1 Calibration & Preventative Maintenance Sally Wolfgang Manager, Quality Operations Merck & Co., Inc.
2 Calibration: A comparison of two instruments or measuring devices one of which is a standard of known accuracy (traceable to national standards) to detect, correlate, report or eliminate by adjustment, any discrepancy in accuracy of the instrument measuring device being compared to the standard.
3 Calibration Programs Required by Regulatory Authorities Automatic, mechanical, or electronic equipment or other types of equipment, including computers, or related systems that will perform a function satisfactorily, may be used in the manufacture, processing, packing, and holding of a drug product. If such equipment is so used, it shall be routinely calibrated, inspected, or checked according to a written program designed to assure proper performance. Written records of those calibration checks and inspections shall be maintained. US Code of Federal Regulations, 21 CFR , 1/20/95 Maintenance at appropriate intervals to prevent malfunction & shall be preventative not reactive maintenance. US Code of Federal Regulations, 21 CFR
4 Calibration Programs Required by Regulatory Authorities Calibration requirements for Laboratory Instruments US Code of Federal Regulations, 21 CFR Specific Directions Schedules Limits of accuracy & precision Remedial Actions Systems to prevent usage of instruments failing calibration Control, weighing, measuring, monitoring and test equipment that is critical for assuring the quality of intermediates or APIs should be calibrated according to written procedures and an established schedule. ICH Good Manufacturing Practice Guide For Active Pharmaceutical Ingredients, 10 November 2000.
5 Each Manufacturing Area: Written calibration procedures that use traceable calibration standards or calibration equipment. Preventive maintenance procedures and / or referenced manuals Qualified individuals (having the appropriate education, training, background and experience) responsible for calibrating & maintaining instrumentation Second person check of all calibration and maintenance Qualified individuals responsible for monitoring the calibration and maintenance program. Ensure the calibration program and procedures are reviewed and approved by Quality
6 Instrument / Equipment Master List System for identification of all GMP related instrumentation in a manufacturing area or laboratory Include instrumentation details (serial number, model number & location) If automation components are tracked separately through the configuration management then it is not necessary to include (must verify) Procedures must exist that identify the calibration and maintenance requirements for each instrumentation / equipment on the master list Retired Equipment: Records pertaining to retired / obsolete equipment must be kept according to company s records retention procedures Records should include date the unit was retired, person responsible and the reason for retirement / discard
7 Instrument Identification & Calibration Status Each instrument given a unique identifier Instrumentation details associated with this number must be documented and available (e.g. serial number, model number, location, etc.) Each instrument should be labeled with the unique identifier Calibration status of each instrument, the date of calibration, the next calibration date and the identification of person performing calibration should be readily available Appropriate systems to document calibration status include calibration logs, MAXIMO, and calibration stickers
8 Instrument Identification & Calibration Status System must be in place to prevent use of an instrument that is not qualified, unusable due to damage or malfunction, or has exceeded its established calibration interval System must be in place that identifies instruments that do not require calibration to be performed beyond the original or factory calibration to distinguish from those instruments that do require scheduled calibrations Documentation required for excluding equipment
9 Traceability of Standards / Calibration Equipment Calibration reference standards / calibration equipment shall be traceable to national standards and be accompanied by certificates of traceability / analysis If recognized standards are not available, an independent reproducible standard may be used The calibration tolerance of a given standard should be as tight or tighter than the tolerance of the instrument to be calibrated A procedure must be in place to ensure tracking and monitoring of standard s expiration date and re-calibration / re-certification requirements Re-calibration records must be retained
10 Instrument Calibration Tolerances Instrument calibration tolerance limits should be established so problems are identified and corrected in a timely manner When assigning tolerances, considerations given to: Capability of the instrument being calibrated (what the manufacturer claims the instrument can achieve). Parameters at which the instrument operates (ex: if testing accuracy of + 0.5% is required, the instrument calibration tolerances should be <0.5%) Work environment - environmental conditions can affect the performance of the instrumentation
11 Instrument Calibration Tolerances Practice of using Alert & Action levels Alert Tolerance ( Adjustment Limit ) Related to instrument performance Level at which the instrument is adjusted back into range Not required - an industry best practice
