1 What does your IV Smart Pump Infusion Data tell you about IV Clinical Practice and Patient Safety? Tim Hoh, BScPhm, John Llaguno, RN, NMD, PhD, Nadia Mamer, BSc., MBA, Mauricio Rodriguez BASc., MBA., PMP., Elizabeth Starenkyj, RN BScN., Department of Clinical Informatics, Baxter Healthcare, Canada Paula Beard, ACP, Canadian Patient Safety Institute Bonnie Bowes, RN., BN., CCNC, Heather Sherrard RN., Timothy J. Zakutney, MHSc., PEng. University of Ottawa Heart Institute, Canada.
2 Abstract SMART pumps can provide retrospective data which, when interpreted in collaboration with clinician users, can provide valuable insight into IV medication practices, compliance with Dose Error Reduction Software (DERS), resource utilization of infusion pumps, and intercepted medication errors and can assist in the promotion of a culture of patient safety. An evaluation of the value the Colleague Infusion Pump with GUARDIAN DERS brings to clinical practice at the University of Ottawa Heart Institute (UOHI) was undertaken through the quantitative analysis of infusion pump data. Interpretation of these reports in collaboration with nurses assisted in identifying various issues impacting best practice. Targeted approaches to practice changes, protocol redesign and further education were identified.
3 Background/Rationale Adverse events (AEs) result in 9,000 24,000 preventable deaths in Canada each year (Baker, Norton, et.al, 2004). Although surgical procedures were found to be the most common type of AE in the Canadian Adverse Events Study, the second most common AEs were associated with drug- or fluid-related events. Adverse drug events (ADE) specifically focus on the delivery of medication to patients and represent a significant vulnerability for patients in hospitalized settings. In addition, ADEs place a financial burden on the healthcare system through an increased length of stay and the implementation of treatment plans to mitigate the injury (Classen & Pestonik, 1997; Rodriguez- Monguio, Otero & Rovira, 2003). In 2004, Health Canada issued a Notice to Hospitals warning of the risks associated with intravenous infusion pumps. Between 1987 and 2003 Health Canada received reports of 425 separate incidents involving infusion pumps. Of these 425 incidents, 23 resulted in deaths, 135 resulted in injury and 127 could have potentially led to injury or death. Of the 23 deaths, 20 were suspected to be the result of pump technology or user error. All 135 injuries were attributed to pump technology or user error (Health Canada, 2004). As a result of these findings, Health Canada recommended a number of safety steps to be implemented in healthcare to protect against infusion device medication errors including the recommendation that hospitals use infusion devices with safety features such as software that provide safeguards against dosing and infusion related errors. These devices are commonly known as SMART Pumps (Wilson & Sullivan, 2004). Upper and lower dosing limits are developed in conjunction
4 with hospital clinical and pharmaceutical protocols and programmed into the pump. Should a clinician program an infusion medication dose outside of those limits considered safe, the pump will alert the clinician to this potential error. The clinician may then choose to accept their choice (since occasionally in clinical practice scenarios the dosing limits must be overridden) or they may recognize a programming error and cancel the dose before the infusion has commenced. The Canadian Adverse Events Study suggested that as many as 37 to 51% of adverse events (AE) in healthcare are preventable (Baker et.al, 2004).. Other studies have shown preventable AE rates as high as 51% (Brennan, Leape, et al, 1991). Although most patients who experienced and AE recovered, many experienced an increased length of stay in hospital or temporary disability and a number of them died as a result of their injury. Costs have been estimated at approximately $8, per ADE in the US (Rodriguez-Monguio, Otero, Rovira, 2003; Brennan, Leape, et al, 1991; Malashock, Gould & Shull, 2004; Bates, Spell, et. al, 1997) and this estimate does not include the additional costs associated with litigation, malpractice insurance premiums and injury related costs such as lost income or family hardship. It is reasonable to suggest that acuity levels in Canada are rising as the population ages. It is also reasonable to suggest that complex infusion therapy usage will also rise creating a greater potential for infusion errors (Rex, Turnbull, Allen, Vande Voorde & Luther, 2000). Nursing workforce shortages place additional burdens on clinicians who
