# Measurement of Inductor Q with the MSA Sam Wetterlin 3/31/11. Equation 1 Determining Resonant Q from Inductor Q and Capacitor Q

Save this PDF as:

Size: px
Start display at page:

Download "Measurement of Inductor Q with the MSA Sam Wetterlin 3/31/11. Equation 1 Determining Resonant Q from Inductor Q and Capacitor Q"

## Transcription

1 Measurement of Inductor with the MSA Sam Wetterlin 3/31/11 The of an inductor, which is its reactance divided by its internal series resistance, is used as an indication of how well it will perform at or near the resonant frequency when it is combined with a capacitor. This is relevant to LC filters and impedance matching circuits. Typically, we assume that the of the resonant circuit will equal that of the inductor, since capacitors typically have far higher than inductors and don t significantly degrade the circuit performance. However, to be precise, the res of the resonant circuit will be as follows: res = Ind Ind * Equation 1 Determining Resonant from Inductor and Capacitor Ind and Cap are the values for the inductor and capacitor. This document describes several methods that can be used to measure inductor using the MSA. Direct Measurement in Reflection Mode The simplest way to measure inductor is to directly measure the S 11 of the inductor in Reflection mode, and graph the equivalent series resistance and reactance. is then the reactance divided by the resistance. The MSA will construct those graphs, and will also do the math and construct the graph. This approach will work fine within a limited frequency range (well below the self-resonant frequency), with inductors whose value is not extremely high. It is difficult to get an accurate measurement of resistance when the resistance is completely swamped by the reactance; for high- inductors, the reactance is by definition far higher than the resistance. Figure 1 shows such a measurement for a hand-wound coil with #14 wire on a ceramic form. The series inductance and the inductor are graphed: + Cap Cap

2 Figure 1 Inductance and of a Hand-Wound Inductor. In Figure 1, is labeled Component. It is probably accurate up to values of 40, and reasonably accurate where it measures up to 85, but above that the determination of the resistive component becomes more questionable, and the higher values become erratic. This sort of measurement is adequate for modestly-high values, but not for pinning down very high values. Still, even though we cannot be confident that the graph accurately shows at 80 MHz, we can conclude that is higher than 100, which in many cases is all we need to know. The graph of Figure 1 is consistent with an inductor whose losses do not increase significantly with increasing frequency, so the increase in reactance causes a corresponding increase in (at least to 40 MHz it is hard to say exactly what is happening above that). Note that the measured L value is increasing near the right side of the graph. This is likely due to the effect of the parallel parasitic capacitance of the inductor, which creates a self-resonant frequency (SRF) and causes the reactance to increase as we approach the SRF. The measurement of L can be affected even at frequencies as low as SRF/10. As discussed at the end of this document, accurate measurement when the apparent L value is rapidly rising requires use of RLC Analysis with a capacitor in parallel with the inductor. Transmission Mode RLC Analysis We could get a better measure of the series resistance if it were not swamped by the series reactance. One way to get rid of the reactance is to add a series capacitor that will cause the circuit to resonate at or near the frequency of interest. (We want to use a capacitor with very high, such as the Cornell-Dubilier MC series of mica capacitors.) If we knew the L value, we could then measure the series resistance, calculate the inductor reactance, and divide reactance by resistance to get. But it is more useful to also measure the L value itself. We can t simply use the above method to measure L and R at resonance, because the reactance at resonance will be zero, giving no information about the size of the inductance. We can measure the value of all the components in Transmission mode by scanning the area around resonance (after first calibrating in the normal manner with the Through) and then using Functions RLC Analysis. This will calculate the R, L, C and values for us. 2

3 Figure 2 shows a Transmission mode scan of our inductor with a series capacitor added to create resonance near 80 MHz (the precise capacitor value was not known, but ends up being determined by RLC Analysis): Figure 2 S 21 of Inductor+15 pf Series Capacitor in Shunt Fixture Figure 2 was created by putting the series inductor+capacitor in a shunt fixture, which is similar to connecting the Tx and Rx lines of the VNA to one end and grounding the other. However, in the series fixture, attenuators are included in both the Tx and Rx lines to be sure the device being tested sees 50 ohms looking toward either Tx or Rx. The markers in Figure 2 were actually placed automatically by the RLC Analysis we are about to describe. After creating the S 21 scan, we invoke Functions RLC Analysis. In the MSA version that created Figure 2, the following dialog was shown: Figure 3 Transmission Mode RLC Analysis Dialog (The current dialog is somewhat different.) The dialog allows us to specify whether the circuit is to be analyzed as a parallel or series RLC circuit and whether it is in a series or shunt fixture. We can also specify the R 0 of the fixture, which is the resistance the RLC device sees looking toward the Tx or Rx ports (through the attenuators.) Finally, for the shunt fixture we can specify the one-way connector delay caused by 3