12 Instrument Calibration Tolerances Action Tolerance ( Calibration Limit or Out-of-Tolerance ) Tied to process performance Level at which the potential for product impact exists Possible atypical investigation or reporting is required Deviations beyond Alert level but not at Action level would not require investigation - May require adjustment, changes to PMs Setting of Alert and Action levels should be described in SOPs, be defendable and have Quality review and approval
13 Calibration and Maintenance Frequencies May be determined for individual instruments or groups of instruments (similarity of construction, reliability, and stability) Considerations when determining calibration frequency: Accuracy of the measurement / instrument range Consequences of incorrect value caused by out of calibration Extent & criticality of use of the instrument & tendency to wear and drift The manufacturer s recommendations Environmental & physical conditions (temperature, humidity, vibration) Previous calibration records, history of calibration problems & repair history Frequency of calibration checks prior to use or in-between intervals Redundant / back-up systems (provides secondary source of information available from other calibrated primary instruments) Results of Qualification studies Process requirements Availability of built-in / automatic calibration checks
14 Calibration and Maintenance Frequencies Changes to frequency must be approved per change control SOPs References to specific instrument procedures listed in compendia (USP, BP or EP) for calibration tests for specific laboratory instruments Calibration time windows should be established around calibration due dates
15 Calibration and Maintenance Frequencies Policy for Calibration and Maintenance Intervals & Schedules Include: Extending Intervals Reducing Intervals Maximum Intervals
16 Maintenance Requirements Manufacturer s recommendations Parts that wear: gaskets, seals & bearings Parts requiring periodic replacement: filters, belts & fluids Parts requiring periodic inspection and cleaning Parts requiring periodic adjustments, tightening and lubrication
17 Calibration and Maintenance Procedures Shall include specific directions and limits for accuracy & precision Shall include guidance for remedial action when accuracy & precision limits are not met Normally provided in the manufacturer s manuals Some compendial requirements exist for some specific laboratory instrumentation Performance checks (e.g. system suitability, daily balance performance checks) are NOT suitable substitutes for regularly scheduled calibrations
18 Calibration and Maintenance Procedures Each calibration & maintenance procedure should include the following: Identification of department responsible to perform the calibration or maintenance Step-by-step calibration instructions, reference to appropriate calibration procedures or instrument manuals Methods for preventive maintenance or reference to appropriate instrumentation manuals Calibration equipment used in the calibration (e.g. spectroscopy filters, voltmeters, digital thermometers, etc) Calibration parameter and tolerance ( + )
19 Calibration and Maintenance Procedures Each calibration & maintenance procedure should include the following: Required environmental controls or conditions, where appropriate Provisions for adjustments, if needed Requirement for recording actual measurements before ( as found ) and after adjustment or preventive maintenance ( as left ) Note: as found recordings are not required where routine performance checks are available to provide evidence indicating instrument is operating properly and is suitable for use (e.g. HPLC system suitability, manual balance checks which encompass range of use, etc.)
20 Calibration and Maintenance Procedures Actions to be taken if instrumentation cannot be calibrated (e.g. contact appropriate service people, label and remove from service) A step to record all calibration & maintenance activities.
21 Use of Contractors / Vendor Service Personnel Must have a procedure for approving contractor activities Contractors must have appropriate training / qualification Responsible to review & approve the contractors calibration and maintenance procedures. Responsible to perform actions or steps not part of the contractors procedures.
22 Calibration & Maintenance Records All calibration records must be retained per document retention procedures Should include as found measurements, results of adjustments ( as left ) and appropriate review & approval of all results Tolerance or limit for each calibration point Identification of standard or test instrument used Identification of persons performing the work and checking the results with dates
23 Calibration & Maintenance Records Review must ensure the approved activities have been completed and all results have passed the established acceptance criteria Periodic review of historic calibration & maintenance data to evaluate appropriateness of established frequencies
24 Procedures for Out-Of-Tolerance Calibration Results Actions required when critical instrument found outside of established instrument calibration limits ( Action Limit,Out-Of-Tolerance, OOT): Critical Instrumentation: Any instrument employed in the production of bulk pharmaceutical chemicals, drug or biologics product that controls and/or is used to measure a parameter that affects the validation state of the product (Critical Process Parameters) and any other quality parameter designated as such. Also includes laboratory instrumentation used to measure conformance to component and/or product specifications.
