# measurement, but almost any pipe elbow can be calibrated Elbow meters are not as potentially accurate as venturi,

Save this PDF as:

Size: px
Start display at page:

Download "measurement, but almost any pipe elbow can be calibrated Elbow meters are not as potentially accurate as venturi,"

## Transcription

1 Lecture 14 Flow Measurement in Pipes I. Elbow Meters An elbow in a pipe can be used as a flow measuring device much in the same way as a venturi or orifice plate The head differential across the elbow (from inside to outside) is measured, and according to a calibration the discharge can be estimated The taps are usually located in the center of the elbow (e.g. at a 45 angle for a 90 elbow), but can be at other locations toward the upstream side of the elbow Some companies manufacture elbow meters for flow measurement, but almost any pipe elbow can be calibrated Elbow meters are not as potentially accurate as venturi, nozzle, and orifice meters Typical accuracy is about ±4% of Q One advantage of elbow meters is that there need not be any additional head loss in the piping system as a result of flow measurement The graph below is a sample calibration curve for a 1½-inch elbow meter in a USU hydraulic lab where the head differential (inside to outside tap) is measured in inches of mercury, using a manometer (data are from Dr. L.S. Willardson) 4.0 Calibration for a 1.5-inch Elbow Meter 3.5 Head differential (inches of Hg) Discharge (cfs) BIE 5300/6300 Lectures 153 Gary P. Merkley

2 II. Variable Area Meters These are vertical cylinders with a uniformly expanding cross-section in the upward direction A float inside the cylinder stabilizes at a certain elevation depending on the flow rate through the cylinder Note that the outside walls are usually transparent to allow direct readings by eye III. Horizontal Trajectory Method From physics, an accelerating object will travel a distance x in time t according to the following equation (based on Newton s nd law): x at = vot+ (1) where x is the distance; v o is the initial velocity at time 0; t is the elapsed time; and a is the acceleration Flow emanating from a horizontal pipe will fall a height y over a distance x The horizontal component (x-direction) has almost no acceleration, and the vertical component (y-direction) has an initial velocity of zero The vertical acceleration is equal to the ratio of weight to mass, or g = 9.81 m/s (3. ft/s ) Therefore, gt x = vot and, y = () Gary P. Merkley 154 BIE 5300/6300 Lectures

3 Then by getting rid of t, knowing that Q = VA, and the equation for the area of a circle, the flow rate is calculated as follows: Q = πd x (3) y 4 g where D is the inside diameter of the circular pipe This equation is approximately correct if x and y are measured to the center of mass of the discharge trajectory Errors occur because in practice it is difficult to measure exactly to the center, and because of possible wind and other turbulent effects Also, the pipe might not be exactly horizontal (although a correction could take this into account, according to the same analysis given above) Tables of coefficient values derived from experiments allow x and y to be measured from the top of the trajectory However, measurements can be difficult because the flow is often very turbulent at a distance from the pipe end The previous equation can be simplified as: x Q = 3.151CD (4) y where C is a coefficient to adjust the calculated discharge value when the ratios of x/d or y/d are smaller than 8 and 5, respectively (otherwise, C equals unity) Equation 4 is valid for x, y and D in ft, and Q in cfs The following table is for C values for use with Eq. 4 BIE 5300/6300 Lectures 155 Gary P. Merkley

4 y/d x/d This method can also be used for pipes flowing partially full (i.e. A < πd /4), and experimental data are available to assist in the estimation of discharge for these conditions IV. California Pipe Method This is the horizontal pipe method for partially-full pipes It is somewhat analogous to the calibration for a weir or free overfall The following equation is in English units: 1 a Q = D (5) D where a and D are defined in the figure below (ft); and Q is discharge in cfs The ratio a/d is limited to: a/d > 0.45 This method was published in the 190 s Measurement accuracy is only ±10%, at best The pipe must be exactly horizontal (level), with circular cross section The pipe must discharge freely into the air, unsubmerged Gary P. Merkley 156 BIE 5300/6300 Lectures

5 V. Vertical Trajectory Method As with pipes discharging horizontally into the air, there is a method to measure the flow rate from vertical pipes This is accomplished by assuming a translation of velocity head into the measurable height of a column of water above the top of the pipe Thus, to estimate the flow rate from pipes discharging vertically into the air it is only necessary to measure the: 1. inside diameter of the pipe, D; and,. the height of the jet, H, above the pipe This is a nice idea on paper, but in practice, it can be difficult to measure the height of the column of water because of sloshing, surging, and splashing Also, the act of measuring the height of the column can significantly alter the measured value The table below gives flow rate values in gpm for several pipe diameters in inches Jet Pipe Diameter (inch) Height (inch) (gpm) (gpm) (gpm) (gpm) (gpm) (gpm) (gpm) (gpm) ½ ½ ½ From Utah Engineering Experiment Station Bulletin 5, June Jet Height (first column) is the height from the top of the pipe to the top of the jet. BIE 5300/6300 Lectures 157 Gary P. Merkley