4 the connector that attaches our device to the fixture ns is typical for a fixture that has a female SMA connector, with the RLC device attached to the back of a mating connector. When we click Analyze, the results of the analysis are shown in the text box. The first step of the analysis is for the software to locate the resonant frequency and -3 db points, which are marked in Figure 2. Note that the -3 db points for a notch response are at the absolute -3 db level, not 3 db above the bottom of the notch. The current software lets you choose which 3 db points to use, and applies the appropriate formula. It is actually better to use the points that are 3 db above the bottom of the notch, as long as the notch is at least 20 db deep. One reason is that it allows the scan to concentrate on a smaller area around the notch, so relevant points can be more precisely located. The results show R=0.26, L=233 nh, C=15.8 pf and U =463 at 83 MHz. This lends credibility to the sky-high values shown near 83 MHz in Figure 1. Note that the U produced by RLC Analysis is that of the inductor/capacitor combination. If the capacitor has an extremely high, the U of the combination will pretty much equal the inductor. But with a U of 463, it is very difficult to have a capacitor whose has a negligible effect on the circuit. If we have an idea of the capacitor, we can make the following adjustment to get inductor Ind. U Cap Ind = Equation 2 Determining Inductor from U and Capacitor Applying this adjustment will result in the inductor being even higher than the measured U of the LC combination. Reflection Mode RLC Analysis We can also perform RLC Analysis in Reflection mode, and there are some advantages to doing so. The OSL calibration in Reflection mode removes many possible sources of error, such as the connector delay and the non-50-ohm source and load impedances of the fixture. We use the same shunt fixture used above, perform OSL calibration, attach the LC circuit to the fixture, and do a scan. We then open Functions RLC Analysis, which opens the dialog shown in Figure 4. Cap U 4

5 Figure 4 RLC dialog in Reflection Mode (Current dialog is somewhat different.) The dialog shown in Figure 4 is an old version. The current version has two changes: 1. The lossy inductor option has been deleted. It was complicated. 2. The option is provided to analyze the circuit at two frequencies pre-marked by the user, as opposed to the method shown in the above dialog, which is referred to as the Slope method. Here we will look at the results produced from use of the Slope method, which is still available in the current version. The results at the resonant frequency are shown in the text box of Figure 4. They differ somewhat from those shown in Figure 3, and may be more accurate, due to the use of OSL calibration. The major difference between this analysis and that of Transmission mode is that values are produced for all frequencies in the scan, not just for the resonant frequency. Unfortunately, the graph of corresponding to Figure 4 no longer exists. However, a similar analysis was done with a different inductor/capacitor combination, this time in parallel (using a series fixture). Figure 5 shows the results. 5

6 Figure 5 Graph of Resistance and over Frequency Different inductor and capacitor than used above. Title lines 2 and 3 show results from RLC Analysis text box. The blue graph in Figure 5 shows the graph of that is produced by Reflection mode RLC Analysis. Marker L was near the resonant frequency. The graph is labeled Parallel simply to distinguish it from the Component graph produced by the simple analysis method first described above. High values not close to resonance are somewhat erratic, but the point is still made that the inductor in question in Figure 5 would produce higher resonance U at lower frequencies. (This is due to the inductor being wound on T-68-6 toroid whose losses start to become large above 50 MHz.) The inductor described in most of this document would have the opposite behavior, with higher at higher frequencies, as was shown in Figure 1. Series vs. Parallel RLC We initially added a series capacitor to our inductor to create resonance, and analyzed it using a shunt fixture, which is optimal for measuring the small impedances that occur near series resonance. The example in Figure 5, however, used a capacitor in parallel with the inductor, and used a series fixture, which is optimal for the large impedances that occur near parallel resonance. The shunt fixture is often more convenient, because it can be built with an SMA connector and the inductor/capacitor combination can be mounted on the back of a mating connector; the components used for calibration can be similarly attached and easily removed. The series fixture requires the components being tested to be directly soldered to the fixture, and the calibration Short and Load must be similarly soldered and unsoldered. The convenience of the shunt fixture suggests that a series RLC arrangement is preferable. However, that arrangement has a significant disadvantage if we are testing the inductor at frequencies higher than 10% of its SRF. The parallel parasitic capacitance of the inductor causes the apparent value of inductance to rise (because it causes the reactance to rise) as we get anywhere close to the SRF. This distorts the measurement of both L and. RLC Analysis of the series combination will separately identify the series L and C components, but the identified L is not the true value of the inductor; it is the value of the inductor distorted by the parasitic capacitance. Only the series capacitor is separated out. 6