25 Procedures for Out-Of-Tolerance Calibration Results Instrument Mechanic: Immediate review of calibration data to verify validity of OOT Notification to supervisor or responsible person
26 Procedures for Out-Of-Tolerance Calibration Results Instrument Services: Corrective Action to the instrument ASAP (repair, re-calibration, replacement or removal from service) Investigation and documentation into root cause of OOT Historical review of instrument performance & factors relevant to setting calibration frequency Corrective Actions to prevent recurrence Re-calibration after maintenance or repair OOT Notification issued to instrument owner (after results validated) OOT Notification should include current calibration data, magnitude of OOT error, and date of last successful calibration
27 Procedures for Out-Of-Tolerance Calibration Results Instrument Owner: Evaluate impact of OOT on product quality. Impact evaluation based on SOP which is approved by Quality and defines how / when Quality will be notified. Ex. of OOT with impact on quality: Instruments used in acceptance / rejection of materials and product Control / monitoring of critical process parameters Process controls affecting the final product quality or yield Instruments required by regulation (e.g. Magnahelic gauges, RH monitors, etc.) Calibration reference standards
28 Procedures for Out-Of-Tolerance Calibration Results Instrument Owner: Needs complete understanding of the process requirements for the parameter measured by the OOT instrument Evaluate magnitude of error in the As Found OOT vs. process requirements to determine quality impact Ex. of OOT with no impact on quality: A temperature transmitter is reading 0.2 o C low and is OOT for that instrument. There is no potential for quality impact if: - Process requires temperature reading accuracy of +/- 1 o C - Measures a range / CPP with max. 85 o C and process normally runs at 75 o C.
29 Procedures for Out-Of-Tolerance Calibration Results Decisions regarding impact on quality must be documented, approved by Quality and documentation retained OOT events should be tracked and trended to identify problem instruments Historical information on OOT events should be readily retrievable
30 Change Control Management Changes to calibration tolerances, frequency, procedures, addition to / deletion from calibration program, changes in location and different replacement parts should be documented in a change control program May also require re-calibration, re-execution / revision of instrument s Installation Qualification (IQ), Operational Qualification (OQ) or Performance Qualification (PQ) Update of qualification records: drawings, parts list, etc. Appropriate review and approval by responsible departments and Quality
31 FDA 483 Calibration Limits Manufacturing processes are not adequately controlled. For example the blender used was shown by calibration records to be in error by nearly one minute for a process that is specified by validation reports to be exactly four minutes. The 25% error in blender agitation timing has not been evaluated as to any effect on drug product quality.
32 FDA 483 Calibration Calibration of scales is scheduled twice a year, however, there is no provision to verify their accuracy with weights between calibrations.
33 FDA 483 Calibration Scale Calibration Records did not include: Percent variance between test readings and actual NIST weight Reproducibility studies and standard deviation calculations Accuracy of the scale regarding the number of decimal points Accuracy of the scale regarding the minimum / maximum weights
34 Out of Tolerance (OOT) for Critical Equipment Suggested OOT Flowchart Instrument Mechanic: Verifies Calibration Result Is Calibration Result Valid? No Invalidate, Document & Recalibrate Yes Instrument Services Group: Repairs & Recalibrates, Replaces, or takes out of service Instrument Services Group: Issues Out of Tolerance Notification Investigates Root Cause Plans & Documents Corrective Action Instrument Owner: Evaluates Potential Impact on Product Quality Complete OOT report & file in Maintenance No Is there potential for impact on product quality? Yes Are Quality-approved procedures in place that define the criteria for potential quality impact? Yes Plan & Document Corrective Action, Document & Retain Atypical Investigation Determine Disposition of Product Yes Notifies Quality Initiates Atypical Investigation Assess Impact on Product Quality Yes Is Quality Impacted?? No No Document & Retain Atypical Investigation
1 of 11 1. Introduction The International Safe Transit Association (ISTA), a non-profit association whose objective is to prevent product damage and excess packaging usage within the distribution environment.