6 VI. Vortex Shedding Meters The vortex shedding meter can be accurate to within ±½% to ±1% of the true discharge The basic principal is that an object placed in the flow will cause turbulence and vortices in the downstream direction, and the rate of fluctuation of the vortices can be measured by detecting pressure variations just downstream This rate increases with increasing velocity, and it can be used to give an estimate of the discharge This requires calibration for a particular pipe material, pipe size, element shape and size, fluid type, and temperature It is essentially a velocity-area flow measurement method, but it is calibrated to give discharge directly Gary P. Merkley 158 BIE 5300/6300 Lectures

7 Vortex shedding meters are commercially available and are used with a variety of fluids, not only with water, and can operate well over a large pressure range and high flow velocities Size selection for these meters is important in order to avoid cavitation There need not be any moving parts in the meter This meter can be used for velocities up to approximately 50 m/s, or 180 kph The response of the device is linear for Reynolds numbers of 10,000 or more Errors can result from pipe vibration due to external machinery or other causes, or when the velocity is too low; however, some sophisticated devices have been developed and tested to correct for such errors VII. Ultrasonic Meters 1. Doppler An emitted pressure wave reflects off a deflector plate Difference between transmitted and reflected frequencies correlates to flow velocity Liquid does not have to be clean in fact, it may not work well if the liquid is too clean because it needs particles to reflect the signal. Transit-time Also called time-of-flight The liquid should be fairly clean with this method Devices generates high-frequency ( 1 MHz) pressure wave(s) Time to reach an opposing wall (inside the pipe) depends on: a) Flow velocity b) Beam orientation (angle) c) Speed of sound through the liquid medium Upstream straightening vanes may be needed to avoid swirling flow May have a single or multiple transmitted sound beams VIII. Other Measurement Devices Collins meters Commercial propeller flow meters Electromagnetic flow meters Volumetric tank BIE 5300/6300 Lectures 159 Gary P. Merkley

8 Weight tank References & Bibliography Brater, E.F. and H.W. King Handbook of hydraulics. McGraw-Hill. Daugherty, R.L. and J.B. Franzini Fluid mechanics with engineering applications. McGraw-Hill. Ginesi, D Putting new technology to work in flow measurement. Chilton s I&CS 60::5-8. Greve, F.W Measurement of pipe flow by the coordinate method. Purdue Engrg. Experiment Station Bulletin #3. Israelson, O.W. and V.E. Hansen Irrigation principles and practices. John Wiley, 3 rd Ed., pp King, L.G Drainage laboratory manual. BIE Dept., USU (BIE 605 course notes). Ledoux, J.W Venturi tube characteristics. Trans. ASCE, vol. 91. Lucas, G.P. and J.T. Turner Influence of cylinder geometry on the quality of its vortex shedding signal. Proc. Int l Conference on Flow Measurement (FLOWMECO 1985), Univ. of Melbourne, Australia, pp Miller, R.W Flow measurement engineering handbook. 3 rd Ed. McGraw-Hill Book Co., New York, N.Y. Sovik, R.E Flow measurement - some new considerations. Mech. Engrg., May, 107(5):48-5. Tily, P Practical options for on-line flow measurement. Process Engrg., London, 67(5): U.S. Bureau of Reclamation Water measurement manual. nd Ed., Denver, CO. Gary P. Merkley 160 BIE 5300/6300 Lectures

### Flow Measurement Options for Pipeline and Open Channel Flow

Flow Measurement Options for Pipeline and Open Channel Flow October 2013 Presented by Molly Skorpik - 2013 Montana Association of Dam and Canal Systems Conference Irrigation Training and Research Center

### Head Loss in Pipe Flow ME 123: Mechanical Engineering Laboratory II: Fluids

Head Loss in Pipe Flow ME 123: Mechanical Engineering Laboratory II: Fluids Dr. J. M. Meyers Dr. D. G. Fletcher Dr. Y. Dubief 1. Introduction Last lab you investigated flow loss in a pipe due to the roughness

### UNIVERSITY OF MINNESOTA DULUTH DEPARTMENT OF CHEMICAL ENGINEERING ChE MEASUREMENTS WITH FLOW METERS

UNIVERSITY OF MINNESOTA DULUTH DEPARTMENT OF CHEMICAL ENGINEERING ChE 3211-4211 MEASUREMENTS WITH FLOW METERS OBJECTIVE The purpose of this experiment is to calculate the coefficient of discharge from

### Flow Measurement in Pipes and Ducts. Flow Measurement in Pipes and Ducts, Course #503. Presented by:

Flow Measurement in Pipes and Ducts, Course #503 Presented by: PDH Enterprises, LLC PO Box 942 Morrisville, NC 27560 www.pdhsite.com This course is about measurement of the flow rate of a fluid flowing

