1 page 1 of 17 Technology for Vacuum Systems Instructions for use DVR 2 Analog/Digital Rough Vacuum Gauge
2 page 2 of 17 Dear customer, Your VACUUBRAND vacuum gauge shall support you at your work for a long time without any trouble and with full load output. Thanks to our large practical experience we attained much information how you could add to an efficient application and to personal safety. Please read these instructions for use prior to the initial start-up of your controller. VACUUBRAND vacuum gauges are the result of many years of experience in construction and practical operation of these vacuum gauges combined with the latest results in material and manufacturing technology. Our quality maxim is the zero fault principle : Every vacuum gauge leaving our company, is tested intensively including an endurance run. Therefore faults, even those which occur rarely, are identified and can be eliminated immediately. The achievement of the specifications after the endurance run is tested for every device. Every VACUUBRAND device achieves the specifications. We are committed to providing our customers with this high quality standard. We know that the device cannot replace all of your real work and hope that our products contribute to an effective and trouble-free realisation of your work. Yours VACUUBRAND GMBH + CO KG After sales service: Contact your local dealer or call The document Safety information for vacuum equipment is part of this manual! Read the Safety information for vacuum equipment and observe the instructions contained therein! Trademark index: VACUU LAN (US-Reg.No 3,704,401), VACUU BUS, VACUU CONTROL TM, chemistry-hybrid TM, Peltronic, TURBO MODE TM, VARIO, VARIO-SP TM, VACUUBRAND (US-Reg.No 3,733,388) and also the shown company logos are trademarks of VACUUBRAND GMBH + CO KG in Germany and/or other countries.
3 page 3 of 17 Contents Safety information!...4 General information...4 Intended use...4 Connecting the device...4 Operating conditions...5 Safety during operation...5 Maintenance and repair...5 Technical data...7 Wetted parts...7 Use and operation...8 Changing the pressure unit...8 Pressure measurement...9 Clock symbol...9 Adjusting the operating time and the measuring cycle...9 Display of the battery s discharge status...10 Troubleshooting...10 Readjustment of the vacuum gauge DVR Adjustment at atmospheric pressure Adjustment under vacuum Adjustment to a reference pressure...12 Calibration in the factory...12 Replacing the battery...13 Notes on return to the factory...14 Health and safety clearance form...15 EC Declaration of Conformity...16 DANGER indicates a hazardous situation which, if not avoided, will result in death or serious injury. WARNING indicates a hazardous situation which, if not avoided, could result in death or serious injury. CAUTION indicates a hazardous situation which, if not avoided, could result in minor or moderate injury. NOTICE NOTICE is used to address practices not related to personal injury. Note: The device contains a battery! Remove battery before disposal (see section Replacing the battery ) and dispose of battery and device separately and according to regulations.
4 page 4 of 17 Safety information! General information Read and comply with this manual before installing or operating the equipment. Do not use the equipment if it is damaged. NOTICE Remove all packing material, remove the product from its packing-box, remove the protective covers and keep, inspect the equipment. If the equipment is damaged, notify the supplier and the carrier in writing within three days; state the item number of the product together with the order number and the supplier s invoice number. Retain all packing material for inspection. If the equipment is not used immediately, replace the protective covers. Store the equipment in suitable conditions. Intended use The device is designed for ambient and gas temperatures at the pressure transducer connection of +10 C to +40 C at continuous operation, up to +80 C for short periods (< 5 minutes) at the pressure transducer. If installing the device into a cabinet or a housing check maximum temperature. Ensure that the maximum permitted gas temperature at the pressure transducer (see Technical data ) is not exceeded. NOTICE Use the equipment and all system parts for the intended use only, i.e. for measurement of vacuum in vessels designed for that purpose. Connecting the device Avoid uncontrolled overpressure (e. g. when connecting to a locked or blocked tubing system). Risk of bursting. Comply with max. permitted ambient and gas temperature (see Technical data ) and make sure ventilation is adequate if the equipment is installed in a housing or if ambient temperature is elevated. Power supply by an internal 9V battery. Do not charge the battery. Do not open the battery. Do not throw the battery into fire. Do not heat over 100 C and do not bring in contact with water. Comply with maximum permissible pressures at the pressure transducer. See section Technical data. Connect hoses gas tight at the pressure transducer connection. Ensure the stability of hose connections. In case of residues or aggressive or condensable media install a gas washing bottle if necessary. NOTICE Avoid high heat supply (e.g., due to hot process gases). Connect the vacuum gauge to the vacuum application. Position the vacuum gauge and its vacuum line in such a way that condensate cannot flow towards the internal pressure transducer.
