASE USER GUIDE INTRODUCTION HOW IT WORKS Thank you for choosing Consumer Cellular! We know you re excited to use your new Wireless Home Phone Base and this short guide will help
AIRAVE 2.5 User Guide 2012 Sprint. Sprint and the logo are trademarks of Sprint. Other marks are trademarks of their respective owners. 911702 Rev. 51.13 Important Messages IMPORTANT: Sprint s policies
ES-CAM2AU IP Camera www.etiger.com EN Box content Connect your smartphone to the camera - ES-CAM2AU x1 - Mounting bracket x1 - AC Adaptor x1 - User Manual x1 and apps Before using the ES-CAM2AU, download
W O R L D C L A S S H I F I O W N E R S M A N U A L Accessories: NAPSC, i-supply, Headline, Stageline IMPORTANT In order to comply with current European safety regulations it is essential that the Naim
WAVE BLUETOOTH MUSIC ADAPTER FOR WAVE SYSTEMS Owner s Guide Bedienungsanleitung Guía de usuario Notice d utilisation Manuale di istruzioni Gebruiksaanwijzing Tab 8, 16 Tab 7, 15 Tab 6, 14 Tab 5, 13 Tab
User s Manual Bluetooth Calculator Keypad Page Regulatory Compliance This device complies with Part 15 of the FCC Rules. Operation is subject to the following two conditions: (1) This device may not cause
Technical Support Thank you for selecting NETGEAR products. After installing your device, locate the serial number on the label of your product and use it to register your product at www.netgear.com/register.
Mercury Helios 2 ASSEMBLY MANUAL & USER GUIDE TABLE OF CONTENTS INTRODUCTION...1 1.1 MINIMUM SYSTEM REQUIREMENTS 1.1.1 Apple Mac Requirements 1.1.2 PC Requirements 1.1.3 Supported PCIe Cards 1.2 PACKAGE
USER GUIDE Digital Satellite Receiver Models DSR207, DSR317, DSR505, and DSR530 CONTENTS IMPORTANT SAFETY INSTRUCTIONS...1 DSR BASICS SYMBOLS/ICONS...6 Front Panel...6 Back Panel...7 CONNECTING YOUR DSR...10
BT-105 Transceiver Product Family Includes DIGIT and ATFS-2 User Manual Table of Contents BT-105... 4 Description... 4 Setup and Pairing... 5 Resetting the Pairing... 5 Turning off the BT-105... 6 Modes...
INFINEA TAB USER MANUAL INFINEA ipad TAB CONTACT INFORMATION National Sales Headquarters: Infinite Peripherals, Inc. 1124 Main Street Suite B Irvine, CA 92614 Toll Free: (866) 278-7860 Office: (949) 222-0300
VOIP Business Phone User Guide Model 25630/25600 MGCP Please read this manual before operating the product for the first time. Interference Information This device complies with Part 15 of the FCC Rules.
BroadBand PowerShield User Manual 990-0375G 12/2006 Chapter 1 General Information The PowerShield provides a power source for broadband telephony and other DC applications. Safety This Safety Guide contains
User Guide for the Kidde RemoteLync Remote Notification Device 120V AC 3 Color LED Indicator Voice Message System For questions concerning your device, please call our Product Support Line at 1-800-880-6788.
reflecta Super 8 Scanner User Manual 1 FEDERAL COMMUNICATIONS COMMISSION (FCC) STATEMENT This Equipment has been tested and found to comply with the limits for a class B digital device, pursuant to Part
CINEMATE 1 SR DIGITAL HOME THEATER SPEAKER SYSTEM Operating Guide Guía del usuario Guide d utilisation Notice This equipment has been tested and found to comply with the limits for a Class B digital device,
ST815 Illumination Sensor with LCD The Illumination Sensor with LCD (refer to as Illumination Sensor hereafter) is a Z-Wave TM enabled device which is fully compatible with any Z-Wave TM enabled network.
P516-098 WPR400 Wireless Portable Reader User guide Para el idioma español, navegue hacia www.schlage.com/support. Pour la portion française, veuillez consulter le site www.schlage.com/support. Contents
Manufacturer: Phonak AG Laubisrütistrasse 28 8712 Stäfa Switzerland www.audeoworld.com 029-3110-02/V1.00/2011-09/na Phonak AG All rights reserved User Guide Audéo PFE 232 with microphone 2 3 Congratulations
Ultrasound Technologies PD1v vascular pocket doppler Operating Instructions Before using your Pocket Doppler for the first time, please read these operating instructions carefully. 1 INTRODUCTION The UltraTec
IDEAL INDUSTRIES, INC. TECHNICAL MANUAL MODELS: 61-763 61-765 The Service Information provides the following information: Precautions and safety information Specifications Performance test procedure Calibration
Home Care Provider Setup Instructions Always use these instructions along with the User Manual when assembling or adjusting this equipment. For clinical systems, refer to the setup guide entitled Respironics
Instruction Bulletin EAV85226 08/2014 Wiser Panel Meter, Model Number WISERCTPM200 Installer s Guide Retain for future use. Product Description Kit Contents The Wiser Panel Meter is for use in energy management
2 Port KVM Switch User Manual CS-82A Read this guide thoroughly and follow the installation and operation procedures carefully in order to prevent any damage to the units and/or any devices that connect
SoftRAID 5 QUICK START GUIDE for OWC ThunderBay TABLE OF CONTENTS INTRODUCTION...1 1.1 MINIMUM SYSTEM REQUIREMENTS 1.2 FEATURES 1.3 ABOUT THIS MANUAL SYSTEM SETUP...2 2.1 GETTING STARTED 2.2 INITIALIZING,
SMART LENS Camera User Manual About this Manual Thank you for purchasing the KODAK PIXPRO SMART LENS Camera. Please carefully read this manual and keep it for future reference. JK Imaging Ltd. reserves
Single Day Alarm Clock with Bluetooth and USB Play-through MODEL : BSC 200 READ THIS INSTRUCTION MANUAL BEFORE OPERATING THE APPLIANCE. SAVE THIS MANUAL FOR FUTURE REFERENCE. TABLE OF CONTENTS IMPORTANT
2 732 Synco living Water detector QFP910 Wireless radio sensor to record water puddles RF communication based on KNX standard (868 MHz unidirectional) Battery operation with standard 1.5 V batteries Use
No. S8351 S8351 External Bluetooth module for mz series transmitter OPERATING INSTRUCTION Prior to use, please read this manual thoroughly. Keep this manual in a convenient place for quick and easy reference.