5 must do more with less, increasing the potential for human error (Milligan, 2003; Wilkins & Shields, 2008; Reason, 2000). Increasingly, those responsible for the delivery of healthcare services understand that individual clinicians are rarely singularly responsible for an adverse event rather, recognizing that the clinical environment is complex and many factors can contribute to errors ( Balas, Scott & Rogers, 2004).. Within the context of a just culture environment, health care providers have been shown to be a reliable source of information about the nature and conditions giving rise to medication errors (Reason, 2000, Neale, Woloshynowych & Vincent, 2001). Escalating healthcare costs in Canada result in hospital administrators having to make difficult decisions about resource allocation and infrastructure upgrading. Efficient use of current technology allows limited funds to be redirected towards more pressing shortfalls. SMART pumps have been shown to intercept potentially costly medication errors, thereby demonstrating their usefulness (Classen et al, 1997). One such device, Baxter s Colleague CXE with GUARDIAN Dose Error Reduction Software (DERS), was implemented throughout the University of Ottawa Heart Institute (UOHI), in Ottawa, Ontario, Canada. The University of Ottawa Heart Institute (UOHI) is Canada s largest and a highly reputable cardiovascular health centre dedicated to understanding, treating and preventing heart disease. The UOHI has three clinical floors (116 beds) offering specialized cardiac care, including cardiac life support, a 90,000- square foot Research Centre, a Prevention and Rehabilitation Centre, a Diagnostic Centre, and the Cardiac Reference Centre. The Institute serves more than 1.5 million residents from Eastern and Northern Ontario, and Western Quebec, and has the largest artificial heart program in Canada (UOHI, n.d.).
6 his paper provides an overview of the efficacy of Baxter s Colleague CXE with TGUARDIAN DERS in the clinical setting of UOHI. Methodology In August, 2007, data from 199 Baxter Colleague CXE infusion pumps at the UOHI was downloaded, representing 73.4 hours, or approximately 3 days, of infusion pump activity. Several quantitative reports were generated from this data including: 1. Unit specific medication error intercepts 2. Medication error intercepts and associated time of day 3. Unit specific data of infusion starts using dose error reduction software 4. Management of high risk infusion medications 5. Infusion pump battery management 6. Asset (infusion pump) utilization These reports were presented to UOHI Nurse Educators, Unit Managers and Biomedical Engineering Services Department members for qualitative feedback. Individual interviews were conducted and recorded. Additionally, 3 focus groups with end users were conducted in December 2007 and audio-recorded. A total of 52 users participated in the focus groups. The reports were presented without interpretation, and users were asked to interpret the findings and comment on their relevancy to clinical
7 practice and error prevention. Field notes were taken. Audio tapes were transcribed and coded to generate themes. Correlation between themes generated by focus groups, individual interviews and group presentation responses was sought. Results he download of data from the 199 infusion pumps translates to approximately 3 Tdays of typical pump activity. The COLLEAGUE GUARDIAN DERS feature allows clinicians to administer intravenous infusions within the institution s pre-defined safety limits. These safety limits are determined in collaboration with Pharmacy, Medicine and Nursing and programmed into the GUARDIAN drug library. If the clinician programs a drug infusion outside of the defined limits, an alert message provides the user with two options that are defined as follows: a) Accept the dose that was entered (OVERRIDE) b) Cancel the dose that was entered and re-enter the correct dose (NEAR MISS) During the three-day time frame, 14 medication errors were intercepted through use of GUARDIAN. Figure 1 demonstrates the drug type and dosage of medication errors that were intercepted.