7 If we use a parallel arrangement of the inductor and capacitor, the parasitic capacitance of the inductor is effectively merged with the parallel capacitor. The L and C values produced by RLC Analysis will then accurately reflect the true underlying inductance. The parasitic capacitance will now be distorting the C value rather than the L value, in that the measured C will not be the true value of the attached capacitor, but will be the sum of the capacitor s value and the parasitic capacitance of the inductor. For present purposes, we don t care about determining the actual value of the capacitor. Therefore, we obtain the best accuracy in the measurement of L, and therefore, by placing the capacitor in parallel with the inductor and using a series fixture. This increased accuracy is significant only if we are testing at frequencies above 10% of the inductor s self-resonant frequency. We can get an approximate idea of the SRF by doing an uncalibrated Transmission mode scan of the inductor with either a series fixture (resonance will produce a notch) or a shunt fixture (resonance will produce a peak). Alternatively, we can measure the inductance by the simple method used to create Figure 1 and see whether the inductance is rising significantly at the frequency of interest. If so, then parallel RLC Analysis is the way to go. 7

### A Shunt Fixture for Low Impedance Measurements Sam Wetterlin 4/28/11

A Shunt Fixture for Low Impedance Measurements Sam Wetterlin 4/28/11 Introduction Measurements of impedances in the 0.01-10 ohm range are useful for measuring the resistance of capacitors, inductors and

### Impedance Matching of Filters with the MSA Sam Wetterlin 2/11/11

Impedance Matching of Filters with the MSA Sam Wetterlin 2/11/11 Introduction The purpose of this document is to illustrate the process for impedance matching of filters using the MSA software. For example,

### Q Measurement with the AIM4170 revised

Q Measurement with the AIM4170 revised 7-22-08 Summary: The AIM4170 can be used to determine the Q of coils and of tuned circuits. A function is included in the program to measure the Q of series and parallel

### Introduction to the Smith Chart for the MSA Sam Wetterlin 10/12/09 Z +

Introduction to the Smith Chart for the MSA Sam Wetterlin 10/12/09 Quick Review of Reflection Coefficient The Smith chart is a method of graphing reflection coefficients and impedance, and is often useful

### VNA Operating Guide. for the MSA 2/1/10

VNA Operating Guide for the MSA 2/1/10 The MSA as VNA Introduction This Guide contains individual pages to describe the use of the MSA in Transmission Mode [functioning as either a Scalar Network Analyzer

### TESTS OF 1 MHZ SIGNAL SOURCE FOR SPECTRUM ANALYZER CALIBRATION 7/8/08 Sam Wetterlin

TESTS OF 1 MHZ SIGNAL SOURCE FOR SPECTRUM ANALYZER CALIBRATION 7/8/08 Sam Wetterlin (Updated 7/19/08 to delete sine wave output) I constructed the 1 MHz square wave generator shown in the Appendix. This

### Yesterday s discrete. Ordinary Vector Network Analyzers Get Differential Port Measurement Capability DIFFERENTIAL MEASUREMENTS

From November 2003 High Frequency Electronics Copyright 2003 Summit Technical Media, LLC Ordinary Vector Network Analyzers Get Differential Port Measurement Capability By Dale D. Henkes Applied Computational

### RF Filter Design Using Coupled Co-axial Resonators

RF Filter Design Using Coupled Co-axial Resonators Liam Devlin, Plextek Ltd (lmd@plextek.co.uk) Plextek Ltd, London Road, Great Chesterford, Essex, CB10 1NY, Tel. 01799 533 261 Abstract This paper describes

### RF POWER AMPLIFIERS - TANK CIRCUITS & OUTPUT COUPLING

1 of 6 RF POWER AMPLIFIERS - TANK CIRCUITS & OUTPUT COUPLING by Lloyd Butler VK5BR The output tuning and coupling of the final RF amplifier is an important part of the transmitter. It is designed to load

### Nano Stepping Notch Filter Jim Hagerman 02 01 12

Nano Stepping Notch Filter Jim Hagerman 02 01 12 INTRODUCTION I worked out equations for the von Newman style high power notch filter. This design is tunable over a fairly wide range, but less than an

### MEASUREMENT SET-UP FOR TRAPS

Completed on 26th of June, 2012 MEASUREMENT SET-UP FOR TRAPS AUTHOR: IW2FND Attolini Lucio Via XXV Aprile, 52/B 26037 San Giovanni in Croce (CR) - Italy iw2fnd@gmail.com Trappole_01_EN 1 1 DESCRIPTION...3

### Circuits with inductors and alternating currents. Chapter 20 #45, 46, 47, 49

Circuits with inductors and alternating currents Chapter 20 #45, 46, 47, 49 RL circuits Ch. 20 (last section) Symbol for inductor looks like a spring. An inductor is a circuit element that has a large

### Chapter 21 Band-Pass Filters and Resonance

Chapter 21 Band-Pass Filters and Resonance In Chapter 20, we discussed low-pass and high-pass filters. The simplest such filters use RC components resistors and capacitors. It is also possible to use resistors