International GMP Requirements for Quality Control Laboratories and Recomendations for Implementation Ludwig Huber, Ph.D. email@example.com Overview GMP requirements for Quality Control laboratories
RTP s NUCLEAR QUALITY ASSURANCE PROGRAM RTP operates under one quality program, whether you purchase products that are commercial grade, nuclear safety-related or industrial safety compliant (IEC 61508).
Pharmaceutical Wholesaler Site Inspection Checklist Date and time of inspection: Legal business name and plant name: Location (address): Phone number: Fax number: Email address: Web site URL: Contact Person:
EVALUATION AND ACCREDITATION OF SUPPLIER TEST FACILITIES GP 10 EVALUATION AND ACCREDITATION OF TESTING FACILITIES USED BY SUPPLIER OF MATERIALS TO GENERAL MOTORS PREFACE TO GP-10 General Motors expects
Guidance for Industry Investigating Out-of-Specification (OOS) Test Results for Pharmaceutical Production U.S. Department of Health and Human Services Food and Drug Administration Center for Drug Evaluation
COMPLIANCE BY DESIGN FOR PHARMACEUTICAL QUALITY CONTROL LABORATORIES INSIGHT FROM FDA WARNING LETTERS Primer CONTENTS INTRODUCTION...3 QUALITY AND COMPLIANCE IN QUALITY CONTROL LABORATORIES...5 Compliance
Calibration System Compliance in the Pharmaceutical Industry Speaker/Author: Kevin J. Daubert Schering-Plough 2000 Galloping Hill Road Kenilworth, NJ 07033, U.S.A. Phone: 908.740.6372 Fax: 908.740.6383
Page 1 of 13 Sections Included in this Document and Change History 1. Purpose 2. Scope 3. Responsibilities 4. Background 5. References 6. Procedure/(6.C.4. added or electronically ) 7. Definitions 8. Records
Supplier Quality Standard 1.0 Purpose The purpose of this Supplier Quality Standard is to communicate the expectations and requirements of Baxter Healthcare Corporation to its suppliers. These expectations
Page 1 Company Name: Date of audit: Date of last audit performed: Name of person performing self-audit: Signature: Name of person responsible for quality system: Signature: Number of non-compliances: Page
DATA ACCEPTANCE PROGRAM UL Client Test Data Program Investigations of products by UL typically involve the testing of product samples. While UL maintains extensive facilities for these purposes, customers
Hong Kong Guide to Good Manufacturing Practice for the Secondary Packaging of Pharmaceutical Products Version 1.3 Drug Office, Department of Health. Contents list 1. QUALITY MANAGEMENT... 4 2. PERSONNEL...
Far West Technology, Inc. ISO 9001 Quality Manual Document No.: QM Revision: 0 Issue Date: 27 August 1997 Approval Signatures President/CEO Executive Vice President Vice President/CFO Change Record Rev
QUALITY ASSURANCE MANUAL JPM OF MISSISSIPPI, INC. Hattiesburg, MS Revision E 01/19/11 Revised to ISO 9001:2008 on July 9, 2009 JPM OF MISSISSIPPI, INC. MANAGEMENT QUALITY POLICY It is the goal of JPM of
Standards for Laboratory Facilities Version 2.1 APPROVED 24 FEBRUARY 2014 Article I. General Provisions Section 1.01 The facility shall be of suitable size and construction to facilitate the proper conduct
Revision History COPY The top row of this table shows the most recent changes to this controlled document. For previous revision history information, archived versions of this document are maintained by
Validation and Calibration Definitions and Terminology ACCEPTANCE CRITERIA: The specifications and acceptance/rejection criteria, such as acceptable quality level and unacceptable quality level, with an
Checklist Standard for Medical Laboratory Name of hospital..name of Laboratory..... Name. Position / Title...... DD/MM/YY.Revision... 1. Organization and Management 1. Laboratory shall have the organizational
Procedure for Procurement and Receipt 1.0 Purpose - To ensure that all supplies, equipment and services purchased by the State Crime Laboratory (Laboratory) are properly procured and received. 2.0 Scope
NOTE TO USERS This Quality Agreement template was developed by the Bulk Pharmaceutical Task Force (BPTF), an affiliate organization of the Society of Chemical Manufacturers and Affiliates (SOCMA), as a
Medical Device Quality Agreement Template Prepared by Dan O Leary Ombu Enterprises, LLC 3 Forest Ave. Swanzey, NH 03446 603-209-0600 This document is intended to form the basis for a Supplier Agreement
Current Good Manufacturing Practice for PET Drugs - CGMP 21 CFR 212 CDER Office of Regulatory Policy Jane Axelrad, JD CDER Office of Compliance Brenda Uratani, Ph.D. CDER ONDQA Ravindra Kasliwal, Ph.D.