8 LOW DOSE HIGH DOSE INCORRECT DOSE DRUG CORRECTED DOSE LIMIT LIMIT ENTERED Heparin units/250 ml 100 units/hr 1500 units/hr 14,000 units/hr 1400 units/hr 70,000 units/hr 700 units/hr 12 units/hr 1200 units/hr 11 units/hr 1100 units/hr 1600 units/hr 1500 units/hr 2000 units/hr 500 units/hr Nitroglycerin 50 mg/250 ml CCU 15 mcg/min 100 mcg/min 1000 mcg/min 100 mcg/min 1000 mcg/min 100 mcg/min 10 mcg/min 100 mcg/min 333 mcg/min 50 mcg/min Nitroglycerin 50 mg/250 ml CSU 0.5 mcg/kg/min 4.9 mcg/kg/min mcg/kg/min 0.25 mcg/kg/min Norepinephrine 8 mg/250 ml mcg/kg/min 1 mcg/kg/min 2.22 mcg/kg/min mcg/kg/min mcg/kg/min mcg/kg/min Vasopressin 100 units/100 ml 0.5 units/hr 2.4 units/hr 0.1 units/hr changed drug to nitroprusside Figure 1: Drug Type and Dosage of Intercepted Errors his data was further stratified by time of day. Figure 2 demonstrates the Tchronogram illustrating specific time of day when errors were intercepted. 18:00H 21:00H 24:00H :00H 06:00H 15:00H 09:00H 12:00H Figure 2: Chronogram of Intercepted Errors
9 Interpreting this data in conjunction with clinician users allowed for the examination of clinical practice and other routine duties. For example, nurses commented that 09:00H was the time when they were typically occupied with reporting activities and on rounds with Physicians, transcribing medications, etc. 11:30H was the time when the first group of nurses generally take their lunch break (and the remaining nurses are particularly short staffed, busy and hungry!) and 19:00H was evening change of shift time for 12-hour shifts. These are all rushing times.you know 9:00, we re putting out our pills, orders are being done, 11:30 well, here we are, we re all rushing to go to first break.and 7:30 in the evening is change of shift and that can be really hectic with people coming and going. Clinicians were particularly surprised at the strong clustering of intercepted errors at these times. Although they could quickly account for the intercepted errors in terms of what activities were taking place in the clinical environment, they remained concerned that these times were resulting in such striking patient vulnerability. We re surprised at what we re seeing.people are tired stuff happens, but the feature is working as it s intended to you can trust it you don t need to check the higher and lower limits. The GUARDIAN DERS captured 67 Overrides in the three-day time frame. The most commonly overridden drugs are illustrated in Figure 3.
10 Dobutamine 250 mg/ 250 ml Eptif ibatide 75 mg/ 100 ml Heparin units/ 250 ml Isoproterenol 1 mg/ 250 ml Nitroglycerin 50 mg/ 250 ml-ccu Nitroglycerin 50 mg/ 250 ml-csu Norepinephrine 8 mg/ 250 ml Phenylephrine 10 mg/ 250 ml Total Figure 3: Most Commonly Overridden Drugs This information can be used to determine appropriateness of pre-set hospital protocols in light of current practice, or alternatively, can be used to examine current practice in light of safe medication management goals. We are always overriding ni-pride it gets annoying for the patient the limits don t seem to be working for us. It is interesting to note in this particular example that despite the perception that We are always overriding ni-pride in fact, nitroprusside was not a commonly overridden drug in these 3 days. The quantitative data may also assist in supporting or refuting commonly held perceptions. Regardless, the collaboration between nursing, pharmacy, medicine and industry suppliers to develop drug libraries that support best clinical practice can only serve to improve patient safety.
11 In the three-day time frame, the majority of infusion starts were in rate-volume (that is, not started within the GUARDIAN DERS) Figure Guardian Infusion Starts Dose Mode Starts Rate Volume Starts Figure 4: Infusion Start Methods The data downloaded provided unit specific information regarding clinical practice and usage of the GUARDIAN safety system. This information, interpreted in conjunction with clinical practice common to specific units, allowed for evaluation of the appropriate level of GUARDIAN usage per unit and assisted in the identification of gaps in education. Clinical Nurse Educators commented that the quantitative data provided by Baxter s Clinical Informatics program had given them the evidence they needed to target education for practice change within specific units. Figure 5 depicts the breakdown of infusion start methods per unit. Hospital protocol dictates that high-risk medications should always be started within GUARDIAN. Quantitative data combined with clinical observation suggested that this was not always the case. Nurses confirmed this in the focus groups.