### Determination of Dielectric Constant Of Printed Circuit Boards Sam Wetterlin 1/24/10

Determination of Dielectric Constant Of Printed Circuit Boards Sam Wetterlin 1/24/10 Introduction The dielectric constant of PCB material is important in determining the trace width required to produce

### THREE-BEAD-BALUN REFLECTION BRIDGE 250 KHz to 1.5 GHz Sam Wetterlin 9/22/08

THREE-BEAD-BALUN REFLECTION BRIDGE 250 KHz to 1.5 GHz Sam Wetterlin swetterlin@comcast.net 9/22/08 This paper presents a reflection bridge useful from 250 KHz to the neighborhood of 1.5 GHz. This bridge

### Inductors in AC Circuits

Inductors in AC Circuits Name Section Resistors, inductors, and capacitors all have the effect of modifying the size of the current in an AC circuit and the time at which the current reaches its maximum

### RLC Series Resonance

RLC Series Resonance 11EM Object: The purpose of this laboratory activity is to study resonance in a resistor-inductor-capacitor (RLC) circuit by examining the current through the circuit as a function

### Capacitor Self-Resonance

Capacitor Self-Resonance By: Dr. Mike Blewett University of Surrey United Kingdom Objective This Experiment will demonstrate some of the limitations of capacitors when used in Radio Frequency circuits.

### MANUAL RETURN LOSS MEASUREMENTS Sam Wetterlin 10/5/08

MANUAL RETURN LOSS MEASUREMENTS Sam Wetterlin swetterlin@comcast.net 10/5/08 This paper describes the procedure for using a reflection bridge (a/k/a return loss bridge ) to manually measure return loss

### Impedance Matching and Matching Networks. Valentin Todorow, December, 2009

Impedance Matching and Matching Networks Valentin Todorow, December, 2009 RF for Plasma Processing - Definition of RF What is RF? The IEEE Standard Dictionary of Electrical and Electronics Terms defines

### GENESYS S/FILTER. Eagleware Corporation. Copyright

GENESYS S/FILTER Copyright 1986-2000 Eagleware Corporation 635 Pinnacle Court Norcross, GA 30071 USA Phone: (678) 291-0995 FAX: (678) 291-0971 E-mail: eagleware@eagleware.com Internet: http://www.eagleware.com

### LOAD RESONANT MEASUREMENTS OF QUARTZ CRYSTALS

LOAD RESONANT MEASUREMENTS OF QUARTZ CRYSTALS Dwane Rose Saunders and Associates, Inc. 2520 E. Rose Garden Lane, Phoenix, Arizona 85050 ABSTRACT The measurement of load resonant frequency can be accomplished

### Chapter 12. RL Circuits. Objectives

Chapter 12 RL Circuits Objectives Describe the relationship between current and voltage in an RL circuit Determine impedance and phase angle in a series RL circuit Analyze a series RL circuit Determine

### RESISTANCE, REACTANCE AND IMPEDANCE A Primer. Douglas Brooks, PhD UltraCAD Design, Inc. PART 2, REACTANCE

RESISTANCE, REACTANCE AND IMPEDANCE A Primer Douglas Brooks, PhD UltraCAD Design, Inc. PART 2, REACTANCE This is part 2 of a 3-part series on resistance, reactance and impedance. Most of us are familiar

### LS RS. Figure 1: Assumed inductor model

Characterizing Inductors at HF and VHF Inductors are a key component in RF circuits. Their performance makes a great difference in the operation of amplifiers, oscillators, and other circuit blocks --

### S-Parameters and Related Quantities Sam Wetterlin 10/20/09

S-Parameters and Related Quantities Sam Wetterlin 10/20/09 Basic Concept of S-Parameters S-Parameters are a type of network parameter, based on the concept of scattering. The more familiar network parameters

### LCR Parallel Circuits

Module 10 AC Theory Introduction to What you'll learn in Module 10. The LCR Parallel Circuit. Module 10.1 Ideal Parallel Circuits. Recognise ideal LCR parallel circuits and describe the effects of internal

### LCR Series Circuits. AC Theory. Introduction to LCR Series Circuits. Module 9. What you'll learn in Module 9. Module 9 Introduction

Module 9 AC Theory LCR Series Circuits Introduction to LCR Series Circuits What you'll learn in Module 9. Module 9 Introduction Introduction to LCR Series Circuits. Section 9.1 LCR Series Circuits. Amazing

### Measuring Impedance and Frequency Response of Guitar Pickups

Measuring Impedance and Frequency Response of Guitar Pickups Peter D. Hiscocks Syscomp Electronic Design Limited phiscock@ee.ryerson.ca www.syscompdesign.com April 30, 2011 Introduction The CircuitGear