### Outdated Publication, for historical use. CAUTION: Recommendations in this publication may be obsolete.

IRRIGATION MANAGEMENT S E R I E S Irrigation Water Measurement Danny H. Rogers Extension Agricultural Engineer Richard D. Black Extension Agricultural Engineers Cooperative Extension Service Kansas State

### Chapter 10. Flow Rate. Flow Rate. Flow Measurements. The velocity of the flow is described at any

Chapter 10 Flow Measurements Material from Theory and Design for Mechanical Measurements; Figliola, Third Edition Flow Rate Flow rate can be expressed in terms of volume flow rate (volume/time) or mass

### FLOW MEASUREMENT 2001 INTERNATIONAL CONFERENCE DERIVATION OF AN EXPANSIBILITY FACTOR FOR THE V-CONE METER

FLOW MEASUREMENT 200 INTERNATIONAL CONFERENCE DERIVATION OF AN EXPANSIBILITY FACTOR FOR THE V-CONE METER Dr D G Stewart, NEL Dr M Reader-Harris, NEL Dr R J W Peters, McCrometer Inc INTRODUCTION The V-Cone

### Laminar and Turbulent flow. Flow Sensors. Reynolds Number. Thermal flow Sensor. Flow and Flow rate. R = Mass Flow controllers

Flow and Flow rate. Laminar and Turbulent flow Laminar flow: smooth, orderly and regular Mechanical sensors have inertia, which can integrate out small variations due to turbulence Turbulent flow: chaotic

### Experiment 3 Pipe Friction

EML 316L Experiment 3 Pipe Friction Laboratory Manual Mechanical and Materials Engineering Department College of Engineering FLORIDA INTERNATIONAL UNIVERSITY Nomenclature Symbol Description Unit A cross-sectional

### CHE 253M Experiment No. 3 LIQUID FLOW MEASUREMENT

Rev 8/15 AW/GW CHE 253M Experiment No. 3 LIQUID FLOW MEASUREMENT The objective of this experiment is to familiarize the student with several types of flow-measuring devices commonly used in the laboratory

### FLUID FLOW Introduction General Description

FLUID FLOW Introduction Fluid flow is an important part of many processes, including transporting materials from one point to another, mixing of materials, and chemical reactions. In this experiment, you

### Figure 1. Head losses in a pipe

53:071 Principles of Hydraulics Laboratory Experiment #1 ENERGY AND HYDRAULIC GRADE LINES IN WATER PIPE SYSTEMS Principle The energy of a real fluid decreases as it moves through a pipe. The energy budget

### Lecture 24 Flumes & Channel Transitions. I. General Characteristics of Flumes. Flumes are often used:

Lecture 24 Flumes & Channel Transitions I. General Characteristics of Flumes Flumes are often used: 1. Along contours of steep slopes where minimal excavation is desired 2. On flat terrain where it is

### Broad Crested Weirs. I. Introduction

Lecture 9 Broad Crested Weirs I. Introduction The broad-crested weir is an open-channel flow measurement device which combines hydraulic characteristics of both weirs and flumes Sometimes the name ramp

### Effects of Beta Ratio and Reynold s Number on Coefficient of Discharge of Orifice Meter

J Agric Rural Dev 7(&2), 5-56, June 29 ISSN 8-86 K wl I cj x Dbœqb zj Available online at http://www.banglajol.info/index.php/jard JARD Journal of Agriculture & Rural Development Effects of Beta Ratio

### Grant Agreement No. 228296 SFERA. Solar Facilities for the European Research Area SEVENTH FRAMEWORK PROGRAMME. Capacities Specific Programme

Grant Agreement No. 228296 SFERA Solar Facilities for the European Research Area SEVENTH FRAMEWORK PROGRAMME Capacities Specific Programme Research Infrastructures Integrating Activity - Combination of

### Superb accuracy in water measurement, Jessica thought. Dune, F. Herbert (1965)

Lecture 2 Flumes for Open-Channel Flow Measurement Superb accuracy in water measurement, Jessica thought. Dune, F. Herbert (1965) I. Procedure for Installing a Parshall Flume to Ensure Free Flow If possible,

### Open Channel Flow Measurement Weirs and Flumes

Open Channel Flow Measurement Weirs and Flumes by Harlan H. Bengtson, PhD, P.E. 1. Introduction Your Course Title Here Measuring the flow rate of water in an open channel typically involves some type of

### Vortex Shedding Meters Class # Curtis Gulaga Business Development Manager CB Engineering Ltd. #20, Street S.E. Calgary, Alberta, Canada

Vortex Shedding Meters Class # 8150 Curtis Gulaga Business Development Manager CB Engineering Ltd. #20, 5920 11 Street S.E. Calgary, Alberta, Canada 1.0 Introduction Vortex meters have proven to be repeatable,

### Sharp-Crested Weirs for Open Channel Flow Measurement, Course #506. Presented by:

Sharp-Crested Weirs for Open Channel Flow Measurement, Course #506 Presented by: PDH Enterprises, LLC PO Box 942 Morrisville, NC 27560 www.pdhsite.com A weir is basically an obstruction in an open channel