12 I ve known some people that once they ve hit the rate sometimes they say Forget It get out of it and just use rate volume because they don t have time to fuss. It s not OK.it s asking for a mistake. Especially for heparin! Department Guardian Starts Dose Mode Starts Rate Volume Starts BED BAY UNIT CARDIAC CATH LAB CARDIAC CT SCAN CARDIAC DAY UNIT CARDIAC SURGICAL - H CARDIOLOGY - H CARDIOLOGY - H CCU CSICU OR-PACU PET-LAB 1 4 REFERENCE CENTRE OTHER* TOTAL Figure 5: Infusion Method Starts by Unit Other = pumps that were located at a neighbouring hospital The data provided by the Clinical Informatics download also provided insight into the management of infusion pump batteries. Figure 6 depicts the breakdown of discharged batteries by unit. Discharged batteries are less capable of maintaining their charge for the full capacity of the battery, thereby posing the risk of unanticipated cessation of infusion therapy to the patient. Fully charged batteries can be counted on to maintain infusion therapy during those times when the pump must run on battery, such as during some diagnostic tests or transport. Nursing users identified that simply plugging
13 the pump in whenever possible can assist in maintaining the health (and safety) of infusion pump batteries. Highlighting areas with poor battery management practices allows for targeted education as well as additional review of biomedical maintenance programs. In some cases, there are insufficient and/or inaccessible plugs and this, too can be addressed with the specific data provided. Sometimes you worry that you ll injure your back trying to get at those plugs.and then we don t want our patients opening up their sternotomy reaching under their bed for a plug! Department Total Pumps Discharged Batteries % Discharge BED BAY UNIT % CARDIAC CATH LAB % CARDIAC CT SCAN % CARDIAC DAY UNIT % CARDIAC SURGICAL - H % CARDIOLOGY - H % CARDIOLOGY - H % CCU % CSICU % OR & PACU % PET LAB % REFERENCE CENTRE % OTHER* % TOTAL % Figure 6: Breakdown of Discharged Batteries by Unit Finally, the reports provided information about infusion device utilization. When determining the number of infusion devices required by an institution, administrators
14 must find the balance between providing sufficient devices for periods of high usage and ensuring that scarce healthcare dollars are not spent on unnecessary equipment. Figure 7 depicts UOHI s infusion pump utilization. Of note, one of the days on which data was collected was a statutory holiday with lower patient volumes. Staff expressed satisfaction that sufficient pumps were available during higher usage times, thereby enhancing the provision of safer patient care. This information further provides clinicians and administrators with the opportunity to review their mix of both single and triple channel pumps, in light of their patient profile and demands for infusion therapy. 92% 50% 72% 80% 57% 55% 62% 73% 60 Pumps 54 Pumps 96 Pumps 69 Pumps 68 Pumps 41 Pumps 74 Pumps 55 Pumps August 7, 2007 August 8, 2007 Augus t 9, 2007 Avg. Total Single CH = 120 Total Triple CH = 75 Figure 7: UOHI s Infusion Pump Utilization The infrastructure required to support this technology is readily available in hospitals in Canada. A number of SMART pumps require wireless connectivity. However, older physical structures, remote areas and/or shrinking budgets often result in wireless connectivity not being an option and/or a priority in terms of spending. As a
15 result the case could be made for utilizing SMART pumps that do not require wireless connectivity in these instances. Timothy Zakutney, UOHI s Manager of Biomedical Engineering, suggested, Wireless connectivity to infusion pumps may be a great benefit to facilitate updating pharmacy profiles in infusion pumps but should not be treat3ed as a prerequisite to implementing SMART pumps. The value of this type of protective software in terms of reducing patient risk and improving patient care outweigh the added value of wireless connectivity. Conclusion Medication delivery errors add to the challenges of delivering healthcare in Canada. Since as many as 51% of AE in healthcare are considered potentially preventable (1), it is incumbent upon healthcare providers and administrators to investigate methods of reducing these avoidable errors. One approach involves collaboration with industry providers to implement technology that can contribute to a safer Canadian healthcare environment. SMART pumps are an easily implemented infusion system proven to intercept medication errors before they can cause harm to the patient. In the 3 day time frame evaluated at UOHI, 14 medication errors were intercepted. If the average cost per ADE is estimated to be $8, (6), one can easily see how a SMART pump system might quickly provide return on investment.
