1 A B VALVE A VALVE B F1 BioFrac Franction Collector F2 F3 F4 F BIOLOGIC DUOFLOW CHROMATOGRAPHY SYSTEM INSTRUCTION MANUAL
2 SAFETY SAFETY! Caution/Warning Disconnect power to any BioLogic DuoFlow system component before servicing. No userserviceable parts are inside any component. Refer servicing to Bio-Rad service personnel. Call Bio-Rad Technical Support Chromatography Group at BioRad. BioLogic DuoFlow systems are certified to meet the I.E.C * standard for safety of laboratory equipment. Certified products are safe to use when operated in accordance with the instruction manual. This safety certification does not extend to other chromatography equipment or accessories not I.E.C certified, even when connected to a BioLogic DuoFlow system. This instrument is intended for laboratory applications only This instrument should not be modified or altered in any way. Alteration of this instrument will void the manufacturer s warranty, void the I.E.C certification, and create a potential safety hazard for the user. BioLogic DuoFlow systems conform to the class A standards for Electromagnetic Emissions, intended for laboratory equipment applications. It is possible that emissions from this product may interfere with some sensitive appliances when placed nearby or on the same circuit as those appliances. The user should be aware of this potential and take appropriate measures to avoid interference. Bio-Rad is not responsible for any injury or damage caused by the use of this instrument for purposes other than for which it is intended or by modifications of the instrument not performed by Bio-Rad or an authorized agent. The BioLogic USB Bitbus Communicator complies with Part 15 of the FCC Rules. Operation is subject to the following two conditions: (1) this device may not cause harmful interference, and (2) this device must accept any interference received, including interference that may cause undesired operation. The USB Bitbus Communicator has been tested and found to comply with the limits for a Class A digital device, pursuant to Part 15 of the FCC Rules. These limits are designed to provide reasonable protection against harmful interference when the device is operated in a commercial environment. This device generates, uses, and can radiate radio frequency energy and, if not installed and used in accordance with the instruction manual, may cause harmful interference to radio communications. Operation of this device in a residential area is likely to cause harmful interference, in which case the user will be required to correct the interference at his own expense. *I.E.C is an internationally accepted electrical safety standard for laboratory instruments. i
3 TABLE OF CONTENTS TABLE OF CONTENTS SAFETY SECTION 1. SYSTEM OVERVIEW Chapter 1.0 Introduction Overview Features Unpacking System Configurations Quick Start Procedure Programming the Maximizer Chapter 2.0 Description of System Components Controller and USB Bitbus Communicator Controller USB Bitbus Communicator Workstation Maximizer Mixers MX-1 Mixer Maximizer Mixer Changing Mixer Capacity Detection Systems UV Detector Conductivity Monitor QuadTec UV/Vis Detector ph Monitor Other Detectors Valves AVR7-3 Sample Inject Valve AVR9-8 Stream Select Valve SV5-4 Buffer Select and Automated Sample Loading Valve SVT3-2 Diverter Valve Fraction Collectors BioFrac Fraction Collector Model 2110 Fraction Collector Model 2128 Fraction Collector Generic Fraction Collectors Sample Loading Options DynaLoop Large Volume Sample Injection Loop Model EP-1 Econo Pump EGP Econo Gradient Pump Other Gradient Pumps System Peripherals System Rack Starter Kit Fittings Kit, including Tubing Kit Fittings Tightener Backpressure Regulator Signal Import Module (SIM) Pump Kits iii
4 TABLE OF CONTENTS Model 1327 Chart Recorder Generic Chart Recorders Uninterruptible Power Supply Printers Columns and Column Fittings Anion Exchange: Q-Strong Anion Exchange Cation Exchange: S-Strong Cation Exchange Anion Exchange: DEAE Weak Anion Exchange Cation Exchange: Carboxy Methyl (CM) Weak Cation Exchange Ceramic Hydroxyapatite (CHT) Size Exclusion Chromatography (SEC) High Pressure Reversed Phase Columns Hydrophobic Interaction Chromatography (HIC) Affinity Chromatography Empty Columns Column Fittings SECTION 2. SYSTEM INSTALLATION AND SETUP Chapter 3.0 System Setup Controller Cable Connections USB Bitbus Communicator Cable Connections Workstation Cable Connections Systems without a Maximizer Systems with a Maximizer System Rack Setup Mixers Detection System Connections UV Detector and Conductivity Monitor QuadTec UV/Vis Detector ph Monitor Non-Bio-Rad Detectors Valve Connections Fraction Collector Connections BioFrac Fraction Collector Model 2110 Fraction Collector Model 2128 Fraction Collector Pump Connections Model EP-1 Econo Pump Econo Gradient Pump (EGP) Model 1327 Chart Recorder Connections Completing System Setup DuoFlow System Network Connections System Power Up Chapter 4.0 System Plumbing General Guidelines for Creating Your Own Tubing Connections Plumbing a DuoFlow System Priming the System iv
5 TABLE OF CONTENTS SECTION 3. SYSTEM OPERATION Chapter 5.0 Introduction to the System Software System Interface Standard Mouse and Keyboard Functions System Menus Toolbar Buttons Drop-down Menus BioLogic Configuration Utility Software Chapter 6.0 Introduction to the Browser Screen Overview Creating and Running a Queue Creating and Viewing a Compare Trace Compare Toolbar Buttons Drop-down Menus Windows Chapter 7.0 Modes of Operation Manual Screen Pressure Limit Settings Setup Screen Protocol Screen Run Screen Pausing/Stopping a Method in Progress Working Offline During a Run Editing a Method During a Run Run Notebook Screen Run Log Screen Post Run Screen Resizing Chromatogram Information (Values at Cursor) Annotating ( tagging ) the Chromatogram Entering Activity Data Exporting Chromatogram Data to other Software Applications Exporting Chromatogram Images SECTION 4. SAMPLE LOADING APPLICATIONS Chapter 8.0 Sample Loading Sample Loading through the Workstation Pump with an SV5-4 Valve System Setup Writing the Protocol Sample Loading using the DynaLoop Sliding-Piston Loop System Setup Writing the Protocol Sample Loading onto a Low Pressure Column Using an Aux Pump System Setup Writing the Protocol Multiple Sample Loading System Setup Writing the Protocol Multiple Injections of a Single Sample System Setup Writing the Protocol v