ITEM #0422362 WIRELESS WTER LEK DETECTOR MODEL #ST812-2 Español p. XX REQUIRES: IRIS Smart Hub FOR FUTURE REFERENCE TTCH YOUR RECEIPT HERE ND SVE THIS MNUL Z-Wave is a registered trademark of Sigma Designs
TM Advanced Tracking Technologies, Inc. 2003 ATTI All rights reserved Copyright 2003, Advanced Tracking Technologies, Inc. All rights reserved. No part of this publication may be reproduced or transmitted
T13C Table Updater for Seating Management U S E R M A N UA L Usage, Installation, Warranty and Service Information Long Range Systems, LLC 4550 Excel Parkway, Suite 200 Addison, TX 75001 800.437.4996 www.lrsus.com
G-BUZZ PORTABLE ALARM CLOCK Dual Alarm FM Radio Wake-up Sounds USB Phone Charger G-1CR Welcome Alarm clocks can be boring. Get ready to shake things up with your new G-BUZZ. Slap the snooze for more ZZZ
ENERGY AWARE Monitor User Guide Table of contents Introduction 3 How it works 4 Have you got everything? 4 Before you get started 5 Setting up your TREC Display 6 Pairing the TREC Display and Transmitter
Video Splitter USER MANUAL VS92A / VS94A / VS98A FCC Information This equipment has been tested and found to comply with the limits for a Class B digital device, pursuant to Part 15 of the FCC Rules. These
FCC COMPLIANCE STATEMENT FOR AMERICAN USERS This equipment has been tested and found to comply with the limits for a CLASS A digital device, pursuant to Part 15 of the FCC Rules. These limits are designed
Portable Air Conditioner OWNER S MANUAL Read these instructions before use Model: MM14CCS Customer Support : 1-800-474-2147 Voltage rating: 115V~60Hz Power rating : 1400W For product inquiries or support
User Guide SURFboard SB6183 DOCSIS 3.0 Cable Modem 2014 ARRIS Enterprises, Inc. All rights reserved. No part of this publication may be reproduced in any form or by any means or used to make any derivative
Intruder Alarm System GSM & PSTN Wireless and Wired Touch Model 2013-01 V1.0.0 Statment This equipment must be installed and used in strict accordance with the instructions given in this user documentation.
StorTrends 3400 Hardware Guide for Onsite Support MAN-3400-SS 11/21/2012 Copyright 1985-2012 American Megatrends, Inc. All rights reserved. American Megatrends, Inc. 5555 Oakbrook Parkway, Building 200
Jarv Nmotion BTHL-201 Bluetooth Stereo Earbuds Users Guide Features: Bluetooth wireless technology Comfortable lightweight in-ear design with detachable sports ear hooks for added support High Definition
WIRELESS EARPHONES with ACTIVE NOISE CANCELLING Thank you for purchasing PHIATON BT 100 NC - Please follow the directions, and read the guidelines carefully before use. Please keep the owner s guide for
THERMAWAVE CERAMIC HEATER Model HZ-850 Series Model HZ-860 Series IMPORTANT SAFETY INSTRUCTIONS WARNING READ AND SAVE THESE OPERATING AND SAFETY INSTRUCTIONS BEFORE USING THIS HEATER. Warning Failure to
Quick Start Guide ATTENTION RISQUE D ÉLECTROCUTION! NE PAS OUVRIR! CAUTION: TO REDUCE THE RISK OF ELECTRIC SHOCK, DO NOT REMOVE COVER (OR BACK). NO USER-SERVICEABLE PARTS INSIDE. REFER SER- VICING TO QUALIFIED
dedicated KVM switch and rackmount screen technology User Manual RK-2d / RK-2t 1U Short Depth Keyboard Drawer Designed and manufactured by Austin Hughes 751 Legal Information First English printing, October
Portable Air Conditioner OWNER S MANUAL Read these instructions before use 8 Model: MN12CES / MN10CESWW Voltage rating: 120V~60Hz Power rating : 1100W (MN12CES) Power rating : 900W (MN10CESWW) Customer
Outdoor Camera Set-Up Guide RC8221 Iris Camera Set-Up Guide 2 2 A B Package Contents Basic A. Outdoor Setup Camera This B. chapter Power provides Adapter details on how to setup and mount the camera. C.
MEMO ALARM CLOCK Operating Manual INTRODUCTION Congratulations on purchasing this state-of-the-art Alarm clock with radio controlled time. The operation of this product is simple and straightforward and