16 Initial evaluation by clinicians using the system indicates that clinicians recognize the value this system brings to the clinical setting. Since the reports generated do not specify which individual clinician is responsible for which intercepted error, clinicians felt that the Clinical Informatics reports could effectively contribute to the promotion of a safer culture through targeting units or areas that require further education rather than blaming individual clinicians. Discussions regarding potential practice and/or protocol changes that can improve safety in infusion therapy further support a just culture through the inclusion of practitioners in the development of strategies, a grass-roots approach to the reduction of adverse drug events. Future Directions Initial evaluation of Baxter s Colleague CXE Infusion Device with GUARDIAN DERS indicates that clinicians and administrators alike find value in the investment in terms of providing safer patient care and promoting a culture of safety. Future directions include a follow up download of data from infusion pumps to validate the efficacy of practice and protocol changes that have been implemented in response to the feedback provided by the Baxter Clinical Informatics team with the goal of improving GUARDIAN utilization in the practice setting. Accreditation Canada (AC) has outlined Required Organizational Procedures (ROPs) for managing medications. SMART pumps addresses ROP 21.0, The organization has a coordinated risk management program to reduce medication related errors and sentinel events, 21.2, The organization s error prevention strategies target
17 the system, not the individual (emphasizing shared accountability), and 21.3, The organization uses a drug use evaluation (DUE) process for medications with heightened error potential (18). References Accreditation Canada website retrieved from Baker, GR., Norton, PG, Flintoft, V., Blais, R., Brown, A., Cox, J., Etchells, E., Ghali, W., Hebert, P., Majumdar, S., O'Beirne, M.,Palacios-Derflingher, L., Reid, R., Sheps, S. & Tamblyn, R. (2004). The Canadian Adverse Events Study: the incidence of adverse events among hospitalized patients in Canada. The Canadian Medical Assoc.Journal: 170(11); Balas, M.C., Scott, L.D. & Rogers, A.E. (2006). Frequency and type of errors and near errors reported by critical care nurses. Canadian Journal of Nursing Research, 38 (2); Balas, M.C., Scott, L.D. & Rogers, A. (2004). The prevalence and nature of errors and near errors reported by hospital staff nurses. Applied Nursing Research, 17 (4); Bates, D. W, Spell, N., Cullen, D. J, Burdick, E., Laird, N., Petersen, L. A, Small, S. D, Sweitzer, B. J & Leape, L. L. (1997). The costs of adverse drug events in hospitalized patients. JAMA, 277: Brennan, T., Leape, L., Laird, N., Hebert, L., Localio, A., Lawthers, A., Newhouse, J., Weiler, P. & Hiatt, H. (1991). Incidence of adverse events and negligence in hospitalized patients. Results of the Harvard Medical Practice Study I. New England Journal of Medicine, 324 (6); Classen, D.C, Pestotnik, S.L, Evans, R.S, Lloyd, J.F, Burke, J.P. (1997). Adverse drug events in hospitalized patients. Excess length of stay, extra costs, and attributable mortality. JAMA, 277: Health Canada (n.d.) retrieved from Malashock, C. M, Gould, D. A, Shull, S S. (2004). Effect of smart infusion pumps on the prevention of potential medication errors resulting from the programming of infusion services. Hospital Pharmacy, 39; Milligan, F. (2003). Adverse healthcare events: Part I. The nature of the problem.