### Minor losses include head losses through/past hydrants, couplers, valves,

Lecture 10 Minor Losses & Pressure Requirements I. Minor Losses Minor (or fitting, or local ) hydraulic losses along pipes can often be estimated as a function of the velocity head of the water within

### Practice Problems on Bernoulli s Equation. V car. Answer(s): p p p V. C. Wassgren, Purdue University Page 1 of 17 Last Updated: 2010 Sep 15

bernoulli_0 A person holds their hand out of a car window while driving through still air at a speed of V car. What is the maximum pressure on the person s hand? V car 0 max car p p p V C. Wassgren, Purdue

### Experiment # 3: Pipe Flow

ME 05 Mechanical Engineering Lab Page ME 05 Mechanical Engineering Laboratory Spring Quarter 00 Experiment # 3: Pipe Flow Objectives: a) Calibrate a pressure transducer and two different flowmeters (paddlewheel

### Michael Montgomery Marketing Product Manager Rosemount Inc. Russ Evans Manager of Engineering and Design Rosemount Inc.

ASGMT / Averaging Pitot Tube Flow Measurement Michael Montgomery Marketing Product Manager Rosemount Inc. Russ Evans Manager of Engineering and Design Rosemount Inc. Averaging Pitot Tube Meters Introduction

### There are a number of criteria to consider when designing canals Below is a list of main criteria, not necessarily in order of importance:

Lecture 15 Canal Design Basics I. Canal Design Factors There are a number of criteria to consider when designing canals Below is a list of main criteria, not necessarily in order of importance: 1. Flow

### UNIT Engineering: Measurement Technology Flow (SCQF level 6)

National Unit Specification: general information CODE H0W6 12 SUMMARY This Unit can be delivered as part of a National Qualification Group Award but can also be taken as a free-standing Unit. This Unit

### Flow Measurement Calibration and Measurement

Calibration and Measurement Flow Measurement Flow measurement for agricultural irrigation delivery can be accomplished with four general approaches. These categories are generalized below. I. Standard

### Energy and Flow Measurement for Hydronic Systems

Energy and Flow Measurement for Hydronic Systems Presented By: George Szyszko Applications Consultant MCR for ONICON Incorporated Why Measure Thermal Energy? To control something, you must first measure

### Lecture 22 Example Culvert Design Much of the following is based on the USBR technical publication Design of Small Canal Structures (1978)

Lecture 22 Example Culvert Design Much of the following is based on the USBR technical publication Design of Small Canal Structures (1978) I. An Example Culvert Design Design a concrete culvert using the

### ABB Flowmeters Review of industrial processing flowmeters

White paper ABB Flowmeters Review of industrial processing flowmeters Understanding flowmeters and their pros and cons By Ron DiGiacomo, ABB Measurement Products One of the most critical measurements in

### ME 120 Experimental Methods. Flow Measurement. BJ Furman 02MAY06

ME 120 Experimental Methods Flow Measurement 02MAY06 Fluid Flow Measurement Classification of flow rate measurements Volume flow Industrial process control Heat transfer and cooling Mass flow Industrial

### Aeration Air & Digester Gas Flow Metering Using Thermal Mass Technology. HWEA 2011 Conference Craig S. Johnson

Aeration Air & Digester Gas Flow Metering Using Thermal Mass Technology HWEA 2011 Conference Craig S. Johnson Presentation Overview Introduction Aeration Air & Digester gas challenges Gas flow metering

### 2.0 BASIC CONCEPTS OF OPEN CHANNEL FLOW MEASUREMENT

2.0 BASIC CONCEPTS OF OPEN CHANNEL FLOW MEASUREMENT Open channel flow is defined as flow in any channel where the liquid flows with a free surface. Open channel flow is not under pressure; gravity is the

### CHME 302 CHEMICAL ENGINEERING LABOATORY-I EXPERIMENT 302-1 FLOW MEASUREMENT TEST

CHME 302 CHEMICAL ENGINEERING LABOATORY-I EXPERIMENT 302-1 FLOW MEASUREMENT TEST OBJECTIVE The purpose of the experiment is to demonstrate the working principles of four most common devices used to measure

### Advanced Differential Pressure Flowmeter Technology V-CONE FLOW METER TECHNICAL BRIEF

Advanced Differential Pressure Flowmeter Technology V-CONE FLOW METER TECHNICAL BRIEF Table of Contents Section 1 - General Introduction 1.1 1 Principles Of Operation 1.2 1 Reshaping The Velocity Profile

### Experimental investigation of fluid flow in horizontal pipes system of various cross-section geometries

EPJ Web of Conferences 67, 006 (014) DOI: 10.1051/ epjconf/ 01467006 C Owned by the authors, published by EDP Sciences, 014 Experimental investigation of fluid flow in horizontal pipes system of various