5 page 5 of 17 Allow the equipment to equilibrate to ambient temperature if you bring it from cold environment into a room prior to operation. Notice if there is water condensation on cold surfaces. Comply with all applicable and relevant safety requirements (regulations and guidelines). Implement the required actions and adopt suitable safety measures. The battery is free from mercury and cadmium. Recommended type of battery see section Technical data. When replacing the battery, always mind its correct orientation when inserting it into the battery compartment. When using a battery of a different chemical system (e. g. alkaline), lifetime may be considerably shorter. Because of their lower capacity, the use of rechargeable batteries is not recommended. Operating conditions The devices are not suitable for applications which involve dangerous or explosive gases or explosive or flammable mixtures. Ensure that the materials of the equipment s wetted parts are compatible with the substances in the vacuum system, see section Technical data. Safety during operation Adopt suitable measures to prevent the release of dangerous, toxic, explosive, corrosive, noxious or polluting fluids, vapours and gases. Prevent any part of the human body from coming in contact with the vacuum. Attention: If the pressure is higher than approximately 1060 mbar the pressure reading becomes incorrect (saturation of the pressure transducer). The display flashes. Immediate pressure relief necessary! Risk of bursting! Use only genuine spare parts and accessories. Otherwise safety and performance of the equipment as well as the electromagnetic compatibility of the equipment might be reduced. Possibly the CE mark becomes void if not using OEM spare parts. NOTICE Electronic equipment is never 100% fail-safe. This may lead to an ill-defined status of the equipment or of other connected devices. Provide protective measures against malfunction and failure. Ensure that in case of failure the device and the vacuum system always will turn into a safe status. Maintenance and repair Attention: Due to the operation the equipment might be contaminated by harmful or dangerous substances, clean or decontaminate prior to maintenance.
6 page 6 of 17 NOTICE Ensure that maintenance is done only by suitably trained and supervised technicians. Interior components of the device can only be repaired at the factory. Returned products will not be repaired or calibrated until the completed health and safety clearance form has been received. In order to comply with law (occupational, health and safety regulations, safety at work law and regulations for environmental protection) vacuum pumps, components and measuring instruments returned to the manufacturer can be repaired only when certain procedures (see section Notes on return to the factory ) are followed.
7 page 7 of 17 Technical data Type DVR 2 Measuring principle Pressure transducer Pressure reading Pressure units Measuring cycle Automatic sleep mode Measuring range (absolute) Maximum permissible pressure at pressure transducer Measurement uncertainty (absolute) after careful adjustment and at constant temperature Temperature coefficient Maximum permissible temperature of gaseous media at pressure transducer Ambient temperature range (operation) Ambient temperature range (storage) Permitted relative atmospheric moisture during operation (no condensation) capacitive, absolute pressure, gas type independent ceramic diaphragm (alumina), analogue and digital LCD display mbar, Torr or hpa (selectable) automatic, once in 3s, once in 1s, three times in 1s (factory-set: automatic) operation time: minutes or continuous operation ( On ), (factory-set: 15 minutes) 1080 mbar - 1 mbar (810 Torr - 1 Torr) 1.5 bar absolute <±1 mbar (0.75 Torr) / ±1 digit <±0.07 mbar/k (0.05 Torr/K) continuous operation: 40 C, for short periods (< 5 minutes): up to 80 C 10 C to +40 C -10 C to +60 C 30% to 85% Degree of protection according to IEC 529 IP 40 Vacuum connection Power supply Internal volume of measurement chamber Weight approx. Dimensions L x W x H small flange KF DN 16 and screw-in hose nozzle DN 6/10 mm battery 9V Lithium, 1.2 Ah, Ultralife U 9VL 4.23 cm 3 (without hose nozzle) 375 g 115 mm x 115 mm x 66 mm Wetted parts Components Sensor Sensor housing Seals Vacuum connection / hose nozzle Wetted materials Aluminium oxide ceramic stainless steel chemically resistant fluoroelastomer PBT / PPS glass fibre reinforced We reserve the right for technical modification without prior notice!
8 page 8 of 17 Use and operation Display adjustment mode/ warning symbol automatic sleep mode replace battery/ display of battery capacity Keys ON/OFF key UP/DOWN key MODE key pressure units, selectable The keys are on the rear side of the device underneath the symbols at the front side. Changing the pressure unit + Torr mbar Switch the vacuum gauge off. Press MODE key and keep it pressed while switching the DVR 2 on ( ON/OFF key). The pressure unit as of the last operation (Torr, mbar or hpa) is displayed. Select the desired pressure unit with the UP / DOWN key. Press ON/OFF key to confirm the selected pressure unit and to terminate the mode.