Content Sheet 3-1: Equipment Management Overview Role in quality management system Equipment management is one of the essential elements of a quality management system. Proper management of the equipment
Welcome Computer System Validation Training Delivered to FDA ISPE Boston Area Chapter February 20, 2014 1 Background Training Conducted on April 24, 2012 Food & Drug Administration Division of Manufacturing
Supplier Quality Assurance Preface We strive to continually improve our process performance by meeting quality objectives. It is imperative that our suppliers operate under the zero defects and 100% on
CMX Professional Calibration software with wide customization possibilities. CMX Professional Calibration software with unlimited possibilities. Who calibrated? What? When? What were the results? Plan,
March 2003 GMP 11 Good Measurement Practice for Assignment and Adjustment of Calibration Intervals for Laboratory Standards 1. Introduction 1.1. Purpose Measurement processes are dynamic systems and often
*Changes from the previous QASE noted by "yellow" highlight of block Evaluation Summary Company: Prepared By: Section Element Manual Audit OK Objective Evidence 2.1 Objective of Quality Assurance Program
Procedure for Equipment Calibration and Maintenance 1.0 Purpose This procedure specifies the schedule and requirements for calibration, performance verification, and maintenance of State Crime Laboratory
for Industry: Hong Kong Guide to GMP for the Secondary Packaging of Pharmaceutical Products Version 2.1 Drug Office Department of Health. Contents PURPOSE... 3 SCOPE... 3 GLOSSARY... 3 GUIDANCE FOR INDUSTRY...
Step-by-Step Analytical Methods Validation and Protocol in the Quality System Compliance Industry BY GHULAM A. SHABIR Introduction Methods Validation: Establishing documented evidence that provides a high
1 CALIBRATION PRINCIPLES After completing this chapter, you should be able to: Define key terms relating to calibration and interpret the meaning of each. Understand traceability requirements and how they
Inspection, Measuring and Test Equipment (IMTE) Calibration Purpose and Scope 2 Responsibilities 2 About IMTE 3 Master IMTE Log 4 Identification of IMTE Calibration Status 5 Handling and Storage of IMTE
Regulatory Requirements for Medical Device Calibration Programs Dan O Leary CBA, CQA, CQE, CRE, SSBB, CIRM President Ombu Enterprises, LLC Dan@OmbuEnterprises.com www.ombuenterprises.com 603-209-0600 1
MARK INTEGRITY PROGRAM UL Calibration Requirements: Equipment Used for UL/C-UL/ULC Mark Follow-Up Services UL defines minimum requirements for calibration of inspection, measuring and test equipment (IMTE)
IMP management at site Dmitry Semenyuta TOP 5 FDA inspections finding 1999-2009 Center of Drug Evaluation and Research (CDER) Failure to follow the protocol Failure to keep adequate and accurate records
This checklist is intended to help laboratories improve the quality of their testing programs with regard to meat and poultry testing samples, in order to ensure those laboratories are at least equal to
White paper System Suitability Tests and AIQ R.D.McDowall PhD Principal, McDowall Consulting Paul Smith European Validation Program Manager, PerkinElmer Nicola Vosloo PhD European Market Development Leader,
INTRODUCTION This document describes an inspection system broken down into four separate levels. When Inspection Level IV is invoked, this system meets the requirements of former MIL-I-45208A. The primary
Accurate & Efficient Weighing Good Weighing Practice in Modern Industry Erik Volek Lab Sales Consultant in Upstate NY and Western PA Jack Blakley Industrial Business Development Manager Accurate and Efficient