6 TABLE OF CONTENTS Chapter 9.0 Multiple Columns Column switching System Setup Writing the Protocol Running One Sample Sequentially over Several Different Columns System Setup Protocol Chapter 10.0 Reverse Flow Chromatography System Setup Protocol SECTION 5. MAINTENANCE AND TROUBLESHOOTING Chapter 11.0 Maintenance Care of the Outer Surfaces of the Instruments Storage of the DuoFlow System Care and Maintenance of the Workstation Pumps Priming the Workstation Pumps and Removing Trapped Air Bubbles Daily Maintenance Routine Maintenance of the Workstation Pumps Maintenance of the UV Detector and the Conductivity Flow Cell Cleaning the UV Detector and the Conductivity Flow Cell Replacing the Lamp in the UV Detector Mixers Valves SVT3-2 Diverter Valve AVR7-3 and AVR9-8 Valves Maximizer Valves Chapter 12.0 Troubleshooting DuoFlow Systems Troubleshooting the DuoFlow Controller and Software Troubleshooting the DuoFlow Workstation Pumps Troubleshooting the UV Detector and UV Trace Troubleshooting the Conductivity Flow Cell and Trace Troubleshooting the Maximizer Buffer Blending Troubleshooting Other Bio-Rad Instruments and Devices SECTION 6. APPENDICES Appendix A. Specifications...A-1 Appendix B. Pressure Conversion Table...B-1 Appendix C. Warranty Statement...C-1 Appendix D. Ordering Information...D-1 Index...IN-1 vi
7 TABLE OF CONTENTS LIST OF FIGURES 1-1. BioLogic DuoFlow System Pathfinder DuoFlow System Components USB Bitbus Communicator MX-1 Mixer and Mixer Barrel Extender Maximizer Mixer and Mixer Barrel Extender Assembly of Mixers UV Detector, with Mercury Lamp, 254 & 280 nm Filters, and Conductivity Flow Cell UV Detector, with Zinc Lamp, 214 nm Filter, and Conductivity Flow Cell Conductivity Monitor QuadTec UV/Vis Detector ph Monitor AVR7-3 Sample Inject Valve Sample Load Positions Examples of AVR7-3 Valve Tubing AVR9-8 Stream Select Valve Two Examples Using AVR9-8 Valves SV5-4 Buffer Select Valve Two Examples Using SV5-4 Valves SVT3-2 Diverter Valve Two Configurations of the SVT3-2 Valve BioFrac Fraction Collector Model 2110 Fraction Collector with Optional Dust Cover Model 2128 Fraction Collector DynaLoop Plumbing the DynaLoop for use with an Inject Valve Model EP-1 Econo Pump Connecting an EP-1 Econo Pump to the BioLogic DuoFlow Workstation Example of Large Sample Loading using an Econo Pump Example of Multiple Sample Loading using an Econo Pump EGP Econo Gradient Pump Example of Large Sample Loading using an Econo Gradient Pump (EGP) Example of Multiple Sample Loading using an Econo Gradient Pump (EGP) Rack Assembly Making 1/4-28 Flat Bottom Fittings Backpressure Regulator Signal Import Module (front and rear views) Cable Connections to the Signal Import Module Model 1327 Chart Recorder Example of a DuoFlow Pathfinder System Configuration DuoFlow Pathfinder Setup in Non-Condensing Environment USB Bitbus Communicator Cabling System Cable Connections (without Maximizer) System Cable Connections (with Maximizer) Rack Assembly Mixers UV Detector and Conductivity Monitor QuadTec Detector Instrument Control Module (ICM) ph Monitor SIM Connections DuoFlow Valves System Plumbing using Pre-cut Tubing from Fittings Kit Making 1/4-28 Flat Bottom Fittings System Plumbing with Maximizer Maximizer Plumbing vii
8 TABLE OF CONTENTS 4-5. Plumbing Connections to the Workstation Pump Inject Valve Plumbing for an AVR Backpressure Device Layout of the Screen Display, showing the Manual Screen BioLogic Configuration Utility Software Screen The Browser Screen The CopyIn Window CopyIn with Information in Browser Tab Window Set Browser Options Window Queues Displayed in the Browser Window Compare Displayed in the Browser Window Layout of the Trace Compare Window Relationships between Modes of Operation Manual Screen, showing a Maximizer, BioFrac Fraction Collector, QuadTec Detector, Econo Gradient Pump, and Four Valves Setup Screen Protocol Screen Run Screen showing a Run in Progress Run Screen s Abort, Pause, and Hold Buttons Editing during a Run Protocol Screen during a Run Run Notebook Screen Run Log Screen Post Run Screen Post Run Tags for UV Detector Activity Trace Editor Activity Trace Export Data Setup Screen Exporting a Chromatographic Image Sample Loading through the Workstation Pump Writing a Protocol for Sample Injection through the Workstation Pump Plumbing the DynaLoop for use with an Inject Valve Load/Inject Sample Window for a Manual Load/Inject of the Sample Valve Positions During a Run using the DynaLoop Connecting an EP-1 Econo Pump to the Workstation Plumbing an AVR7-3 Inject Valve with an Auxiliary Load Pump Writing the Protocol: Direct Sample Loading AVR7-3 Valve Positions During a Run with Direct Sample Loading and Injection Using an Auxiliary Load Pump and an AVR9-8 Valve to Load Multiple Samples Load/Inject Sample Window for Loading Multiple Samples Column Switching using Two AVR7-3 Valves Protocol for using an AVR7-3 Valve Inject Valve for Column Switching Column Switching using two AVR9-8 Valves Programming a Sequential Run Injection Valve Plumbing for Use in Affinity Chromatography Programming for Reverse Flow Chromatography Workstation Pump Mechanism Parts Piston Assembly and Access to the Piston Seal Pumphead Assembly Replacing a Mercury Lamp in a Model OM-II UV Detector Replacing a Zinc Lamp in a Model OM-II UV Detector Mixer Assembly SVT3-2 Valve Assembly AVR7-3 and AVR9-8 Valve Assembly Replacing a Maximizer Valve viii
9 TABLE OF CONTENTS LIST OF TABLES 1-1. DuoFlow System Configurations Front View of the Dell PC Computer as the DuoFlow Controller Rear View of the Dell PC Computer as the DuoFlow Controller F10 and F40 Pumphead Flow Rates Workstation Front Panel Controls Workstation Rear Panel Connectors Maximizer Front Panel Controls Maximizer Rear Panel Connectors Maximizer Screens Mixer Barrels and Mixer Capacity for the MX-1 Mixer Mixer Barrels and Mixer Capacity for the Maximizer Mixer BioFrac Racks Available Model 2128 Racks Available Workstation Pump Configuration Flow Rates Columns and Column Fittings Mixer Flow Rates Tubing Guidelines Special Function Keys Toolbar Buttons File Drop-down Menu Edit Drop-down Menu View Drop-down Menu Utilities Drop-down Menu Options Drop-down Menu Window Drop-down Menu Toolbar Buttons File Drop-down Menu Options Drop-down Menu View Drop-down Menu Window Drop-down Menu Valve Setup Information Isocratic Flow Button Load/Inject Sample Button Linear Gradient Button Change Valve Button Column Switching Button Hold Button Miscellaneous Buttons Fraction Collection Button Protocol Screen s Editing Toolbar Buttons Run Screen s Control Buttons B-1. Pressure Conversion...B-1 ix