### LR Phono Preamps. Pete Millett ETF.13. pmillett@hotmail.com

LR Phono Preamps Pete Millett ETF.13 pmillett@hotmail.com Agenda A bit about me Part 1: What is, and why use, RIAA? Grooves on records The RIAA standard Implementations of RIAA EQ networks and preamps

### Since any real component also has loss due to the resistive component, the average power dissipated is 2 2R

Quality factor, Q Reactive components such as capacitors and inductors are often described with a figure of merit called Q. While it can be defined in many ways, it s most fundamental description is: Q

### Filter Considerations for the IBC

application note AN:202 Filter Considerations for the IBC Mike DeGaetano Application Engineering July 2013 Contents Page Introduction 1 IBC Attributes 1 Damping and 2 Converter Bandwidth Filtering 3 Filter

### Extra Questions - 1. 1. What current will flow in a 20Ω resistor when it is connected to a 50V supply? a) 0.4A b) 1.6A c) 2.5A

Extra Questions - 1 1. What current will flow in a 20Ω resistor when it is connected to a 50V supply? a) 0.4A b) 1.6A c) 2.5A 2. A current of 500mA flows in a resistance of 12Ω. What power is dissipated

### Understanding Adjustable Speed Drive Common Mode Problems and Effective Filter Solutions

Understanding Adjustable Speed Drive Common Mode Problems and Effective Filter Solutions September 22, 2014 Todd Shudarek, Principal Engineer MTE Corporation N83 W13330 Leon Road Menomonee Falls WI 53051

### EXPANDED-RANGE LOG DETECTOR Sam Wetterlin 6/22/08, updated 9/25/10

EXPANDED-RANGE LOG DETECTOR Sam Wetterlin 6/22/08, updated 9/25/10 Log detectors take an RF input and produce an output proportional to the RF signal level expressed in db. Limiting log detectors such

### Basic AC Reactive Components IMPEDANCE

Basic AC Reactive Components Whenever inductive and capacitive components are used in an AC circuit, the calculation of their effects on the flow of current is important. EO 1.9 EO 1.10 EO 1.11 EO 1.12

### Voltage Transformers. VT Types

Voltage Transformers VT - Voltage Transformer, also know as Potential Transformer (PT) CVT - Capacitive Voltage Transformer also know as CCVT (Capacitive Coupled VT) BPD - Bushing Potential Device also

### RESONANCE AND FILTERS

1422-1 RESONANCE AND FILTERS Experiment 4, Resonant Frequency and Impedance of a Parallel Circuit, For more courses visit www.cie-wc.edu OBJECTIVES: 1. To demonstrate the resonant frequency of a parallel

### Frank Lynch, W4FAL. 24 April 2008 SCARS Page 1

SCARS http://smithchart.org 1 Smith Chart History Frank Lynch, W4FAL Invented by Phillip H. Smith in 1939 Used to solve a variety of transmission line and waveguide problems Basic Uses For evaluating the

### Eðlisfræði 2, vor 2007

[ Assignment View ] [ Pri Eðlisfræði 2, vor 2007 31. Alternating Current Circuits Assignment is due at 2:00am on Wednesday, March 21, 2007 Credit for problems submitted late will decrease to 0% after the

### Constructing and Using Chokes

Constructing and Using Chokes The first installment of this series introduced basic antenna modeling and demonstrated the generation and control of feed line currents. The control of the feed line currents

### Basic Electrical Theory

Basic Electrical Theory Impedance PJM State & Member Training Dept. PJM 2014 10/24/2013 Objectives Identify the components of Impedance in AC Circuits Calculate the total Impedance in AC Circuits Identify

### Direct versus Alternating Current Things We Can Measure

Phil Sherrod W4PHS Direct versus Alternating Current Things We Can Measure Direct Current (DC) Alternating Current (AC) Voltage Voltage (peak, RMS) Current Current (peak, effective) Power True power, Apparent

### Chapt ha e pt r e r 12 RL Circuits

Chapter 12 RL Circuits Sinusoidal Response of RL Circuits The inductor voltage leads the source voltage Inductance causes a phase shift between voltage and current that depends on the relative values of

### April 8. Physics 272. Spring Prof. Philip von Doetinchem

Physics 272 April 8 Spring 2014 http://www.phys.hawaii.edu/~philipvd/pvd_14_spring_272_uhm.html Prof. Philip von Doetinchem philipvd@hawaii.edu Phys272 - Spring 14 - von Doetinchem - 218 L-C in parallel

### Agilent Advanced impedance measurement capability of the RF I-V method compared to the network analysis method. Application Note 1369-2

Agilent Advanced impedance measurement capability of the RF I-V method compared to the network analysis method Application Note 1369-2 1. Introduction Explosive demand for the electronic devices used in

### Filters & Wave Shaping

Module 8 AC Theory Filters & Wave Shaping Passive Filters & Wave Shaping What you'll learn in Module 8. Module 8 Introduction Recognise passive filters with reference to their response curves. High pass,