9 page 9 of 17 Pressure measurement Upon switching on, the absolute pressure (total pressure, independent of gas type) ranging from 1 mbar to 1080 mbar is displayed in the selected pressure unit (Torr, mbar or hpa). The pressure transducer was adjusted using factory standards. In general, there is no need for readjustment by the user because of the excellent longtime stability (see section Readjustment ). Clock symbol Adjusting the operating time and the measuring cycle The vacuum gauge is equipped with an automatic sleep mode to save battery power when the instrument is not used. When the clock symbol appears on the display, the vacuum gauge will switch off automatically after approx. 1 minute. Press ON/OFF key. The clock symbol will disappear and the preselected operation time will start again. The time until automatic switching off (operating time) can be set by the user within a range from 1 to 1000 minutes (factory-set: 15 minutes) or continuous operation On can be selected. Press MODE key to select the time setting mode. The clock symbol and the currently set operating time in minutes are displayed. The arrow to the right indicates that the time can be increased using UP/DOWN key. Press UP/DOWN key to increase operating time. To decrease the operating time: Adjusting the measuring cycle: Press MODE key to change the arrow to the left. Press UP/DOWN key to decrease the operating time. Press ON/OFF key to confirm. The device switches to the mode Adjusting the measuring cycle. The measuring cycle can be set to Automatic, 1 measurement in 3s, 1 measurement in 1s and 3 measurements in 1s. The measuring cycle as of the last operation (factory-set: A ) is displayed. Press UP/DOWN key to select the measuring cycle: A = Automatic (automatic determination of the measuring cycle; frequent measurement in case of great pressure change), S1 = 1 measurement in 3s, S2 = 1 measurement in 1s, S3 = 3 measurements in 1s. The flashing of the arrow (only S1, S2 or S3 ) symbolizes the frequency of the pressure measurement). Press MODE key to change the direction of the adjustment.
10 page 10 of 17 Press ON/OFF key to confirm the measuring cycle and to terminate the mode. Display of the battery s discharge status When switching off the device, the remaining battery capacity is displayed qualitatively by the number of arrows in the display. If the battery symbol appears in the display during measurement, the battery should be replaced. The correct operation of the device is no longer ensured. Troubleshooting Fault Possible cause Remedy No display or display disappears. Incorrect pressure reading. Display is flickering and/ or battery symbol appears. Adjustment mode cannot be activated. All segments of the LCD are displayed or no display although battery has been replaced. Automatic sleep mode? Switch on, increase operating time if necessary. Battery flat? Replace battery. Other cause? Contact local distributor. Pressure transducer not correctly adjusted? Moisture in the pressure transducer? Battery flat, battery symbol is displayed? Readjust pressure transducer. Dry or evacuate the pressure transducer and readjust if necessary. Replace battery. Battery flat? Replace battery. At the applied pressure, readjustment is not possible. Pressure transducer or measuring electronics defective? Admit air to atmospheric pressure or evacuate to a pressure below 20 mbar. Contact local distributor.
11 page 11 of 17 Readjustment of the vacuum gauge DVR 2 The equipment has been adjusted at the factory. In general, there is no need for adjustment by the user due to the excellent longtime stability. Depending on operation conditions, type of application and accuracy requirements, an inspection and readjustment may become necessary. Adjustment at atmospheric pressure Admit air to the vacuum gauge. Make sure that the vacuum connection at the vacuum gauge is at atmospheric pressure. Attention: Determine the exact actual atmospheric pressure, e. g. by using an accurate barometer or get accurate reading from the weather service, e. g. at the next airport, etc. (take into account the difference in altitude between e. g. airport and laboratory). + Press UP/DOWN key and keep it pressed while switching on. The vacuum gauge switches into the adjustment mode (indicated by a warning triangle). The arrow at the right indicates, that the pressure reading can be increased using UP/ DOWN key. Press UP/DOWN key to increase the reading to the actual atmospheric pressure. To reduce the reading: Press MODE key to change the direction of adjustment. The arrow moves to the left indicating that the pressure reading can be reduced. Press UP/DOWN key to reduce the reading. Press ON/OFF key to confirm the adjustment and to terminate the adjustment mode. Adjustment under vacuum Evacuate the vacuum gauge to a pressure <0.5 mbar (<0.4 Torr) e. g. by applying a good rotary vane pump. Press UP/DOWN key and keep it pressed while switching on. + The vacuum gauge switches into the adjustment mode (indicated by a warning triangle). The reading is adjusted automatically to zero. Attention: Adjustment under vacuum with a pressure higher than 0.5 mbar (0.4 Torr) reduces the accuracy of measurement. If the pressure is significantly higher than 0.5 mbar (0.4 Torr), adjustment to a reference pressure is recommended. Press ON/OFF key to confirm the adjustment and to terminate the adjustment mode.