### CEE 410: Hydraulic Engineering Lab 2 Flow Measurement Test at Unit Well 30 November 26 th 2014. Tyler Gates Bill Hussey Megan Long Senn Mettel

CEE 410: Hydraulic Engineering Lab 2 Flow Measurement Test at Unit Well 30 November 26 th 2014 Tyler Gates Bill Hussey Megan Long Senn Mettel Introduction This laboratory experiment involved the collecting

### Applied Fluid Mechanics

Applied Fluid Mechanics Sixth Edition Robert L. Mott University of Dayton PEARSON Prentkv Pearson Education International CHAPTER 1 THE NATURE OF FLUIDS AND THE STUDY OF FLUID MECHANICS 1.1 The Big Picture

### Testing Protocol for Differential Pressure Measurement Devices API MPMS Chapter 22.2

Testing Protocol for Differential Pressure Measurement Devices API MPMS Chapter 22.2 Steve Baldwin Chevron Energy Technology Company Houston, Texas USA Casey Hodges CEESI Measurement Solutions. Nunn, Colorado

### Pipe Flow-Friction Factor Calculations with Excel

Pipe Flow-Friction Factor Calculations with Excel Course No: C03-022 Credit: 3 PDH Harlan H. Bengtson, PhD, P.E. Continuing Education and Development, Inc. 9 Greyridge Farm Court Stony Point, NY 10980

### The Buyer s Guide to Ultrasonic Flowmeters

Spire Metering Technology The Buyer s Guide to Ultrasonic Flowmeters Selecting the Right Flowmeter for Your Liquid Application Introduction Today s advanced ultrasonic flow meters enable the accurate measurement

### New Method for Accurate High Reynolds Metering Uses Water Calibration for Significant Cost Savings White Paper

New Method for Accurate High Reynolds Metering Uses Water Calibration for Significant Cost Savings White Paper Page 1 of 3 White Paper Jonathan Hollist, V-Cone Engineer Nick Voss, V-Cone Product Manager

### Equivalents & Conversion Factors 406 Capacity Formulas for Steam Loads 407 Formulas for Control Valve Sizing 408-409

Engineering Data Table of Contents Page No. I II Formulas, Conversions & Guidelines Equivalents & Conversion Factors 406 Capacity Formulas for Steam Loads 407 Formulas for Control Sizing 408-409 Steam

### Chapter 8: Flow in Pipes

Objectives 1. Have a deeper understanding of laminar and turbulent flow in pipes and the analysis of fully developed flow 2. Calculate the major and minor losses associated with pipe flow in piping networks

### Smart Electromagnetic Flowmeter Open channel Flowmeter Detector

Magne3000 PLUS Smart Electromagnetic Flowmeter Open channel Flowmeter Detector Model NNK150/951 OVERVIE The Magne3000 PLUS Electromagnetic Flowmeter is submersible type of flowmeter mainly used for flow

### Understanding the Flo-Dar Flow Measuring System

Understanding the Flo-Dar Flow Measuring System Independent tests verify non-contact flowmeter is highly accurate under both open channel and surcharge conditions TM Flo-Dar is the only non-contact open

### WHITE PAPER. A New Approach to Accurate Water Flow Measurement - The Study of the FPI Mag Accuracy

WHITE PAPER A New Approach to Accurate Water Flow Measurement - The Study of the FPI Mag Accuracy A New Approach to Accurate Water Flow Measurement - The Study of the FPI Mag Accuracy Author: Nicholas

### 6. FLOW MEASUREMENT. A. Evaluation of Permittee's Flow Measurement

6. FLOW MEASUREMENT Contents Page A. Evaluation of Permittee's Objectives and Requirements... 6-1 Evaluation of Facility-Installed Flow Devices and Data... 6-1 Evaluation of Permittee Data Handling and

### 11. IMPACT OF A JET. Introduction

11. IMPACT OF A JET Introduction Water turbines are widely used throughout the world to generate power. In the type of water turbine referred to as a Pelton wheel, one or more water jets are directed tangentially

### Open Channel Flow. M. Siavashi. School of Mechanical Engineering Iran University of Science and Technology

M. Siavashi School of Mechanical Engineering Iran University of Science and Technology W ebpage: webpages.iust.ac.ir/msiavashi Email: msiavashi@iust.ac.ir Landline: +98 21 77240391 Fall 2013 Introduction

### FIGURE P8 50E FIGURE P8 62. Minor Losses

8 48 Glycerin at 40 C with r 1252 kg/m 3 and m 0.27 kg/m s is flowing through a 4-cm-diameter horizontal smooth pipe with an average velocity of 3.5 m/s. Determine the pressure drop per 10 m of the pipe.