10 A B VALVE A VALVE B F1 BioFrac Franction Collector F2 F3 F4 F SYSTEM OVERVIEW INTRODUCTION 1.0 INTRODUCTION 1.1 OVERVIEW The BioLogic DuoFlow chromatography system is specifically designed for the high resolution purification of proteins, peptides, and other biomolecules where recovery of biological activity is of primary concern. The DuoFlow F10 pumphead operates at up to 20 ml/min and 3500 psi when used with the Maximizer. The DuoFlow F40 pumphead operates at up to 80 ml/min and 1000 psi when used with the Maximizer. The BioLogic DuoFlow system software provides an easy-to-use graphic interface and menu-driven software for manual operation, system setup, method editing, and run operations. The system software may be run on any PC running Microsoft Windows The flexible control architecture allows the seamless integration of a wide variety of configurations with other Bio-Rad and non-bio-rad components to meet your purification requirements. UNO TM Q1 COLUMN AVR7-3 SAMPLE INJECT VALVE MAXIMIZER MIXER QUADTEC DETECTOR ph MONITOR USB BITBUS COMMUNICATOR CONDUCTIVITY MONITOR WORKSTATION DELL CONTROLLER MAXIMIZER BIOFRAC KEYBOARD MOUSE Figure 1-1. BioLogic DuoFlow System 1-1
11 INTRODUCTION SYSTEM OVERVIEW 1.2 FEATURES BioLogic DuoFlow systems provide the following features: Flexibility of setup. A space saving modular design that is stackable enables configuring the system to meet your exact needs and fit into your desired bench space. Trays and vertical bars are moveable and removable. Horizontal bars can be placed in the optimal position for valves, columns, detectors, etc. The system fits easily into a cold box. Modular components provide an easy upgrade path to fit all applications and financial requirements. For example, the DuoFlow Basic system may be purchased and upgraded to a Pathfinder system as needed. (See Section 1.4 for a description of the DuoFlow systems and upgrades.) Intuitive, user-friendly software programming. Users become experts in only a few runs. Four easy steps to set up new devices/instruments, create a new method, and start to run samples: Step 1. Browser, to enter new user name and method name. Step 2. Setup, to specify decises required for the method. Step 3. Protocol, to enter sequential method steps. Step 4. Run, to inject sample and view real time chromatogram. Two easy steps, if you want to run a sample using a current Setup and Protocol: Step 1. Browser, to select user and method. Step 2. Run, to inject sample and view real time chromatogram. On screen Help. Easily accessed extensive, detailed explanations and troubleshooting guide. USB Bitbus Communicator. Bitbus communication allows users to connect their own USB compatible PC computer running Windows 2000 and the DuoFlow version 4.0 or higher software. Bio-Rad provides a Dell computer running Windows 2000 and the DuoFlow software version 4.0 or higher software with each DuoFlow system. Buffer Blending for ph methods development applications. The Maximizer add-on component provides this additional capability to the DuoFlow system. ph Monitor. The optional in-line BioLogic ph monitor allows direct monitoring of ph conditions during a run and comes with the Maximizer. It has a Calomel sealed electrode with both ph and reference electrodes in a single body. The sealed reference design eliminates the need to add electrolyte solutions and minimizes reference dry out. The electrode is fully compatible with buffers, like Tris, that may not be compatible with Ag/AgCl (silver/silver chloride) containing electrodes. It has a PEEK biocompatible flow cell. Flow Rate Flexibility. The Workstation has pumpheads that permit higher or lower flow rates. The DuoFlow system may be equipped with F10 pumpheads for slower flow rates from 0.01 ml/min, or F40 pumpheads for higher flow rates to 80 ml/min, when used with the Maximizer. Detection Flexibility UV detector with fixed 254 nm and 280 nm filters, long life mercury lamp, and additional drop-in expansion filters available. UV detector can be expanded to 214 nm filter with zinc lamp. QuadTec UV/Vis detector analyzes samples simultaneously at 4 different wavelengths from nm with a deuterium lamp or with a halogen lamp. Third party detectors, such as refractive index or fluorescent, may be utilized with our DuoFlow systems via a SIM or Maximizer. Conductivity Monitor monitors salt concentration to assure reliable gradient formation and pump functions. 1-2
12 SYSTEM OVERVIEW INTRODUCTION Multiple Valve Capabilities Workstation provides connection for 3 low pressure and 3 high pressure valves. Workstation with the addition of the Maximizer doubles the capacity to 6 low pressure and 6 high pressure valves. Starter Kit and UNO Q1 Anion Exchange Column provides the reagents, protein sample, and column. Easy to follow tutorial style instructions enable users to learn how to run a sample under known conditions. IQ/OQ Protocols. Validation protocols are available or can be performed by certified Bio-Rad service engineers. 1.3 UNPACKING When you receive the BioLogic DuoFlow system, carefully inspect the shipping containers for any damage which may have occurred in shipping. Severe damage to a container may indicate damage to its contents. If you suspect damage to the contents, immediately file a claim with the carrier in accordance with their instructions before contacting Bio-Rad Laboratories.! Caution Lift items from the bottom as you remove them from their containers! Open each of the shipping cartons and lift the contents out of its packing. Check the contents of each box against the supplied packing list. Remove the plastic bag from each unit and inspect the unit for external damage. If any part is missing or damaged, contact Bio-Rad Laboratories immediately. Bio-Rad ships DuoFlow systems in a number of different configurations, each with its own catalog number. These systems are described in the following table. Because of its modular design, any of these systems can be upgraded any time simply by adding system options. 1-3