12 page 12 of 17 Adjustment to a reference pressure Evacuate the vacuum gauge to an exactly known reference pressure within the range of mbar ( Torr). Switch the vacuum gauge to the adjustment mode (see Adjustment under vacuum ). Press UP/DOWN key to adjust the reading from 0 ( zero ) to the actual reference pressure in the vacuum line in the range of 0 to 20 mbar (0 to 15 Torr). Press ON/OFF key to confirm the adjustment and to terminate the adjustment mode. Attention: The accuracy of the value of the reference pressure will directly affect the accuracy of the adjustment. If the nominal ultimate vacuum of a diaphragm pump is used as reference vacuum, the accuracy of the adjustment of the vacuum gauge might be doubtful. The diaphragm pump may not achieve the specified value due to condensate, poor state, failure of the valves or the diaphragm. Calibration in the factory Control of measuring equipment The VACUUBRAND DAkkS calibration laboratory is accredited by the Deutsche Akkreditierungsstelle GmbH (national accreditation body of the Federal Republic of Germany) for the measurable variable pressure in the pressure range from 7.5*10-4 Torr to 975 Torr (10-3 mbar to 1300 mbar) in accordance with the general criteria for the operation of testing laboratories defined in the DIN EN ISO/IEC 17025:2000 series of standards (accreditation number D-K ). The DAkkS is signatory to the multilateral agreements of the European cooperation for Accreditation (EA) and of the International Laboratory Accreditation Cooperation (ILAC) for the mutual recognition of calibration certificates. Rely on calibration in the VACUUBRAND calibration laboratory: - To meet the requirements of the DIN ISO 9000ff and series of standards regarding the calibration of inspection, measuring and test equipment at specified intervals. - To document that the vacuum gauges calibrated are traceable to national standards of the PTB (Physikalisch-Technische Bundesanstalt; German national institute for science and technology and the highest technical authority of the Federal Republic of Germany for the field of metrology and certain sectors of safety engineering). DAkkS calibration DVR
13 page 13 of 17 Replacing the battery Note: Some of the components on the printed circuit board are sensitive to electrostatic discharge. Do not touch components. If necessary, ground the printed circuit board and the person replacing the battery suitably. Attention: Use only a battery of the recommended type (see Technical data ). The battery must not be charged or connected to an external voltage. Unscrew the four countersunk head screws at the rear side of the housing with a Phillips screw driver. Disassemble the housing parts carefully. Attention: The two housing parts are connected by the sensor cable. printed circuit board battery switch pad Remove the battery. Insert the new battery into the battery compartment correctly matching the positive and negative polarity (as shown by the engraving inside the compartment). When reassembling the vacuum gauge, make sure that the switch pad and the printed circuit board are positioned correctly. Carefully screw both parts of the housing together.
14 page 14 of 17 Notes on return to the factory Repair - return - DAkkS calibration NOTICE If the measures listed in section Troubleshooting do not work: Return the equipment to the factory for inspection and repair. Interference with the equipment (except replacement of the battery) voids all warranties. Safety and health of our staff, laws and regulations regarding the handling of dangerous goods, occupational health and safety regulations and regulations regarding safe disposal of waste require that for all pumps and other products the Health and safety clearance form must be sent to our office duly completed and signed before any equipment is dispatched to our premises. Fax or post a completed copy of the health and safety clearance form to us in advance. The declaration must arrive before the equipment. Enclose a second completed copy with the product. If the equipment is contaminated you must notify the carrier. No repair / DAkkS calibration is possible unless the correctly completed form is returned. Inevitably, there will be a delay in processing the equipment if information is missing or if this procedure is not obeyed. Eventually, acceptance of the returned equipment will be refused. If the product has come in contact with chemicals, radioactive substances or other substances dangerous to health or environment, the product must be decontaminated prior to sending it back to the factory. To expedite repair and to reduce costs, please enclose a detailed description of the problem and the product s operation conditions with every product returned for repair. We submit quotations only on request and always at the customer s expense. If an order is given, the costs incurred are offset from the costs for repair or from the purchase price, if the customer prefers to buy a new product instead of repairing the defective one. If you do not wish a repair on the basis of our quotation, the equipment might be returned to you disassembled and at your charge! We replace parts due to optical aspects upon your request (at your expense). NOTICE Before returning the equipment ensure that (if applicable): Equipment has been cleaned and/or decontaminated. All ports have been sealed. Equipment has been properly packed, if necessary, please order an original packaging (costs will be charged), marked as appropriate and the carrier has been notified. Ensure that the completed health and safety clearance form is enclosed. We hope for your understanding for these measures, which are beyond our control. Scrapping and waste disposal: Dispose of the equipment and any components removed from it safely in accordance with all local and national safety and environmental requirements. Particular care must be taken with components which have been contaminated with dangerous substances from the process. Do not incinerate fluoroelastomer seals and O-rings. You may authorize us to dispose of the equipment at your expense.