18 Professional Nurse, 18 (9); Neale, G, Woloshynowych, M, Vincent, C. (2001). Exploring the causes of adverse events in NHS hospital practice. Journal of the Royal Society of Medicine, 94; Nuckols, T.K, Paddock, S. M, Bower, A.G., Rothschild, J. M., Fairbanks, R.J., Carlson, B., Panzer, R. J, & Hilborne. L. H (2008) Costs of Intravenous Adverse Drug Events in Academic and Non-academic ICUs..Medical Care, 46(1), Reason, J. (2000). Human error: models and management. British Med. Journal, 320; Rex, J.H., Turnbull, J. E., Allen, S.J., Vande Voorde, K & Luther, K. (2000). Systematic root cause analysis of adverse drug events in a tertiary referral hospital. Joint Commission Journal of Quality Improvement, 26 (10); Rodriguez-Monguio R, Otero M. & Rovira J. (2003). Assessing the economic impact of adverse drug effects. Pharmacoeconomics, 21: UOHI website retrieved from Wilkins, C. & Shields, M. (2008). Correlates of medication error in hospitals. Health Reports, 19 (2); 1-12 Wilson, K & Sullivan, M. (2004). Preventing medication errors with smart infusion technology. American Journal Health System Pharmacy, 61(2);
19 Glossary of Terms: A SMART pump is an infusion device, which includes both hardware and Dose Error Reduction software (DERS). Dose Error Reduction Software (DERS) is software integrated into the pump which allows clinicians to incorporate their best practice guidelines into infusion therapy through dosing limits. Adverse Event (AE) is an unintended injury or complication that results in disability at the time of discharge, death or prolonged hospital stay and that is caused by health care management rather than by the patient s underlying disease process. (Canadian Adverse Events Study) Adverse Drug Event (ADE) is any preventable event that may cause or lead to inappropriate medication use or patient harm, while the medication is in the control of the health care professional, patient or consumer. Such events may be related to professional practice, health care products, procedures and systems including: prescribing; order communication; product labeling, packaging and nomenclature; compounding; dispensing; distribution; administration; education; monitoring; and use. (Institute for Safe Medication Practices) EDITORS: a) Agnese Bianchi, RN, BScN, MN b) Heidi Chevrier RN BScN BA
BIBLIOGRAPHICAL REVIEW ON COST OF PATIENT SAFETY FAILINGS IN ADMINISTRATION OF DRUGS. SUMMARY. Bibliographical review on cost of Patient Safety Failings in administration of drugs. Summary This has been
GUIDELINES GUIDELINES ON PREVENTING MEDICATION ERRORS IN PHARMACIES AND LONG-TERM CARE FACILITIES THROUGH REPORTING AND EVALUATION Preamble The purpose of this document is to provide guidance for the pharmacist
北 里 大 学 一 般 教 育 紀 要 16(2011) 137-150 原 著 The Analysis of Job Satisfaction of Japanese Medical Safety Managers Eiichi KATO Kyoko TAMURA Abstract Medical safety systems are different in every country. The
MEDICATION RECONCILIATION TOOL A Practical Tool to Reduce Medication Errors During Patient Transfer from an Intensive Care Unit Peter Pronovost, MD, PhD, Deborah Baugher Hobson, BSN, Karen Earsing, RN,
SafetyFirst Alert Massachusetts Coalition for the Prevention of Medical Errors January 2001 This issue of Safety First Alert is a publication of the Massachusetts Coalition for the Prevention of Medical
Effective Approaches to Standardization and Implementation of Smart PumpTechnology A CONTINUING EDUCATION PROGRAM FOR PHARMACISTS AND NURSES PROGRAM FACULTY Michael R. Cohen, RPh, MS, ScD, FASHP President