13 INTRODUCTION SYSTEM OVERVIEW 1.4 SYSTEM CONFIGURATIONS The BioLogic DuoFlow is available in the following system configurations. Each system configuration is identified by its name, its standard components and devices, and optional components and devices that may be ordered separately for use with the system. Catalog Number BioLogic DuoFlow System Guide S = Standard U = Upgradable F10 Workstation F10 Pump Kit F40 Workstation F40 Pump Kit BioFrac Fraction Collector QuadTec UV/Vis Detector Maximizer (incl ph) ph Monitor DuoFlow Basic System 100/120 V DuoFlow Basic System Japan and Korea DuoFlow Basic System 220/240 V 10 ml/min flow rate to 3500 psi 254/280 nm detection DuoFlow Standard System 100/120 V DuoFlow Standard System Japan/Korea DuoFlow Standard System 220/240 V 10 ml/min flow rate to 3500 psi 254/280 nm detection Fraction collection DuoFlow QuadTec Basic System 100/120 V DuoFlow QuadTec Basic System Japan/Korea DuoFlow QuadTec Basic System 220/240 V 10 ml/min flow rate to 3500 psi UV/Vis detection with 4 simultaneous wavelengths DuoFlow QuadTec Standard System 100/120 V DuoFlow QuadTec Standard System Japan/Korea DuoFlow QuadTec Standard System 220/240 V 10 ml/min flow rate to 3500 psi UV/Vis detection with 4 simultaneous wavelengths Fraction collection DuoFlow Maximizer 20 System 100/120 V DuoFlow Maximizer 20 System Japan/Korea DuoFlow Maximizer 20 System 220/240 V 20 ml/min flow rate to 1000 psi 254/280 nm detection Buffer blending Fraction collection DuoFlow Maximizer 80 System 100/120V DuoFlow Maximizer 80 System Japan/Korea DuoFlow Maximizer 80 System 220/240 V 80 ml/min flow rate to 1000 psi 254/280 nm detector Buffer blending ph monitoring Fraction collection DuoFlow Pathfinder 20 System 100/120 V DuoFlow Pathfinder 20 System Japan/Korea DuoFlow Pathfinder 20 System 220/240 V 20 ml/min flow rate to 1000 psi UV/Vis detection with 4 simultaneous wavelengths Buffer blending ph monitoring Fraction collection DuoFlow Pathfinder 80 System 100/120 V DuoFlow Pathfinder 80 System Japan/Korea DuoFlow Pathfinder 80 System 220/240 V 80 ml/min flow rate to 1000 psi UV/Vis detection with 4 simultaneous wavelengths Buffer blending ph monitoring Fraction collection S S U U U U U U S S U U S U U U S S U U U S U U S S U U S S U U S S U U S U S S U U S S S U S S S S U U S S S S U U S S S S S S 1-4
14 SYSTEM OVERVIEW INTRODUCTION 1.5 QUICK START PROCEDURE The following describes the general procedure to run a sample with your DuoFlow system. 1. Install the relevant devices and instruments. Refer to figures 3-4 and 5-5 for electrical connections and figures 4-1 and 4-3 for plumbing connections. 2. Flush all plumbing with DDI H 2 O to ensure that the DuoFlow system is clean and free of air. Prime the pumps with your starting buffer or mobile phase. Refer to Chapter 4 for more information. 3. Equilibrate the system and your column with the starting buffer or mobile phase via the Manual screen options (see pg 7-2). Use the Manual screen to set the Workstation pump pressure high and low limit settings. The high limit setting should be the maximum backpressure acceptable for your column. 4. Select Browser from the Tool bar. In the Browser screen you will enter a username and name your new method. Refer to page 6-1 for more information on the Browser screen. Select the Browser icon from the tool bar menu. Select the NEW icon from the upper left side of the Browser screen. Select NEW and enter your desired username in the dialog box. Click on the Project icon for your username. Select NEW and NEW METHOD. Enter your method name (or use default Method 1). Click OK to proceed to the hardware Setup screen. 5. In the Setup screen select the devices and instruments you will use for this method. You may name and save this particular configuration so that it can be recalled to be used with a different method protocol. To do this, select Save Setup under the File menu and enter the desired name. For more information on the Setup screen, refer to page Select Protocol from the tool bar. In the Protocol screen create your method. For more information on this screen, refer to page 7-9. If you are running an ion exchange experiment, you might want to begin your method as follows by selecting from the tool bar icons along the left side of the screen: Select the fraction collector icon on the side bar menu. Select your fraction collection technique (e.g. collect all, threshold, etc.) and the volume of sample you wish to collect. After entering your entire method, return to the fraction collector and you will find the number of tubes required for collection has been calculated. Add an Isocratic flow step and enter a flow rate and the amount of time required to equilibrate your column. The flow rate you select here will automatically appear in any subsequent steps, but can be changed at any time. Refer to page 7-10 for more information. Add a Zero baseline step to zero your selected detector before injecting your sample. Add an Inject sample step to select the sample volume to be injected. If you are using an AVR7-3 inject valve, select static loop. Your options will vary according to the type of injection method you chose when you set up your hardware. Refer to page 7-11 for more information. Add a Linear Gradient step to select buffers or solvents to mix a gradient that will elute your sample. Refer to page 7-13 for mor information. Add an Isocratic flow step with 100% of the final buffer or solvent in the linear gradient. This assures that all sample is removed from the column. Add an Isocratic flow step with 100% of the starting buffer or solvent to ensure that your system is ready to inject the next sample. You may also wish to include a Hold step in your method which will hold your method while the pumps continue working. This is useful if you wish to hold your gradient at some point for a specific amount of time. You may also enter a continue time to stop the hold and continue the method. See Chapter 7 for more information. For more information refer tp page Use the Run screen to run your method. Select the RUN icon from the tool bar menu to bring up the Run screen. For more information on this screen, refer to Chapter