15 page 15 of 17 Health and safety clearance form Devices will not be accepted for any handling before we have received this declaration. Please read and comply with Notes on return to the factory. Oil filled pumps: Drain oil prior to shipping absolutely! 1. Device (Model): Serial no.: Reason for return / malfunction: Has the device been used in a copper process step (e.g., semiconductor production). yes no 5. Substances (gases, liquids, solids) in contact with the device / which have been pumped: Prior to return to the factory the device has been decontaminated. yes no Description of the decontamination method and the test / verification procedure: The device is free of hazardous, harmful substances. yes no 8. Protective measures required for VACUUBRAND employees: If the paint is damaged, we wish a repaint or a replacement of parts for reason of appearance (repaint and replacement at customer s expense). yes no 10. Legally binding declaration We assure for the returned device that all substances, which have been in contact with the device are listed in section 5 and that the information is complete and that we have not withheld any information. We declare that all measures - where applicable - have been taken listed in section Return to the factory. By our signature below, we acknowledge that we accept liability for any damage caused by providing incomplete or incorrect information and that we shall indemnify VACUUBRAND from any claims as regards damages from third parties. We are aware that as expressed in 823 BGB (Public Law Code of Germany) we are directly liable for injuries or damages suffered by third parties, particularly VACUUBRAND employees occupied with handling/repairing the product. Shipping of the device must take place according to regulations. Name:... Signature:... Job title:... Company s seal: Date:... Release for repair grant by VACUUBRAND (date / signature):... VACUUBRAND GMBH + CO KG Tel.: Alfred-Zippe-Straße 4 Fax: Wertheim, Germany
16 page 16 of 17 EG-Konformitätserklärung EC Declaration of Conformity Déclaration CE de conformité Hersteller / Manufacturer / Fabricant: VACUUBRAND GMBH + CO KG Alfred-Zippe-Str Wertheim Germany Hiermit erklärt der Hersteller, dass das Gerät konform ist mit den Bestimmungen der Richtlinie 2004/108/ EG. Hereby the manufacturer declares that the device is in conformity with the directive 2004/108/EC. Par la présente, le fabricant déclare, que le dispositif est conforme à la directive 2004/108/CE. Vakuummessgerät / Vacuum gauge / Vacuomètre Typ / Type / Type: DVR 2 Artikelnummer / Order number / Numéro d article: Seriennummer / Serial number / Numéro de série: Siehe Typenschild / See rating plate / Voir plaque signalétique Angewandte harmonisierte Normen / Harmonized standards applied / Normes harmonisées utilisées: DIN EN Bevollmächtigter für die Zusammenstellung der technischen Unterlagen / Person authorised to compile the technical file / Personne autorisée à constituer le dossier technique: Dr. J. Dirscherl VACUUBRAND GMBH + CO KG Alfred-Zippe-Str Wertheim Germany Wertheim, Ort, Datum / place, date / lieu, date (Dr. F. Gitmans) Geschäftsführer / Managing director / Gérant ppa (Dr. J. Dirscherl) Technischer Leiter / Technical Director / Directeur technique VACUUBRAND GMBH + CO KG Alfred-Zippe-Str Wertheim T F
17 page 17 of 17 Disclaimer: Our technical literature is only intended to inform our customer. The validity of general empirical values and results obtained under test conditions for specific applications depend on a number of factors beyond our control. It is therefore strictly the users responsibility to very carefully check the validity of application to their specific requirements. No claims arising from the information provided in this literature will, consequently, be entertained. VACUUBRAND GMBH + CO KG -Technology for Vacuum Systems VACUUBRAND GMBH + CO KG Printed in Germany Alfred-Zippe-Str Wertheim / Germany T F
page 1 of 75 Technology for Vacuum Systems Instructions for use MZ 2C EX MD 4C EX MV 10C EX MZ 2C EX + AK + EK MZ 2C EX + IK + EK MD 4C EX + AK + EK MV 10C EX + AK + EK Chemistry diaphragm pumps with ATEX
Titrette and Digital Burette (all models) Testing Instructions (SOP) May 2013 1. Introduction The standard ISO DIS 8655 describes both the design and the testing of bottle-top burette. The following Testing
page 1 of 63 Technology for Vacuum Systems Instructions for use PC 510 NT / PC 511 NT / PC 520 NT PC 610 NT / PC 611 NT / PC 620 NT Chemistry pumping unit page 2 of 63 Dear customer, Your VACUUBRAND diaphragm
Infrarot-Bewegungsmelder IP44 infrared motion sensors IP44 ODA (weiß) slim ODA (schwarz) slim 95174 96000 ODA (weiß) ODA (schwarz) 95175 96001 Betriebsanleitung User s Manual User s Manual infrared motion