### Understanding Power Factor and How it Affects Your Electric Bill. Presented by Scott Peele PE

Understanding Power Factor and How it Affects Your Electric Bill Presented by Scott Peele PE Understanding Power Factor Definitions kva, kvar, kw, Apparent Power vs. True Power Calculations Measurements

### Components in Series, Parallel, and Combination

Components in Series, Parallel, and Combination Kirchoff s Laws VOLTAGE LAW: A series circuit of voltages across the various components must add up to be equal to the voltage applied to the circuit. CURRENT

### VJ 6040 Mobile Digital TV UHF Antenna Evaluation Board

VISHAY VITRAMON Multilayer Chip Capacitors Application Note GENERAL is a multilayer ceramic chip antenna designed for receiving mobile digital TV transmissions in the UHF band. The target application for

### Henry Lin, Department of Electrical and Computer Engineering, California State University, Bakersfield Lecture 3 (Electric Circuits) July 16 th, 2013

Henry Lin, Department of Electrical and Computer Engineering, California State University, Bakersfield Lecture 3 (Electric Circuits) July 16 th, 2013 1 What is an electrical circuit? An electrical network

### When designing. Inductors at UHF: EM Simulation Guides Vector Network Analyzer. measurement. EM SIMULATION. There are times when it is

Inductors at UHF: EM Simulation Guides Vector Network Analyzer Measurements John B. Call Thales Communications Inc., USA When designing There are times when it is circuits for necessary to measure a operation

### RADIO AMATEUR EXAM GENERAL CLASS

RAE-Lessons by 4S7VJ 1 RADIO AMATEUR EXAM GENERAL CLASS By 4S7VJ CHAPTER- 3 3.1 REACTANCE (X) When a.c. voltage applied to a capacitor or an inductor the current (r.m.s.) is proportional to the voltage

### Chapter 15 10/14/2014

Chapter 15 Analyze series and parallel ac circuits to find Voltage Current Power Total impedance, admittance Apply known circuit theories Kirchhoff s current, voltage laws Voltage or current divider rule

### AN-837 APPLICATION NOTE

APPLICATION NOTE One Technology Way P.O. Box 916 Norwood, MA 262-916, U.S.A. Tel: 781.329.47 Fax: 781.461.3113 www.analog.com DDS-Based Clock Jitter Performance vs. DAC Reconstruction Filter Performance

### RESONANCE AND FILTERS

1422-1 RESONANCE AND FILTERS Experiment 1, Resonant Frequency and Circuit Impedance For more courses visit www.cie-wc.edu OBJECTIVES 1. To verify experimentally our theoretical predictions concerning the

### Experiment #11: LRC Circuit (Power Amplifier, Voltage Sensor)

Experiment #11: LRC Circuit (Power Amplifier, Voltage Sensor) Concept: circuits Time: 30 m SW Interface: 750 Windows file: RLC.SWS EQUIPMENT NEEDED Science Workshop Interface Power Amplifier (2) Voltage

### LRC Circuits. Purpose. Principles PHYS 2211L LAB 7

Purpose This experiment is an introduction to alternating current (AC) circuits. Using the oscilloscope, we will examine the voltage response of inductors, resistors and capacitors in series circuits driven

### IC-EMC v2 Application Note. A model of the Bulk Current Injection Probe

IC-EMC v2 Application Note A model of the Bulk Current Injection Probe This work has been conducted by S. Akue Boulingui and A. Cisse Ndoye within the French project EPEA-Aerospace Valley funded by the

### Chapter 3. Simulation of Non-Ideal Components in LTSpice

Chapter 3 Simulation of Non-Ideal Components in LTSpice 27 CHAPTER 3. SIMULATION OF NON-IDEAL COMPONENTS IN LTSPICE 3.1 Pre-Lab The answers to the following questions are due at the beginning of the lab.

### Practice Problems - Chapter 33 Alternating Current Circuits

Multiple Choice Practice Problems - Chapter 33 Alternating Current Circuits 4. A high-voltage powerline operates at 500 000 V-rms and carries an rms current of 500 A. If the resistance of the cable is

### ATE-A1 Testing Without Relays - Using Inductors to Compensate for Parasitic Capacitance

Introduction (Why Get Rid of Relays?) Due to their size, cost and relatively slow (millisecond) operating speeds, minimizing the number of mechanical relays is a significant goal of any ATE design. This

### ALTERNATING CURRENTS

ALTERNATING CURRENTS VERY SHORT ANSWER QUESTIONS Q-1. What is the SI unit of? Q-2. What is the average value of alternating emf over one cycle? Q-3. Does capacitor allow ac to pass through it? Q-4. What