15 INTRODUCTION SYSTEM OVERVIEW 1.6 PROGRAMMING THE MAXIMIZER The Maximizer valve system added to the DuoFlow system enables automated buffer blending protocols, doubles the flow rate of the Workstation pumps, and doubles the number of valves that may be connected to the DuoFlow system. Following are the basic procedures for running a buffer blending method. Refer to pages 2-11; 3-5; 4-3,4; 7-6; for more information regarding installation and maintenance of your Maximizer system. Overview to run a buffer blending method: Manual Screen In the Manual screen select the buffer blending icon from the tool bar and select your desired buffer from the menus of available buffers. Prepare your solutions according to the recipes provided and place them on the correct valve position of the Maximizer acid at A1, base at A2, salt at B1, and water at B2. Assure that the "Use ph correction" box is not checked. In the Maximizer*Pump control panel enter the ph value you are using (Desired), the salt composition in %B and you desired flow rate, press run. After the system has equilibrated collect some effluent and measure the ph (Observe ph at 0% B). These values will be entered into the buffer blender in Setup screen when you run you method. Browser Select a User name or enter a new name. Select -> New -> Method and enter a method name or accept the default of Method.1. Setup Select and enter the relevant information as prompted by the software for the pumps, valves, detectors, etc., in use for the method. Select Buffer Blender and enter the Desired ph and Observed ph values obtained from the Manual run. Assure that Use ph correction, box is checked. Protocol Enter desired method steps. When you select Isocratic Flow and Linear Gradient your chosen buffer system will appear. Load your sample, Start and Run Solution Preparation The predefined buffer recipes provided for use with the Maximizer have been formulated to give optimum accuracy near the pka of the buffer. Buffer solutions should always be made with reagent grade materials and filtered or degassed. ph Measurement and Corrections Maximizer compensates for changes in temperature and ionic strength that affect the ph of a buffer. A sensor in the conductivity flow cell monitors temperature when connected to the Maximizer. Single Point Correction (best for isocratic applications) Accuracy can be improved by applying a single or double point ph correction. 1. Run your buffers in Manual mode and measure the ph of the eluant after the system has calibrated. When selecting buffers from Buffer Blending in the Manual screen assure that the ph correction box is not checked. 2. In the Browser select your desired User and method. 3. In the Setup screen select Buffer Blender and select the buffer in use. 4. Place a check in the "use ph Correction" box and assure that the "use two point correction" option is not selected. 5. Enter the ph value that was measured in step 1 in the "observed at 0% B" box and the desired ph in the "Desired" box and press OK. The software will calculate the correction. 1-6
16 SYSTEM OVERVIEW INTRODUCTION Two Point Correction 1. Follow the same protocol for single point correction as defined in step 1, however set the salt composition to 0% B, press run. 2. After the system has equilibrated collect some low salt effluent to use to measure the ph. 3. Set the salt composition to the maximum to be used in the method. Take the ph electrode out of line as it can take a long time to equilibrate the electrode after exposure to high salt. 4. Press run, after the system has calibrated collect some high salt effluent to be used to measure the ph. 5. Measure the ph of the low and high salt solutions collected in steps 2 and In the Browser screen select your desired User and method. 7. In the Setup screen select Buffer Blender and select the buffer in use. 8. Place a check in the "use ph Correction" box and the "use two point correction" option. 9. Enter the low salt and high salt ph values collected in step 5 in the observed and the %B that was used to collect the high salt ph. 10. Enter the desired ph in "Desired." 11. The software will calculate the correction. High flow rate chromatography without buffer blending An option is available allowing you to run high flow rates, up to 80 ml/min with F40 pumpheads, without doing buffer blending method protocols as follows: 1. In the Browser select User and method you wish to run. 2. In the Setup screen double click on the Buffer Blender icon, choose "High Flow Non-Blending" from the ph recipe list. 3. Prepare the buffer that is placed at valve A1 at a 1X concentration and at the desired ph, plug the A2 port. 4. Prepare the buffer that is placed at valve B2, probably high salt, at 2X concentration and at the desire ph. Place a plug in port B2. 5. Prime the system and start your run. 1-7
17 A B VALVE A VALVE B F1 BioFrac Franction Collector F2 F3 F4 F SYSTEM OVERVIEW DESCRIPTION OF SYSTEM COMPONENTS 2.0 DESCRIPTION OF SYSTEM COMPONENTS The DuoFlow s modular design supports many types of system components, allowing for a wide variety of system configurations. This section discusses in detail the function of each component and its connection to the system. Dell PC Computer/Controller and USB Bitbus Communicator: see Section 2.1. Workstation: see Section 2.2. Maximizer: see Section 2.3. Mixers, which consist of the MX-1 mixer and the Maximizer mixer: see Section 2.4. Detection Systems, which consist of the UV detector, Conductivity monitor, and QuadTec detector: see Section 2.5. Valves, which consist of the AVR7-3, AVR9-8, SV5-4, and SVT3-2 valves: see Section 2.6. Fraction Collectors, which include the BioFrac, Model 2110, and Model 2128: see Section 2.7. Sample Loading Options, including the DynaLoop, EP-1 Econo pump, and EGP Econo Gradient Pump: see Section 2.8. System Peripherals include the SIM, ph monitor, and universal power supply: see Section 2.9. Columns and Column Fittings: see Section UNO TM Q1 COLUMN AVR7-3 SAMPLE INJECT VALVE MAXIMIZER MIXER QUADTEC DETECTOR ph MONITOR USB BITBUS COMMUNICATOR CONDUCTIVITY MONITOR WORKSTATION DELL CONTROLLER MAXIMIZER BIOFRAC KEYBOARD MOUSE Figure 2-1. Pathfinder DuoFlow System Components 2-1