NIST HANDBOOK 150-5 CHECKLIST CONSTRUCTION MATERIALS TESTING Instructions to the Assessor: This checklist addresses specific accreditation criteria prescribed in NIST Handbook 150-5, Construction Materials
Luminus Devices, Inc Quality Management Systems Manual ISO 9001:2008 This document belongs to Luminus Devices, Inc. It cannot be reproduced without authorized authority. Area: Quality System Document Page
DEBUNKING THE TWO GREAT MYTHS ABOUT CALIBRATION: TRACEABILITY TO NIST IF YOU CANNOT ADJUST, YOU CANNOT CALIBRATE! BY DEFINITION First, let s get everyone on the same page with the definitions of two critical
Validation of Process Gas Systems By Jeff Hargroves Alza Corporation Validation of process gas systems involve documenting the expected system behavior, and verifying that the system performs as expected.
CHAPTER 7 QUALITY ASSESSMENT Chapter 7 QUALITY ASSESSMENT 7.1 OVERVIEW OF CBB-SPECIFIC QUALITY ASSURANCE/QUALITY CONTROL PROGRAM An extensive, site-specific Quality Assurance/Quality Control (QA/QC) program
Implementing New USP Chapters for Analytical Method Validation March 2010 Ludwig Huber Fax.: +49 7243 602 501 E-mail: Ludwig_Huber@labcompliance.com Today s Agenda Handling Method Changes vs. Adjustments
Quality Policy Manual (ISO 9001:2000) Page 1 of 21 Management Approval of Quality Manual The signatures that appear below indicate that the executive management of has reviewed and approved the following
Corrective and Preventive Actions A Five Step Approach Topics to Be Covered What is CAPA? Governing authority Five steps to a good CAPA process Where companies have difficulty Example citations Recap What
Appendix A manual Example of a food company quality Contents Date: 13/03/95 RME-QLMN-OO Page 1 of 3 Section Title ISO 9001 reference 01 In trod uction 02 Purpose 03 Scope 04 Definitions 05 Management responsibility
This document summarizes the additional ISO/TS 16949:2009 requirements, beyond those included from ISO 9001:2008. ISO 9001:2008 clauses and titles are boxed to set them off from the ISO/TS 16949:2009 requirements.
Risk-Based Validation and Requalification of Processes & Equipment Nancy Tomoney Associate Validation Manager QPharma Inc. Order of Operations US Predicate law always comes first US Guidances Other non
Page 2 of 15 Document Revision History Revision Date Author Description of change 10 07Jun13 Mark Benton Removed Admin. Manager from approval 12Feb13 Mark Benton 08 01Oct12 Mark Benton 07 8/30/2012 Refer
World Health Organization WHO Technical Report Series, No. 908, 2003 Annex 9 Guide to good storage practices for pharmaceuticals 1 1. Introduction 125 2. Glossary 126 3. Personnel 128 4. Premises and facilities
Effective Personnel Qualification Does the Job Description Play a Role in CGMP Compliance? Eldon Henson Critical quality attributes API QbR EMEA CALIBRATION Design space Deviations IQ ICH Maintenance Validation
Minimizing Calibration Costs for Measuring and Test Equipment EASA Convention 2015 Henry B. Gonzalez Convention Center San Antonio, TX June 15, 2015 Presented by Mike Howell Technical Support Specialist
GLP vs GMP vs GCP Dominique Pifat, Ph.D., MBA The Biologics Consulting Group firstname.lastname@example.org Common Misconception Good Laboratory Practices 1) A quality system concerned with the organizational process
Schweppes Australia Head Office Level 5, 111 Cecil Street South Melbourne Victoria 3205 www.schweppesaustralia.com.au Quality Management Systems 1. Quality Management Systems develop, implement, verify
ISO 9001 Quality Systems Manual Revision: D Issue Date: March 10, 2004 Introduction Micro Memory Bank, Inc. developed and implemented a Quality Management System in order to document the company s best