### Alternating Current. Asist. Prof. Dr. Aytaç Gören Asist. Prof. Dr. Levent Çetin

Asist. Prof. Dr. Aytaç Gören Asist. Prof. Dr. Levent Çetin 30.10.2012 Contents Alternating Voltage Phase Phasor Representation of AC Behaviors of Basic Circuit Components under AC Resistance, Reactance

### Fall 12 PHY 122 Homework Solutions #10

Fall 12 PHY 122 Homework Solutions #10 HW10: Ch.30 Q5, 8, 15,17, 19 P 1, 3, 9, 18, 34, 36, 42, 51, 66 Chapter 30 Question 5 If you are given a fixed length of wire, how would you shape it to obtain the

### Application Note. So You Need to Measure Some Inductors?

So You Need to Measure Some nductors? Take a look at the 1910 nductance Analyzer. Although specifically designed for production testing of inductors and coils, in addition to measuring inductance (L),

### Understanding Power Impedance Supply for Optimum Decoupling

Introduction Noise in power supplies is not only caused by the power supply itself, but also the load s interaction with the power supply (i.e. dynamic loads, switching, etc.). To lower load induced noise,

### EMI Conducted Interference

Although the concepts stated are universal, this application note was written specifically for Interpoint products. Switching power converters are natural generators of input narrowband spectral noise

### The Smith chart. Smith chart components. The normalized impedance line CHAPTER

26 CHAPTER The Smith chart The mathematics of transmission lines, and certain other devices, becomes cumbersome at times, especially when dealing with complex impedances and nonstandard situations. In

### Parallel Inductor-Resistor-Capacitor (RLC) Circuits

Parallel Inductor-Resistor-Capacitor (RLC) Circuits Session 4b for Basic Electricity A Fairfield University E-Course Powered by LearnLinc 7/9/2002 Basic Electricity 1 Module: Basic Electronics (AC Circuits

### S-PARAMETER MEASUREMENTS OF MEMS SWITCHES

Radant MEMS employs adaptations of the JMicroTechnology test fixture depicted in Figure 1 to measure MEMS switch s-parameters. RF probeable JMicroTechnology microstrip-to-coplanar waveguide adapter substrates

### FREQUENCY RESPONSE AND PASSIVE FILTERS LABORATORY. We start with examples of a few filter circuits to illustrate the concept.

FREQUENCY RESPONSE AND PASSIVE FILTERS LABORATORY In this experiment we will analytically determine and measure the frequency response of networks containing resistors, AC source/sources, and energy storage

### Common Mode Filter Inductor Analysis

Document 2-1 Common Mode Filter Inductor Analysis Abstract Noise limits set by regulatory agencies make solutions to common mode EMI a necessary consideration in the manufacture and use of electronic equipment.

### Teacher s Guide Physics Labs with Computers, Vol C P52: LRC Circuit. Teacher s Guide - Activity P52: LRC Circuit (Voltage Sensor)

Teacher s Guide Physics Labs with Computers, Vol. 2 012-06101C P52: LRC Circuit Teacher s Guide - Activity P52: LRC Circuit (Voltage Sensor) Concept DataStudio ScienceWorkshop (Mac) ScienceWorkshop (Win)

### Agilent Ultra-Low Impedance Measurements Using 2-Port Measurements. Application Note

Agilent Ultra-Low Impedance Measurements Using 2-Port Measurements Application Note Table of Contents Ultra-Low Impedance Measurements Using a Vector Network Analyzer... 3 Limitations of 1-Port VNA Impedance

### Application Note AN:005. FPA Printed Circuit Board Layout Guidelines. Introduction Contents. The Importance of Board Layout

FPA Printed Circuit Board Layout Guidelines By Paul Yeaman Principal Product Line Engineer V I Chip Strategic Accounts Introduction Contents Page Introduction 1 The Importance of 1 Board Layout Low DC

### Keywords: input noise, output noise, step down converters, buck converters, MAX1653EVKit

Maxim > Design Support > Technical Documents > Tutorials > Power-Supply Circuits > APP 986 Keywords: input noise, output noise, step down converters, buck converters, MAX1653EVKit TUTORIAL 986 Input and

### EE133 Winter 2002 Cookbook Filter Guide Welcome to the Cookbook Filter Guide!

Welcome to the! Don t have enough time to spice out that perfect filter before Aunt Thelma comes down for dinner? Well this handout is for you! The following pages detail a fast set of steps towards the

### ε: Voltage output of Signal Generator (also called the Source voltage or Applied

Experiment #10: LR & RC Circuits Frequency Response EQUIPMENT NEEDED Science Workshop Interface Power Amplifier (2) Voltage Sensor graph paper (optional) (3) Patch Cords Decade resistor, capacitor, and

### Electronic Components. Electronics. Resistors and Basic Circuit Laws. Basic Circuits. Basic Circuit. Voltage Dividers

Electronics most instruments work on either analog or digital signals we will discuss circuit basics parallel and series circuits voltage dividers filters high-pass, low-pass, band-pass filters the main