### inter.noise 2000 The 29th International Congress and Exhibition on Noise Control Engineering 27-30 August 2000, Nice, FRANCE

Copyright SFA - InterNoise 2000 1 inter.noise 2000 The 29th International Congress and Exhibition on Noise Control Engineering 27-30 August 2000, Nice, FRANCE I-INCE Classification: 2.1 WHISTLING PHENOMENA

### Chapter (1) Fluids and their Properties

Chapter (1) Fluids and their Properties Fluids (Liquids or gases) which a substance deforms continuously, or flows, when subjected to shearing forces. If a fluid is at rest, there are no shearing forces

### Module 9: Basics of Pumps and Hydraulics Instructor Guide

Module 9: Basics of Pumps and Hydraulics Instructor Guide Activities for Unit 1 Basic Hydraulics Activity 1.1: Convert 45 psi to feet of head. 45 psis x 1 ft. = 103.8 ft 0.433 psi Activity 1.2: Determine

### Module 3. Irrigation Engineering Principles. Version 2 CE IIT, Kharagpur

Module 3 Irrigation Engineering Principles Lesson 10 Distribution and Measurement Structures for Canal Flows Instructional objectives On completion of this lesson, the student shall learn: 1. What are

### hose-to-hose coupling pump-to-hose coupling

pipe_02 A homeowner plans to pump water from a stream in their backyard to water their lawn. A schematic of the pipe system is shown in the figure. 3 m 1 m inlet pipe-to-pump coupling stream re-entrant

### Fluid Mechanics. Fluid Statics [3-1] Dr. Mohammad N. Almasri. [3] Fall 2010 Fluid Mechanics Dr. Mohammad N. Almasri [3-1] Fluid Statics

1 Fluid Mechanics Fluid Statics [3-1] Dr. Mohammad N. Almasri Fluid Pressure Fluid pressure is the normal force exerted by the fluid per unit area at some location within the fluid Fluid pressure has the

### Practice Problems on Pumps. Answer(s): Q 2 = 1850 gpm H 2 = 41.7 ft W = 24.1 hp. C. Wassgren, Purdue University Page 1 of 16 Last Updated: 2010 Oct 29

_02 A centrifugal with a 12 in. diameter impeller requires a power input of 60 hp when the flowrate is 3200 gpm against a 60 ft head. The impeller is changed to one with a 10 in. diameter. Determine the

### Custody Transfer Measurement. with the V-Cone Flowmeter

Custody Transfer Measurement with the V-Cone Flowmeter Stephen A. Ifft McCrometer Inc. Hemet, California, USA Abstract This paper will discuss the approval of the McCrometer V-Cone flowmeter for custody

### Pipe Loss Experimental Apparatus

Pipe Loss Experimental Apparatus Kathleen Lifer, Ryan Oberst, Benjamin Wibberley Ohio Northern University Ada, OH 45810 Email: b-wibberley@onu.edu Abstract The objective of this project was to develop

### FLUID MECHANICS IM0235 DIFFERENTIAL EQUATIONS - CB0235 2014_1

COURSE CODE INTENSITY PRE-REQUISITE CO-REQUISITE CREDITS ACTUALIZATION DATE FLUID MECHANICS IM0235 3 LECTURE HOURS PER WEEK 48 HOURS CLASSROOM ON 16 WEEKS, 32 HOURS LABORATORY, 112 HOURS OF INDEPENDENT

### ME 305 Fluid Mechanics I. Part 4 Integral Formulation of Fluid Flow

ME 305 Fluid Mechanics I Part 4 Integral Formulation of Fluid Flow These presentations are prepared by Dr. Cüneyt Sert Mechanical Engineering Department Middle East Technical University Ankara, Turkey

### EXPERIMENT NUMBER 6 PERFORMANCE TEST OF AN IMPULSE TURBINE

EXPERIMENT NUMBER 6 PERFORMANCE TEST OF AN IMPULSE TURBINE OBJECTIVES The primary objective of this experiment is to determine the characteristics of a small impulse turbine and to compare these characteristics

### Entrance Conditions. Chapter 8. Islamic Azad University

Chapter 8 Convection: Internal Flow Islamic Azad University Karaj Branch Entrance Conditions Must distinguish between entrance and fully developed regions. Hydrodynamic Effects: Assume laminar flow with

### THE THERMAL FLOW METER, A GAS METER FOR ENERGY MEASUREMENT

THE THERMAL FLOW METER, A GAS METER FOR ENERGY MEASUREMENT Kazuto Otakane, Tokyo Gas Co., Ltd Katsuhito Sakai, Tokyo Gas Co., Ltd Minoru Seto, Tokyo Gas Co., Ltd 1. INTRODUCTION Tokyo Gas s new gas meter,

### Fluid Mechanics Definitions

Definitions 9-1a1 Fluids Substances in either the liquid or gas phase Cannot support shear Density Mass per unit volume Specific Volume Specific Weight % " = lim g#m ( ' * = +g #V \$0& #V ) Specific Gravity