IMPORTANT SAFETY NOTICE To: Our Valued Customers User safety is a major focus in the design of our products. Following the precautions outlined in this manual will minimize your risk of injury. ITT Goulds
Industrial Process Pump Safety Manual IMPORTANT SAFETY NOTICE To: Our Valued Customers User safety is a major focus in the design of our products. Following the precautions outlined in this manual will
Technical Product Information No. 1290 EN Hydraulic control unit series 0086-372-01 Contents Page About this Technical Product Information (TPI) 2 The ORTLINGHAUS numbering system 2 About the product 3
1 919 1919P01 1919P03 QBM65/C, QBM75-1U/C Differential Pressure Sensors for air and non-corrosive gases, with calibration certificate QBM751-1/C QBM65/C QBM75/C Highly accurate measurements with calibration
aseries A13B Mini Bullet Camera User Manual Thank you for purchasing our product. If there are any questions, or requests, please do not hesitate to contact the dealer. This manual applies to the MicroView
BE17 EN OPERATING MANUAL VOLTAGE DETECTOR TRT-BA-BE17-TC-001-EN table of contents Notes regarding the operating manual... 1 Safety... 2 Information about the device... 3 Technical data... 3 Transport and
Rp 3/4 Ergo-Pro Single Line Solar Station Installation and Operating Instructions Item No Pump type 677.21.70 WILO ST 15/6 13 2 4 11 Technical Specifications Max. operating pressure: 6 bar Max. operating
Defender 3000 Series Base Instruction Manual Defender 3000 Series Base EN-1 INTRODUCTION This manual covers installation, and maintenance instructions for the Ohaus Defender TM 3000 Series Base. Please
EPSOLAR LS1024B / LS2024B/ LS3024B Solar Charge Controller USER MANUAL Thank you very much for selecting our product! This manual offers important information and suggestions with respect to installation,
Art.S001-S002 SOLAR MODULE WITH DELIVERY AND RETURN CONNECTIONS Technical Information sheet 0001/07/Rev01 ENG FUNCTION Series S001 and S002 circulation units are applied to the primary circuit of solar
Runtastic rechargeable battery for iphone 4, 4S Thank you very much for purchasing our product, the runtastic rechargeable battery (model no.: RUNBATT1) for the iphone 4. This product is only to be used
Alarm Clock USER GUIDE Jazwares, Inc. 2012 CONTENTS Please read the instructions along with the Alarm Clock carefully before you use it, so that you can operate it conveniently. WELCOME & Warnings Page
This document is hosted by Old Will Knott Scales www.oldwillknottscales.com Defender 3000 Series Base Instruction Manual Compliance This product conforms to the EMC directive 89/336/EEC and the Low Voltage
Operating instructions Supply system for flammable liquids Labor Table of contents Safety General safety instructions.................... 2 Warning instructions used in this operating instruction 3 Correct
0102 Model Number Features 15 mm quasi flush Usable up to SIL2 acc. to IEC 61508 Accessories BF 30 Mounting flange, 30 mm V1-G-N-2M-PUR Female cordset, M12, 2-pin, NAMUR, PUR cable V1-W-N-2M-PUR Female
RI-215A Operator s Manual Part Number: 71-0045RK Revision 0 Released: 10/3/05 Warranty RKI Instruments, Inc., warrants gas alarm equipment sold by us to be free from defects in materials and workmanship,
FT-100 Fire Training Fog Machine User Manual English 2015 Antari Lighting and Effects Ltd. User Manual - English Safety Information Please read the following safety information carefully before operating
MAKING MODERN LIVING POSSIBLE Instruction ATEX instruction for PAHT G 2-90 in dust applications danfoss.high-pressurepumps.com Table of Contents 1. Common demands for Category 2 and 3 installations...3
6262100/6 IM-P626-01 ST Issue 6 PBX Series Quick Fit Connector Bimetallic Steam Traps Installation and Maintenance Instructions 1. Safety information 2. General product information 3. Installation 4. Commissioning
Cable Tester Pro ORDERCODE D1909 Congratulations! You have bought a great, innovative product from DAP Audio. You can rely on DAP Audio, for more excellent audio products. We design and manufacture professional
Operating Instuctions Shaking device SV 14 / 22 SV 29 / 45 page 1 General notes, safety notes, notes on electrical equipment and electromagnetic compatibility, as well as notes on maintenance of the unit
instrumentation for fluids Electromagnetic sensor Series FLOMID FX Instructions Manual! Conforms with the Pressure Equipment Directive 97/23/EC. 0830 This equipment is considered as being a pressure accessory
8 716 473 216-00.3O Gas boiler WBN 6000-30-H-E-N/L-S2400 Operating instructions for the end customer en 2 Contents Contents 1 Key to symbols and safety instructions................... 2 1.1 Key to symbols..................................