A cluster randomized trial evaluating electronic prescribing in an ambulatory care setting Merrick Zwarenstein, MBBCh, MSc Senior Scientist Institute for Clinically Evaluative Science 2075 Bayview Avenue
, 2001: The First Statewide Assessment Wu Xu, Ph.D.; Stephen Pickard, M.B.A.; Michael P. Silver, M.P.H.; Paul Hougland, M.D.; Carol Masheter, Ph.D.; Steve Donnelly, Ph.D.; Jonathan Nebeker, M.D.; Matthew
Optimizing Medication Administration in a Pediatric ER ER Pharmacist Review of First Dose Non-Emergent Medications Penny Williams, RN, MS Clinical Program Manager, Emergency Center Children s Medical Center
Medication Reconciliation Transfer of medication information across settings keeping it free from error. By Jane H. Barnsteiner, PhD, RN, FAAN Medication error is the most common type of error affecting
Research Article Drug Utilization and Evaluation of Cephalosporin s At Tertiary Care Teaching Hospital, Bangalore HS. Shekar 1*, HR. Chandrashekhar 2, M. Govindaraju 3, P. Venugopalreddy 1, Chikkalingiah
5/12/2011 Objective Failure Modes & Effects Analysis: A U-500 Insulin Case Study Understand the role of a failure mode and effects analysis (FMEA) in developing U-500 insulin use criteria Ryan J. Bickel,
Prevalence and Causes of Wrong Time Medication Administration Errors at Tertiary Care Hospital Karachi, Pakistan When categorized, the Medication administration error can relate to: Wrong Time Wrong Patient
Multiple IV Infusions Webinar Questions and Answers Document Webinar sponsored by: AAMI Foundation/Health Technology Safety Institute Infusion Nurses Society Institute for Safe Medication Practices (ISMP)
Record Review Studies in Switzerland Potential and Limitations Charles Vincent Smith and Nephew Foundation Professor of Clinical Safety Research Clinical Safety Research Unit www.csru.org.uk Drivers of
Menu Case Study 3: Medication Administration Record Applicant Organization: Ontario Shores Centre for Mental Health Sciences Organization s Address: 700 Gordon Street, Whitby, Ontario, Canada, L1N5S9 Submitter
Back to contents Double pumping of intravenous vasoactive drugs in the critical care setting. Approved by & date of Publication Review date September 2006 Lead and contact details * Distribution/ availability
ehealth: the future of health care Royal College of Nursing position statement EHEALTH: THE FUTURE OF HEALTH CARE 2 Nursing staff form the largest group of health care professionals, and because of their
Acknowledgements Anticoagulant Project Co-leads: Carmine Stumpo, TEGH Kris Wichman, ISMP Canada Donna Walsh, ISMP Canada Funded by the Ontario Ministry of Health and Long-Term Care Test Sites: Royal Victoria
Medication Errors In Relation To Education & Medication Errors In Relation To Years of Nursing Experience CHRIS G. BAILEY, BRYAN S. ENGEL, JILLIAN N. LUESCHER, MAEGAN L.TAYLOR This paper was written for
PART 1 Keeping patients safe when they transfer between care providers getting the medicines right Good practice guidance for healthcare professions July 2011 Endorsed by: Foreword Taking a medicine is
Meeting the Standard for Pharmacist Order Review: The Override Challenge Impact on Pharmacy and Nursing Gae M. Ryan, PharmD, Director, Pharmacy Services, Oregon Health Sciences University Barbara Jennings-Garant,
1 Pharmacology Review Quiz: 1. Order: Xylocaine 1 g IV in 500 ml D5W at 3 mg/min. Set the infusion pump at ml/hr 2. Order: Nipride 500 mg IV in 250 ml D5w at 2 mcg/kg/min for a patient weighing 125 lb.