18 DESCRIPTION OF BIOLOGIC DUOFLOW SYSTEM SYSTEM OVERVIEW 2.1 CONTROLLER AND USB BITBUS COMMUNICATOR The DuoFlow systems are controlled by a PC computer, referred to throughout this document as the Controller. The Dell Controller available from Bio-Rad includes a color display monitor, a keyboard, a mouse device, a CD-ROM drive, and a floppy disk drive. The Controller communicates with the workstation and other external devices via its USB connector. These devices include the DuoFlow system components, which communicate via the instrument bus connectors. The USB Bitbus Communicator serves as the link between the Controller s USB connection and the DuoFlow s Instrument bus connections. This link allows the Controller to control the Maximizer and Workstation, as well as any instruments connected to the Maximizer or Workstation, such as automatic valves, UV detector and conductivity monitor, and peripheral instruments such as the BioFrac fraction collector, the QuadTec UV/Vis detector, the Model EP-1 Econo pump, the Econo Gradient Pump (EGP), and Signal Import Modules (SIM) for additional devices Controller The Controller runs BioLogic DuoFlow version 4.0 or higher system control software on Windows 2000 or higher operating system. From the Controller you can set up and run methods, perform simple data analysis, and store method and run data. The following tables show the key features on the Dell computer provided by Bio-Rad. Table 2-1. Front View of the Dell PC Computer as the DuoFlow Controller COLOR DISPLAY MONITOR POWER SWITCH USB CONNECTORS (2) (BEHIND COVER PANEL) CD-ROM DRIVE FLOPPY DISK DRIVE Feature Description Power Switch Turns on/off the Controller and monitor. Floppy Disk Drive To backup methods to - and restore methods from - a floppy disk. Press the button next to the drive slot to manually eject a floppy disk from the drive. 2-2
19 SYSTEM OVERVIEW DESCRIPTION OF BIOLOGIC DUOFLOW SYSTEM Table 2-1. (continued) Front View of a Dell PC Computer as the DuoFlow Controller Feature Description CD ROM Drive Keyboard, Mouse, & Function Keys F2 F1 Esc To load updates of the BioLogic DuoFlow operating software. To open the drive, press the button on the front of the drive. The keyboard and mouse interface devices control the system. They are standard PC compatible input devices. The keyboard includes the following special function keys: Hold until Keypress: To start a method that is on Hold during a run when a method includes a Hold until Keypress step. Help: Displays the Help menu for the currently displayed screen. Esc: Functions as an alternative to the Cancel selection in a Dialog box. Alt USB Connectors Alt: Some system commands can be executed either by selecting them from a drop-down menu or by holding down the Alt key and then pressing the appropriate character key. These connect to the USB Bitbus Communicator, which in turn connects to the Instrument bus, thereby allowing components in the BioLogic DuoFlow system to communicate with the Controller. Components connect to the Instrument bus in a daisy-chain. The presence of a component on the Instrument bus is recognized when it is switched on. Even when one component is switched off, another component daisy-chained to it can be controlled by the Controller. 2-3
20 DESCRIPTION OF BIOLOGIC DUOFLOW SYSTEM SYSTEM OVERVIEW Table 2-2. Rear View of the Dell PC Computer as the DuoFlow Controller MONITOR POWER CONNECTOR MOUSE CONNECTOR KEYBOARD CONNECTOR PARALLEL CONNECTOR DELL CONTROLLER MONITOR SIGNAL CABLE ETHERNET CONNECTOR COLOR MONITOR CONNECTOR USB BUS CONNECTORS CONTROLLER POWER CONNECTOR Connector Description USB connectors: To connect to the USB Bitbus Communicator, which connects to the Instrument bus, enabling components in the BioLogic DuoFlow system to communicate with the Controller. Components connect to the Instrument bus cable in a daisy-chain. The presence of a component on the Instrument bus is recognized when it is switched on. Even when one component is switched off, another component daisy-chained to it can be controlled by the Controller. There are two USB ports behind a door on the front of the Controller. Monitor connector: To connect the color monitor to the Controller. Keyboard connector: To connect the keyboard to the Controller. Mouse connector: To connect the mouse to the Controller. Parallel connector: Devices designed for connection to the parallel port include printers and external storage devices. Refer to Windows 2000 and/or the device documentation for installation instructions. (Some printer drivers are pre-installed on the Controller.) Power connectors: To connect the power cable. 2-4
A B VALVE A VALVE B F1 BioFrac Franction Collector F2 F3 F4 F5 1 2 3 4 5 6 7 8 9 0 BioLogic DuoFlow Chromatography System Starter Kit Instruction Manual Catalog # 760-0135 B L D F Table of Contents Introduction...1
www.bio-rad.com/chromsystems Bio-Rad offers a complete line of laboratory-scale chromatography instruments that are flexible, upgradable, and easy to use. These instruments are specifically designed for
Waters Corporation Waters 2690/5 USER & TROUBLESHOOTING GUIDE Contents 2690/5 Theory Setup procedures. Troubleshooting the 2690/5 User maintenance of the 2690/5 Spare Parts 2 2690/5 Theory 2690/5 Solvent
Chromatography BioLogic LP System Low-Pressure Liquid Chromatography BioLogic LP System Low Pressure Liquid Chromatography The BioLogic LP system is a valuable asset for biomolecule purification. Protein
ChromQuest 5.0 Chromatography Data System User Guide for the Surveyor LC CHROM-97257 Revision A March 2008 2008 Thermo Fisher Scientific Inc. All rights reserved. Surveyor and ChromQuest are trademarks
ServersCheck Monitoring Appliance Quick Overview This document is intended to make you familiar with the ServersCheck Monitoring Appliance Although it is possible, we highly recommend not to install other
USB 2.0 LAPTOP DOCKING STATION WITH VIDEO 410-1864-001B / ACP51US 1 YEAR LIMITED WARRANTY: We pride ourselves on the quality of our products. For complete warranty details and a list of our worldwide offices,
Ethernet Radio Configuration Guide for Gateway, Endpoint, and Repeater Radio Units April 20, 2015 Customer Service 1-866-294-5847 Baseline Inc. www.baselinesystems.com Phone 208-323-1634 FAX 208-323-1834
User and installation manual aquaero 5 The information contained in this manual is subject to change without prior notice. All rights reserved. Current as of April 2011 ENGLISH: PAGE 1 DEUTSCH: SEITE 13
SA-9600 Surface Area Software Manual Version 4.0 Introduction The operation and data Presentation of the SA-9600 Surface Area analyzer is performed using a Microsoft Windows based software package. The
AZI P/N 700-0042-B Last update May 2009 USER MANUAL JEROME COMMUNICATIONS SOFTWARE (JCS) INSTALLATION and OPERATION MANUAL V1.02 for Windows XP May 2009 ARIZONA INSTRUMENT LLC 3375 N Delaware Street Chandler,
GE Healthcare Life Sciences UNICORN 5.31 Administration and Technical Manual Table of Contents Table of Contents 1 Network setup... 1.1 Network terms and concepts... 1.2 Network environment... 1.3 UNICORN
Quick Start Guide NVR DS-7104NI-SL/W NVR NOTE: For more detailed information, refer to the User s Manual on the CD-ROM. You must use your PC or MAC to access the files. www.hikvision.com Quick Start 1.