### THE USE OF AIR ATOMIZING NOZZLES TO PRODUCE SPRAYS WITH FINE DROPLETS

14th International Water Mist Conference THE USE OF AIR ATOMIZING NOZZLES TO PRODUCE SPRAYS WITH FINE DROPLETS Dr.Gökhan BALIK Etik Muhendislik Danismanlik Tasarim ve Egitim Hizmetleri Ltd.Şti. Istanbul

### Chapter 13 OPEN-CHANNEL FLOW

Fluid Mechanics: Fundamentals and Applications, 2nd Edition Yunus A. Cengel, John M. Cimbala McGraw-Hill, 2010 Lecture slides by Mehmet Kanoglu Copyright The McGraw-Hill Companies, Inc. Permission required

### Using meters to measure steam flow

Page 1 of 6 April 1998 Plant Engineering SELECTING NEW TECHNOLOGY: PROBLEM SOLVING Using meters to measure steam flow Jesse L. Yoder, Ph.D., Senior Analyst, Automation Research Corp., Dedham, MA Key Concepts

### Transient Mass Transfer

Lecture T1 Transient Mass Transfer Up to now, we have considered either processes applied to closed systems or processes involving steady-state flows. In this lecture we turn our attention to transient

### Training programme on flow measurements

WMO / OMM Autorite du Bassin du Niger Niger-HYCOS and Volta-HYCOS Projects Training programme on flow measurements 2 nd part : Flow measurement techniques Flow measurements and calculations Definitions

### Lecture 25 Design Example for a Channel Transition. I. Introduction

Lecture 5 Design Example for a Channel Transition I. Introduction This example will be for a transition from a trapezoidal canal section to a rectangular flume section The objective of the transition design

### 1. Sprinkler selection 2. Design of the system layout 3. Design of the laterals 4. Design of the mainline 5. Pump and power unit selection

Lecture 3 Sprinkler Characteristics I. Hardware Design Process 1. Sprinkler selection 2. Design of the system layout 3. Design of the laterals 4. Design of the mainline 5. Pump and power unit selection

### Fluid Flow Instrumentation

Fluid Flow Instrumentation In the physical world, mechanical engineers are frequently required to monitor or control the flow of various fluids through pipes, ducts and assorted vessels. This fluid can

### INTRODUCTION TO FLUID MECHANICS

INTRODUCTION TO FLUID MECHANICS SIXTH EDITION ROBERT W. FOX Purdue University ALAN T. MCDONALD Purdue University PHILIP J. PRITCHARD Manhattan College JOHN WILEY & SONS, INC. CONTENTS CHAPTER 1 INTRODUCTION

### FLOW METER SPECIFICATION AND APPLICATION ANALYSIS Class #41

FLOW METER SPECIFICATION AND APPLICATION ANALYSIS Class #41 Curtis Gulaga Business Development Manager CB Engineering Ltd. #20, 5920 11 th Street SE Calgary, AB T2H 2M4 1.0 Introduction Today s working

### These slides contain some notes, thoughts about what to study, and some practice problems. The answers to the problems are given in the last slide.

Fluid Mechanics FE Review Carrie (CJ) McClelland, P.E. cmcclell@mines.edu Fluid Mechanics FE Review These slides contain some notes, thoughts about what to study, and some practice problems. The answers

### Determining Optimal Discharge and Optimal Penstock Diameter in Water Turbines

Utah State University DigitalCommons@USU 6ᵗʰ International Symposium on Hydraulic Structures Jun 9th, 1:30 PM - 3:30 PM Determining Optimal Discharge and Optimal Penstock Diameter in Water Turbines Arturo

### What is the most obvious difference between pipe flow and open channel flow????????????? (in terms of flow conditions and energy situation)

OPEN CHANNEL FLOW 1 3 Question What is the most obvious difference between pipe flow and open channel flow????????????? (in terms of flow conditions and energy situation) Typical open channel shapes Figure

### ME 144: Heat Transfer Convection Relations for External Flows. J. M. Meyers

ME 144: Heat Transfer Convection Relations for External Flows Empirical Correlations Generally, convection correlations for external flows are determined experimentally using controlled lab conditions

### 191: Calibration and Standards in Flow Measurement. Richard Paton National Engineering Laboratory, Scotland, UK 1 GENERAL PRINCIPLES

191: Calibration and Standards in Measurement Richard Paton National Engineering Laboratory, Scotland, UK 1 General Principles 1 2 Gravimetric Calibration of Liquid meters 2 3 Volumetric Calibration of

### Gertrudys B. Adkins, Ph.D. Division of Water Rights 2006

FLOW MEASUREMENT DEVICES By Gertrudys B. Adkins, Ph.D. Division of Water Rights 2006 TABLE OF CONTENT INTRODUCTION...1 FLOW MEASUREMENT DEVICES...1 FLOW MEASUREMENT BASICS...2 OPEN CHANNEL FLOW METHODS...3

### Applied Fluid Mechanics

Applied Fluid Mechanics 1. The Nature of Fluid and the Study of Fluid Mechanics 2. Viscosity of Fluid 3. Pressure Measurement 4. Forces Due to Static Fluid 5. Buoyancy and Stability 6. Flow of Fluid and