BroadBand PowerShield User Manual 990-0375G 12/2006 Chapter 1 General Information The PowerShield provides a power source for broadband telephony and other DC applications. Safety This Safety Guide contains
OCTOPLUS WARNING AND MEASUREMENT CONTROL UNIT FOR TOXIC AND FLAMMABLE GASES USER'S MANUAL Copyright 2013 by Dalemans s.a. Article IMP00000062 Revision V2R0 of 25/11/2013 Any reproduction, whether partial
VD series 53 Vacuum drying ovens Vacuum drying oven for non-flammable solvents A BINDER vacuum drying oven of the VD series is impressive while gently drying with its homogeneous temperature distribution.
SDX Submersible Depth Transmitter User Manual January 2011 USER INFORMATION Stevens makes no warranty as to the information furnished in these instructions and the reader assumes all risk in the use thereof.
GE Sensing & Inspection Technologies Druck PV 6x pneumatic/hydraulic pressure stations safety and quick reference guide - K08 A. PV 6: Pneumatic pressure station (Figure A.) -90 mbar to 0 bar (-3. to 300
For the user Instructions for use ecomax Wall hung room sealed fan assisted condensing boilers ecomax 63/ E ecomax 68/ E ecomax 6/ E ecomax 635 E ecomax 84/ E ecomax 88/ E ecomax 835 E GB Table of contents
Installation and Service Instructions for use by heating contractors Neutralization System Part no. 7441823 Grünbeck part no. 410480 Safety and Installation Requirements Please ensure that these instructions
Digital inductive conductivity transmitter Optimal solution for conductivity measurements in difficult fluids (polluted, dirty,...) PEEK/PPA version for CIP applications Large range of process connections
Programmable indoor transmitter of temperature T0118 Programmable indoor transmitter of atmospheric pressure T2118 Programmable indoor transmitter of temperature, relative humidity and other derived humidity
Operating Instructions Drill rig DRU160 Index 000 / 001 Original operating instructions 10988825 en / 20.10.2009 Congratulations! With a Hydrostress unit from TYROLIT you have chosen a tried and tested
ENGINEERING SPECIFICATIONS FOR CO2 EXTINGUISHING AGENT PRE-ENGINEERED FIRE SUPPRESSION SYSTEMS SECTION 1 GENERAL SPECIFICATIONS I. SCOPE This specification outlines the requirements for a pre-engineered
CAUTION! Instruction manual of pumps WQ, WQF, SN, Furiatka, Furia, KRAKEN, BIG, IP, IPE, SWQ, CTR NOTE: read the manual prior to commencement of use. For safety reasons, the pump can only be operated by
Transferpette Testing Instructions (SOP) February 2016 1. Introduction The standards DIN EN ISO 8655 describe both the design and the testing of the piston operated pipette Transferpette. The following
MAKING MORN LIVING POSSIBLE Danfoss heating living connect Installation and User Guide Contents 1.0 System overview... 3 2.0 Overview of display and control buttons... 3 3.0 Installation - step by step...