reflecta Super 8 Scanner User Manual 1 FEDERAL COMMUNICATIONS COMMISSION (FCC) STATEMENT This Equipment has been tested and found to comply with the limits for a class B digital device, pursuant to Part
Technical Note Overview The uses sensors to measure and display the concentrations of oxygen and carbon dioxide in a sample as the percentage of a gas in the sample by volume. This method of expressing
FCC COMPLIANCE STATEMENT FOR AMERICAN USERS This equipment has been tested and found to comply with the limits for a CLASS A digital device, pursuant to Part 15 of the FCC Rules. These limits are designed
ServersCheck Monitoring Appliance Quick Overview This document is intended to make you familiar with the ServersCheck Monitoring Appliance Although it is possible, we highly recommend not to install other
Computech Systems, Inc. 301-884-5712 30071 Business Center Dr. Charlotte Hall, MD 20622 RACEAIR REMOTE PAGER SYSTEM TM Introduction: Computech s RaceAir Remote Competition Weather Station with the Data
TR-3 Channel Editor Software Manual Trilithic Company Profile Trilithic is a privately held manufacturer founded in 1986 as an engineering and assembly company that built and designed customer-directed
Wireless Internet Keyboard & Optical Mouse User s Manual Model KB990W FCC Statement This device complies with part 15 of FCC Rules. Operation is subject to the following two conditions: (1) this device
NEWTECH PRO 1-2-3 KVA ONLINE UNINTERRUPTIBLE POWER SUPPLY USER MANUAL email@example.com / www.tuncmatik.com Table of Contents 1. Important Safety Warning 2 1-1. Transportation 2 1-2. Preparation 2 1-3.
Mercury Helios 2 ASSEMBLY MANUAL & USER GUIDE TABLE OF CONTENTS INTRODUCTION...1 1.1 MINIMUM SYSTEM REQUIREMENTS 1.1.1 Apple Mac Requirements 1.1.2 PC Requirements 1.1.3 Supported PCIe Cards 1.2 PACKAGE
DMX USB PRO User Manual www.enttec.com Firmware V1.43 February 2007 Package Contents Your DMX USB PRO package should contain these items: DMX USB PRO (Part No. 70304) Driver for Windows software on the
1.0 Key Features Addonics T E C H N O L O G I E S NAS Adapter Model: NASU2 User Manual Convert any USB 2.0 / 1.1 mass storage device into a Network Attached Storage device Great for adding Addonics Storage
Wave/PC Interactive System USB Adapter Kit Installation Guide Safety Information 1. Read these instructions for all components before using this product. 2. Keep these instructions for future reference.
Installing the Printer Software Checking the Package Contents Connecting the Printer Unpacking the Printer Installing the Ink Cartridges Plugging in the Printer Attaching the Printer Parts 4032216-00 EUL
OxySense Gen III Operations Manual OxySense GEN III 300 Series OxySense GEN III 5000 Series COMPLIES WITH ASTM 2714-08 OxySense, Inc. 13111 North Central Expressway, Suite 440 Dallas, Texas 75243 Phone
USB 2.0 Peripheral Switch USER MANUAL US221A / US421A FCC Information This equipment has been tested and found to comply with the limits for a Class B digital device, pursuant to Part 15 of the FCC Rules.
Agilent Automated Card Extraction Dried Blood Spot LC/MS System SCAP DBS Software User Guide Notices Agilent Technologies, Inc. 2012 No part of this manual may be reproduced in any form or by any means
RFID Logger Software User Manual Rev 1.0 Table of Contents 1. System Overview... 2 2. Installation... 2 3. USB RFID Interface Box... 3 4. Operation... 4 4.1. Start Up... 4 4.2. Menu overview... 4 4.2.1.
SoftRAID 5 QUICK START GUIDE for OWC ThunderBay TABLE OF CONTENTS INTRODUCTION...1 1.1 MINIMUM SYSTEM REQUIREMENTS 1.2 FEATURES 1.3 ABOUT THIS MANUAL SYSTEM SETUP...2 2.1 GETTING STARTED 2.2 INITIALIZING,
Transmitter Interface Program Operational Manual Version 3.0.4 1 Overview The transmitter interface software allows you to adjust configuration settings of your Max solid state transmitters. The following
R Installation Guide APA-1460 SlimSCSI PCMCIA-to-SCSI Host Adapter Introduction This document explains how to install and use Adaptec s APA -1460 SlimSCSI PCMCIA-to-SCSI adapters. The SlimSCSI adapters
SecureLinx Spider Duo Quick Start Guide SecureLinx Spider Duo Quick Start Guide SecureLinx Spider Duo QUICK START GUIDE CONTENTS Overview... 2 What s In The Box... 3 Installation and Network Settings...
How to read this guide The following shows the symbols used in this Quick start guide with descriptions and examples. Symbol Description Example P oint Reference Caution [ ] This symbol explains information
Notice to Users 2003 2Wire, Inc. All rights reserved. This manual in whole or in part, may not be reproduced, translated, or reduced to any machine-readable form without prior written approval. 2WIRE PROVIDES
PU-USBX USB over Ethernet Extender OPERATION MANUAL Safety Precautions Please read all instructions before attempting to unpack or install or operate this equipment, and before connecting the power supply.
Machine type: 10087/3110 Lenovo IdeaCentre Q180 Series User Guide Version 1.0 2011.09 31500665 Important Safety Information Before using this manual, it is important that you read and understand all of
XPanel V2 Remote Control Panel User Manual XILICA Audio Design Important Safety Instructions 1. READ THESE INSTRUCTIONS All the safety and operating instructions should be read before the product is operated.
GV-Keyboard User's Manual V2.0 Before attempting to connect or operate this product, please read these instructions carefully and save this manual for future use. 2007 GeoVision, Inc. All rights reserved.
Wireless Asset Monitoring - Local Food Safety Software BA/WAM-L-F Software Installation and User Guide System Overview The BAPI Wireless Asset Monitoring Local (WAM Local) Software receives temperature
2100 POS System 2100 User Guide Thank you for selecting UTC RETAIL s innovative Model 2100 Point of Sale solution! This guide is designed to acquaint you with the features and functionality of the 2100
Package Content DVR-4TL/8TL/16TL Setup Guide Inspect the DVR s packaging. Make sure that the DVR-4TL/8TL/16TL is packed properly. Remove all items from the box and make sure it contains the following items.