Page 1 of 6 U.S. Food and Drug Administration Protecting and Promoting Your Health Ambco Electronics, A California Corp 9/26/14 Department of Health and Human Services Public Health Service Food and Drug
ISO 9001:2008 QUALITY MANUAL Revision B Because we want you to achieve the highest levels of performance, we connect care Because with leading we want science you to achieve continuously the highest improve,
Library Guide: Pharmaceutical GMPs Table of Contents Overview...3 Courses Listed by Functional Area... 4 Course Descriptions: A Step-by-Step Approach to Process Validation (PHDV79)... 7 A Tour of the FDA
NC SBI QUALITY ASSURANCE PROGRAM for the SBI Reviewed by: Deputy Assistant Director Bill Weis Date: Approved by: Assistant Director Jerry Richardson Date: Originating Unit: SBI Effective Date: July 25,
18.104.22.168 The NTEP Certificate of Conformance is updated periodically to provide information consistent with current practices of NTEP. NOTE: During the phase in period, NTEP will use special procedures
GE Measurement & Control Solutions 4Sight Calibration Management Software 4Sight calibration and maintenance management software provides visibility to the assets, data, and resources that affect maintenance,
CENTRIS CONSULTING Quality Control Manual ISO 9001:2008 Introduction Centris Consulting developed and implemented a Quality Management System in order to document the company s best business practices,
Analytical Method Validation A. Es-haghi Ph.D. Dept. of Physico chemistry vaccine and serum research institute email@example.com http://www.rvsri.ir/ Introduction Test procedures for assessment of the
Cold Chain Shipping: Protecting Temperature Sensitive Products Diane McLean Pfizer Global Manufacturing January 8, 2009 1 What is Cold Chain? The term cold chain or cold chain management refers to controlled
United States Department of Agriculture Marketing and Regulatory Programs Agricultural Marketing Service Livestock and Seed Program Meat Grading & Certification Branch MGC Instruction 800 May 9, 2005 Page
Company Name Street Address City, State, Zip code Phone Number Fax Company Website Email Address ORGANIZATION NAME PHONE NUMBER EMAIL ADDRESS President/CEO General Manager Engineering Manager Production
g GE Power & Water ISO 9001:2008 Audit Checklist Organization Auditor Date Page 1 Std. 4.1 General s a. Are processes identified b. Sequence & interaction of processes determined? c. Criteria for operation
OMCL Network of the Council of Europe QUALITY ASSURANCE DOCUMENT PA/PH/OMCL (08) 69 3R Full document title and reference Document type VALIDATION OF COMPUTERISED SYSTEMS Legislative basis - CORE DOCUMENT
Regulatory Asset Management: Harmonizing Calibration, Maintenance & Validation Systems 800.982.2388 1 Introduction Calibration, maintenance and validation activity, despite operating within the same department
Analytical Method Validation for Assay, Related substances & Dissolution. N A Y A N K S H A T R I 1 8 / 0 1 / 2 0 1 3 Definitions The objective of validation of an analytical procedure is to demonstrate
QCS-001 PROCESS SOURCE QUALITY REQUIREMENTS Notice: A hard copy of this document may not be the document currently in effect. The current version is always the version on the Lockheed Martin network at:
1. Introduction QUALITY ASSURANCE MANUAL This manual contains a description of the Quality Assurance Program of Phoenix Chemical Laboratory, Inc., as required by the relevant sections of 10CFR50 App. B,
Designation: E 1212 99 AMERICAN SOCIETY FOR TESTING AND MATERIALS 100 Barr Harbor Dr., West Conshohocken, PA 19428 Reprinted from the Annual Book of ASTM Standards. Copyright ASTM Standard Practice for