2 Contents Components Attachments Guidance Installation Operation Maintenance Service Technical Troubleshooting Symbols used to mark instructions...3 Included Attachments...4 Congratulations...5 Getting
INSTRUCTION MANUAL NFC Communication Tray Model: HHX-IT3-Z ENGLISH CONTENTS Introduction...3 Important safety information...4 Know your unit...6 Transferring data to the PC...7 Care and maintenance...8
1263050/6 IM-P126-07 ST Issue 6 BPW32 DN15, DN20 and DN25 Balanced Pressure Wafer Steam Trap Installation and Maintenance Instructions 1. Safety information 2. General product information 3. Installation
StarLight 3000 OPERATING INSTRUCTIONS h h 2 4 1 1. Quick Start Power Button 2. SOS/Flashlight Power Switch 3. USB Out/In 4. Fuel Gauge (Battery Status Indicator) 3 Figure 1 A B Included items: - 1 HALO
DAB/FM Tuner User Manual Order code: CRAM21 Safety advice WARNING FOR YOUR OWN SAFETY, PLEASE READ THIS USER MANUAL CAREFULLY BEFORE YOUR INITIAL START-UP! Before your initial start-up, please make sure
ITE Indirect calorifier ITE - 400/500/600/750/1000 0310 117 Installation, User and Service Manual Innovation has a name. Read this manual carefully Warning Read this manual carefully before starting the
Electronic pressure measurement Digital pressure gauge for industrial applications Model DG-10-S, Standard version Model DG-10-E, Enhanced version WIKA data sheet PE 81.66 Applications Machine building
TM-47 ORDERCODE D1370 Congratulations! You have bought a great, innovative product from DAP Audio. The DAP Audio Microphone range brings excitement to any venue. Whether you want simple plug-&-play action
The Third Rail System User Manual System Contents: Slim Case and Smart Battery for iphone 4 Slim Case for iphone 4 Smart Battery micro-usb Cable USB Adapter Cable User Manual Smart Battery only Smart Battery
transmitter (2 wire) / switch for continuous or On/Off Control Indication, monitoring, transmitting and continuous or On/Off control in one device Output signal 4...20 ma, 2-wire for continuous control
BRAVER UPS (Uninterruptible Power System) User s Manual Safety CAUTION! This UPS utilizes voltages that may be hazardous. Do not attempt to disassemble the unit. The unit contains no user replaceable parts.
User Guide Nokia Portable Wireless Charging Plate DC-50 Issue 1.1 EN User Guide Nokia Portable Wireless Charging Plate DC-50 Contents For your safety 3 About your accessory 4 Keys and parts 5 Top up your
The device type and date of manufacture (week/year) can be found on the device rating plate. In the event of any queries about the device, please quote all the details given on the rating plate. For further
Instruction Manual Window Wiper AFW900 (Read carefully before use) When using electrical appliances, basic precautions should always be followed, including the fact READ ALL INSTRUCTIONS IN THIS MANUAL
SALTER AIRE Plus COMPRESSOR Model 8350 / 8352 / 8353 Instruction Manual Page 1 of 9 TABLE OF CONTENTS Introduction...3 Important Safeguards..3 System Components...5 Setup and Operation...5 Setup...5 Operation....6
AE PV Heating Element Screwing in Heating Unit Installation Manual Version 01.7 IA_AE_PV_Heating_element_V01.7_EN Title Type of Documentation Purpose of the Documentation Publisher Legal Reservation Copyrights
2 732 Synco living Water detector QFP910 Wireless radio sensor to record water puddles RF communication based on KNX standard (868 MHz unidirectional) Battery operation with standard 1.5 V batteries Use
Bogobit Bremsmodul Oneway Manual bogobit Bremsmodul Oneway 2011-12-29 page 1 / 11 1 Introduction Please read the instructions completely before using the product. Observe all operating and safety instructions!
User s manual FLIR VP50/VP52 Non-contact AC voltage detector User s manual FLIR VP50/VP52 #T559851; r. AD/ 9134/9134; en-us Table of contents 1 Disclaimers... 1 1.1 Copyright... 1 1.2 Quality assurance...
Milwaukee Milwaukee USER MANUAL PORTABLE DISSOLVED OXYGEN METER MODEL: SM600 Smart DO Meter Authorized Dealer: ISMIL600 11/01 PROBE PREPARATION: The meter is supplied with a 9V battery. Slide off the battery
Air Conditioner Water Heater - A Product of HotSpot Energy LLC PLEASE READ THIS BEFORE YOU INSTALL THE UNIT 1. This air conditioner must be installed and/or repaired by a qualified technician. If you perform
Series 510 Submersible Level Transmitters The Series 510 Submersible Level Transmitters are solid state instruments designed for direct submergence into many types of liquid for quick, accurate and reliable
APPENDIX II Guide on the Segregation, Packaging, Labelling and Storage of Laboratory Chemical Wastes for Schools (Prepared by Environmental Protection Department) 1. Introduction The Waste Disposal (Chemical
Uninterruptible Power Supply ERA LED 1.5 ERA LED 2.0 ERA LED 2.6 User s manual Index Safety Warnings... 2 1 Introduction... 3 2 General Characteristics... 4 3 Receipt and site selection... 4 4 EXTERNAL
Electronic pressure measurement Miniature pressure transmitter Model M-10, standard version Model M-11, version with flush diaphragm WIKA data sheet PE 81.25 for further Approvals see page 3 MicroTronic
MultiDim MKII V.2 ORDERCODE 50386 Congratulations! You have bought a great, innovative product from Showtec. The Showtec MultiDim brings excitement to any venue. Whether you want simple plug-&-play action