User Manual V2.1 2007.7.3 C o n t e n t s 1. Introduction... 3 1.1 Product Line... 4 1.2 Main Features... 5 1.3 Package contents... 6 1.4 Cables Diagrams... 6 1.4.1 3-in-1 DB15 Cable...6 1.4.2 3-in-1 CAT5
mysensors Information to Users mysensors Wireless Sensors and Ethernet Gateway Quick Start Guide This equipment has been tested and found to comply with the limits for a Class B digital devices, pursuant
SNMP Web card User s Manual Management Software for Uninterruptible Power Supply Systems Table of Contents 1. Overview... 3 1.1 Introduction... 3 1.2 Features... 3 1.3 Overlook... 3 1.4 Installation and
Getting Started This Quick User Guide helps you get started with the IRIScan TM Anywhere Wifi. Please read this guide before operating this scanner and its software. All information is subject to change
User Manual / Installation Guide AV Over Ethernet Gigabit Adaptor Model No. ZE6000 Warning! It will cause malfunction if the AV Adaptor is operating with unspecified power supply adaptor or incorrect power
POS 208 / 808 DISTRIBUTED BY INTELLIGENT CASH DRAWER WEDGES MADE IN CANADA USER S MANUAL 2 Table of Contents Introduction............................................... 3 POS 208 Installation........................................
Evolution Digital HD Set-Top Box Important Safety Instructions 1. Read these instructions. 2. Keep these instructions. 3. Heed all warnings. 4. Follow all instructions. 5. Do not use this apparatus near
Evolution Digital HD Set-Top Box Important Safety Instructions 1. Read these instructions. 2. Keep these instructions. 3. Heed all warnings. 4. Follow all instructions. 5. Do not use this apparatus near
Computer Lane Monitor for Windows User Guide Supplement BRUNSWICK SERVICE PARTS & SUPPLIES MUSKEGON, MICHIGAN 1995, Brunswick Bowling and Billiards Corporation, 525 W. Laketon Avenue, P.O. Box 329, Muskegon,
Rubbermaid Medical Solutions Cart Interface Installation & Control Board Manual for M38 XP and RX Carts and Legacy M39 Carts REVISED FEBRUARY 25, 2010 Filename: Cart Interface Installation & Control Board
Networking General networking Networking overview A network is a collection of devices such as computers, printers, Ethernet hubs, wireless access points, and routers connected together for communication
OWNERS MANUAL Status Monitor for Windows 95, 98, ME, NT 4, 2000 & XP SIGNALCRAFTERS TECH, INC. www.signalcrafters.com 57 Eagle Rock Avenue, East Hanover, NJ 07936 Tel: 973-781-0880 or 800-523-5815 Fax:
Enterprise Level Digital Intercom and Management System USER GUIDE DC3400UG-02 February 2005 Table of Contents 1 OVERVIEW...3 1.1 Hardware Components...3 1.1.1 3406 Gateway Card...3 1.1.2 3401 Dual Station
IP Box Camera ACM-5711 Ver. 081016 Hardware User s Manual 0 0 PRECAUTIONS 1. Read these instructions All the safety and operating instructions should be read before the product is operated. 2. Heed all
Wireless Optical Mouse User s Manual Model PD950P FCC STATEMENT PD950P This device complies with part 15 of FCC Rules. Operation is subject to the following two conditions: (1) this device may not cause
TROUBLESHOOTING GUIDE HPLC Before starting any troubleshooting, whether it is related to instruments or columns, it is essential that safe laboratory practices be observed. The chemical and physical properties
3 Introducing Windows CE Windows CE Operating System Windows CE Services Software Applications Online Help Introducing Windows CE 3-1 WINDOWS CE OPERATING SYSTEM Windows CE is the smallest member of the
Firmware version: 1.10 Issue: 7 AUTODIALER GD30.2 Instruction Manual Firmware version: 2.0.1 Issue: 0.6 Version of the GPRS transmitters configurator: 188.8.131.52 Date of issue: 07.03.2012 TABLE OF CONTENTS
WUA-0605 300Mbps Wireless USB Network Adapter User Manual V1.0 Certification FCC CE FCC Statement This equipment has been tested and found to comply with the limits for a Class B digital device, pursuant
Power Log PC Application Software Users Manual May 2007 Rev. 1, 5/08 2007-2008 Fluke Corporation. All rights reserved. Specifications are subject to change without notice. All product names are trademarks
s Contents Product Overview 1 Installation 2 Working with the Automation License Manager 3 Glossary Manual Index 12/2008 A5E02389428-01 Legal information Warning notice system This manual contains notices
User Guide HUAWEI UML397 Welcome to HUAWEI 1 Huawei UML397 4G LTE Wireless Modem User Manual CHAPTER 1 BEFORE USING YOUR UML397 WIRELESS MODEM ABOUT THIS USER MANUAL...4 WHAT S INSIDE THE PRODUCT PACKAGE...4
SwitchView 1000 Switch Installer/User Guide SwitchView 1000 Switch Installer/User Guide Avocent, the Avocent logo, The Power of Being There and SwitchView are registered trademarks of Avocent Corporation.
2800 Series POS System Installation Guide Congratulations on your purchase of UTC RETAIL s innovative 2800 Series POS System! This fully integrated point-of-sale hardware solution, with its modular design
Congratulations on your purchase of the GIGABYTE Notebook! This Manual will help you to get started with setting up your notebook. For more detailed information, please visit our website at http://www.gigabyte.com.
Business Audio System: Music & Messaging MP3 Player by Grace Digital Audio User Guide Model No. GDI-USBM10 User Guide Contents Introduction 2 Safety & General Use Information 2 Features 3 Set Up & Operation
user manual chromatography software Administration and Technical Manual um 03-0014-93 UNICORN 5.0 Administration and Technical Manual 03-0014-93 Edition AC 2004-03 Office addresses: Amersham Biosciences
OMBEA Response User Guide ver. 1.4.0 OMBEA Response User Guide Thank you for choosing an Audience Response System from OMBEA. Please visit www.ombea.com to get the latest updates for your system. Copyright
Wireless Optical Stow-N-Go TM Notebook Mouse USER GUIDE Visit our Website at www.targus.com Features and specifications subject to change without notice. 2008 Targus Group International, Inc. and Targus,
NETWORK PRINT MONITOR User Guide Legal Notes Unauthorized reproduction of all or part of this guide is prohibited. The information in this guide is subject to change without notice. We cannot be held liable
DCT-16 USB HD Audio Converter Operation Manual DISCLAIMERS The information in this manual has been carefully checked and is believed to be accurate. Cypress Technology assumes no responsibility for any
Optilab rex User s Guide (M1500 Rev. D) WYATT TECHNOLOGY Corporation makes no warranties, either express or implied, regarding this instrument, computer software package, its merchantability or its fitness
Kramer Electronics, Ltd. Site-CTRL and Web Access Online User Guide (Documentation Revision 2) Software Version 1.9. 2902 To check that you have the latest version, go to the DOWNLOADS section of our Web
Your consent to our cookies if you continue to use this website.