### METER GATE CALIBRATION AND ACCURACY ANALYSIS

METER GATE CALIBRATION AND ACCURACY ANALYSIS by Ryan Fulton BioResource and Agricultural Engineering BioResource and Agricultural Engineering Department California Polytechnic State University San Luis

### Friction Factor Tests on High Density Polyethylene Pipe

Utah State University DigitalCommons@USU Reports Utah Water Research Laboratory January 1988 Friction Factor Tests on High Density Polyethylene Pipe Steven L. Barfuss J. Paul Tullis Follow this and additional

### physics 1/12/2016 Chapter 20 Lecture Chapter 20 Traveling Waves

Chapter 20 Lecture physics FOR SCIENTISTS AND ENGINEERS a strategic approach THIRD EDITION randall d. knight Chapter 20 Traveling Waves Chapter Goal: To learn the basic properties of traveling waves. Slide

### Theory overview of flow measurement using differential pressure devices based on ISO-5167 standard.

Theory overview of flow measurement using differential pressure devices based on ISO-567 standard. rian FL40 flow computer description. Flow Cad software users manual. Technical note, Differential pressure

### Technical Publication

Technical Publication Flow Meter Selection for Improved Gas Flow Measurements: Comparison of Differential Pressure and Thermal Dispersion Technologies Art Womack, Sr. Applications Engineer Fluid Components

### δy θ Pressure is used to indicate the normal force per unit area at a given point acting on a given plane.

2 FLUID PRESSURES By definition, a fluid must deform continuously when a shear stress of any magnitude is applied. Therefore when a fluid is either at rest or moving in such a manner that there is no relative

### Characteristic Study of Ultrasonic Intensity in a Gas-Liquid Flow System

Characteristic Study of Ultrasonic Intensity in a Gas-Liquid Flow System Characteristic Study of Ultrasonic Intensity in a Gas-Liquid Flow System Nattadate Fuangworawong Dept. of Mechanical Engineering,

### DEVELOPMENT OF HIGH SPEED RESPONSE LAMINAR FLOW METER FOR AIR CONDITIONING

DEVELOPMENT OF HIGH SPEED RESPONSE LAMINAR FLOW METER FOR AIR CONDITIONING Toshiharu Kagawa 1, Yukako Saisu 2, Riki Nishimura 3 and Chongho Youn 4 ABSTRACT In this paper, we developed a new laminar flow

### E 490 Fundamentals of Engineering Review. Fluid Mechanics. M. A. Boles, PhD. Department of Mechanical & Aerospace Engineering

E 490 Fundamentals of Engineering Review Fluid Mechanics By M. A. Boles, PhD Department of Mechanical & Aerospace Engineering North Carolina State University Archimedes Principle and Buoyancy 1. A block

### Pumps: Convert mechanical energy (often developed from electrical source) into hydraulic energy (position, pressure and kinetic energy).

HYDRAULIC MACHINES Used to convert between hydraulic and mechanical energies. Pumps: Convert mechanical energy (often developed from electrical source) into hydraulic energy (position, pressure and kinetic

### Steven R. McManus Environmental Product Support Teledyne Isco

Steven R. McManus Environmental Product Support Teledyne Isco Flow Measurement Applications and Technologies. Methods of flow measurement Technologies of flow measurement Flow Rate Applications Channel

### Flow Like An Egyptian: The Basics of Open channel flow

Flow Like An Egyptian: The Basics of Open channel flow Walt Boyes,as ISA Life Fellow, Fellow of InstMC, Chartered Measurement and Control Technologist, Editor of Control and ControlGlobal.com and principal

### APPENDIX A CONTROL VALVE TESTING PROCEDURES AND EQUATIONS FOR LIQUID FLOWS

APPENDIX A CONTROL VALVE TESTING PROCEDURES AND EQUATIONS FOR LIQUID FLOWS Section A.1. Flow Coefficients Definition The flow coefficient or pressure loss coefficient is used to relate the pressure loss

### Pacific Pump and Power

Pacific Pump and Power 91-503 Nukuawa Street Kapolei, Hawaii 96707 Phone: (808) 672-8198 or (800) 975-5112 Fax: (866) 424-0953 Email: sales@pacificpumpandpower.com Web: www.pacificpumpandpower.com Table

### The Unique Accelabar Flow Meter

The Unique Accelabar Flow Meter The Accelabar is a new and unique flow meter that combines two differential pressure technologies to produce operating ranges never before attainable in a single flow meter.

### CE 6303 MECHANICS OF FLUIDS L T P C QUESTION BANK PART - A

CE 6303 MECHANICS OF FLUIDS L T P C QUESTION BANK 3 0 0 3 UNIT I FLUID PROPERTIES AND FLUID STATICS PART - A 1. Define fluid and fluid mechanics. 2. Define real and ideal fluids. 3. Define mass density