### Line Reactors and AC Drives

Line Reactors and AC Drives Rockwell Automation Mequon Wisconsin Quite often, line and load reactors are installed on AC drives without a solid understanding of why or what the positive and negative consequences

### RESONANCE AND FILTERS

1422-1 RESONANCE AND FILTERS Experiment 5, Current in a Parallel Resonant Circuit, For more courses visit www.cie-wc.edu OBJECTIVES: 1. To verify by experiment, that the line current (I line ) is at its

### Yaesu FT-847 70 MHz PA output and filter simulations Marc Vlemmings, PA1O - Eindhoven, The Netherlands

Yaesu FT-847 70 MHz PA output and filter simulations Marc Vlemmings, PA1O - Eindhoven, The Netherlands The efficiency of the HF transmitter chain which is also used for 50 MHz and 70 MHz decreases with

### EMI Filter Hints and Tips

EMI Filter Hints and Tips 2010 Stephen Hopwood - Snr Engineer Syfer Technology are a Dover CMP Company based in Norwich England. World leading manufacturers of ceramic passive components, their filter

### CHAPTER 16 OSCILLATORS

CHAPTER 16 OSCILLATORS 16-1 THE OSCILLATOR - are electronic circuits that generate an output signal without the necessity of an input signal. - It produces a periodic waveform on its output with only the

### EDEXCEL NATIONAL CERTIFICATE/DIPLOMA UNIT 5 - ELECTRICAL AND ELECTRONIC PRINCIPLES NQF LEVEL 3 OUTCOME 4 - ALTERNATING CURRENT

EDEXCEL NATIONAL CERTIFICATE/DIPLOMA UNIT 5 - ELECTRICAL AND ELECTRONIC PRINCIPLES NQF LEVEL 3 OUTCOME 4 - ALTERNATING CURRENT 4 Understand single-phase alternating current (ac) theory Single phase AC

### Electronic filters design tutorial - 3

High pass, low pass and notch passive filters In the first and second part of this tutorial we visited the band pass filters, with lumped and distributed elements. In this third part we will discuss about

### APN1001: Circuit Models for Plastic Packaged Microwave Diodes

APPLICATION NOTE APN11: Circuit Models for Plastic Packaged Microwave Diodes Abstract This paper reports on the measurement and establishment of circuit models for SOT-23 and SOD-323 packaged diodes. Results

### LCR Meter Measurement Accuracy

LCR Meter Measurement Accuracy Introduction LCR meters make taking L, C and R measurements very easy. One can simply hook up the test component between the clip terminals and get the reading on the display

### 2. A conductor of length 2m moves at 4m/s at 30 to a uniform magnetic field of 0.1T. Which one of the following gives the e.m.f. generated?

Extra Questions - 2 1. A straight length of wire moves through a uniform magnetic field. The e.m.f. produced across the ends of the wire will be maximum if it moves: a) along the lines of magnetic flux

### Filters and Waveform Shaping

Physics 333 Experiment #3 Fall 211 Filters and Waveform Shaping Purpose The aim of this experiment is to study the frequency filtering properties of passive (R, C, and L) circuits for sine waves, and the

### VCO Design Project ECE145B

VCO Design Project ECE145B VCO DESIGN. Design, build, and test a voltage-controlled oscillator (VCO). The intent is that the VCO can be used as part of a VHF radio. Because the radio will be battery powered,

### Quality Factor, Bandwidth, and Harmonic Attenuation of Pi Networks

Bill Kaune, W7IE 6 Cedarview Dr, Port Townsend, WA 98363: wtkaune@ieee.org uality Factor, Bandwidth, and Harmonic Attenuation of Pi Networks The author looks at several definitions of, and makes some interesting

### Maintaining Signal Integrity when Selecting ESD Protection for USB 3.0 Interface

Maintaining Signal Integrity when Selecting ESD Protection for USB 3.0 Interface Introduction As today s lifestyle becomes more content rich through the use of streaming and sharing high quality video,

### Diodes have an arrow showing the direction of the flow.

The Big Idea Modern circuitry depends on much more than just resistors and capacitors. The circuits in your computer, cell phone, Ipod depend on circuit elements called diodes, inductors, transistors,

### The Evolution of an EQ Design By Fred Forssell, Forssell Technologies Inc. 1 st draft

The Evolution of an EQ Design By Fred Forssell, Forssell Technologies Inc. st draft This discussion covers the steps used to design a functional multi-band equalizer for use in professional audio applications

### TRANSISTOR AMPLIFIERS AET 8. First Transistor developed at Bell Labs on December 16, 1947

AET 8 First Transistor developed at Bell Labs on December 16, 1947 Objective 1a Identify Bipolar Transistor Amplifier Operating Principles Overview (1) Dynamic Operation (2) Configurations (3) Common Emitter