Patient safety alert 18 Alert 28 March 2007 Immediate action Action Update Information request Ref: NPSA/2007/18 4 Actions that can make anticoagulant therapy safer Anticoagulants are one of the classes
Interactive Voice Response Technology To Prevent Type 2 Diabetes in Cardiac Population Overview The University of Ottawa Heart Institute (UOHI) with an annual rate of 6000 admissions is committed to understanding,
DRUg NAme 15 mcg/kg/min Designed with you in mind. The Next-Generation SIGMA Spectrum Infusion System DRUg NAme 15 mcg/kg/min The infusion system that moves with your patient, evolves with your facility
ACCESS TO HEALTH CARE OCTOBER 2005 BriefingPaper Towards faster treatment: reducing attendance and waits at emergency departments Key messages based on a literature review which investigated the organisational
Risk Management and Patient Safety Evolution and Progress Madrid February 2005 Charles Vincent Professor of Clinical Safety Research Department of Surgical Oncology & Technology Imperial College London
KENYATTA UNIVERSITY ESRI 2 ND EDUCATIONAL CONFERENCE UNIVERSITY OF DAR ES SALAAM TITLE: APPLICATION OF GIS TECHNOLOGY TO HOSPITAL MANAGEMENT, PATIENT CARE AND PATIENT FLOW SYSTEMS BY MUGAMBI KELVIN MWENDA
Disclosing Medical Errors to Patients: Developing and Implementing Effective Programs Thomas H. Gallagher, MD University of Washington School of Medicine Accelerating Interest in Disclosure Growing experimentation
From the: U.S. Bureau of Labor Statistics Pharmacy Tech Pharmacy technicians fill prescriptions and check inventory. Pharmacy technicians help licensed pharmacists dispense prescription medication. They
EMR Benefits and Benefit Realization Methods of Stage 6 and 7 Hospitals Hospitals with advanced EMRs report numerous benefits February, 2012 2 Table of Contents 3 Introduction... 4 An Important Question...
Flex Monitoring Team Briefing Paper No. 3 Critical Access Hospital Patient Safety Priorities and Initiatives: Results of the 2004 National CAH Survey September 2004 The Flex Monitoring Team is a consortium
Statement on the Importance of Collaboration between Biomedical Engineering and Clinical Staff and the Simulation Laboratory Activities to Patient Safety and Reporting at Brigham & Women's Hospital (BWH)
IV Clinical Integration Some Things Work Better Together Your smart pumps help enhance safety through guidance at the bedside. Your bar code system, connected to electronic medical records (EMRs), electronically
In this place, it takes all the running you can do, to stay in the same place Lewis Carroll Alice Through The Looking Glass By 2010 we are promised (Direct Quotations) Patients will be admitted for treatment
White Paper WellSpan Cerner Closing the Loop in IV Medication Administration The challenge: 400,000 injuries, $3.5 billion in added costs 1 For nurses, expectations are high. Responses must be quick, but
ANNALS OF HEALTH LAW ADVANCE DIRECTIVE VOLUME 19 SPRING 2010 PAGES 306-315 The Effect of Medical Malpractice Jonathan Thomas * I. INTRODUCTION: WHAT IS MEDICAL MALPRACTICE Every year, medical malpractice
1 Author(s) Year Published 1 Taxis & Barber 2003 (BMJ) 2 Taxis & Barber 2003 (Qu Saf Heth Care) 3 Rothschild, Landrigan, et 4 Husch et Study Objectives incidence and importance of in preparation and of
Disclosing Medical Errors to Patients: International Developments and Future Directions Thomas H. Gallagher, MD University of Washington School of Medicine Accelerating Interest in Disclosure Growing experimentation
healthcare associated infection A C T I O N G U I D E 1.2 AUSTRALIAN SAFETY AND QUALITY GOALS FOR HEALTH CARE What are the goals? The Australian Safety and Quality Goals for Health Care set out some important
doi:10.1111/j.1365-2044.2009.06134.x ORIGINAL ARTICLE The impact of computerised physician order entry on prescribing practices in a cardiothoracic intensive care unit* J. Ali, 1 L. Barrow 2 and A. Vuylsteke
Winchcombe Healthcare Limited CURRENT STATUS: The following summary has been accepted by the Ministry of Health as being an accurate reflection of the Surveillance audit conducted against the Health and
FIP Pharmacy Information Section Requirements for Drug Information Centres Summary All countries should provide drug information services either independently or as part of a regional network. The service
Using Bar Code Verification to Improve Patient Care and Tracking and Traceability William W. Churchill MS, R.Ph. Chief of Pharmacy Services Brigham and Women s Hospital The Brigham and Women s Hospital
Defining the Boundaries between NHS and Private Healthcare (Adapted from NHS Commissioning Board Interim Commissioning Policy: NHSCB cp-12) Produced by: Trish Campbell Version control: V2 March 2013 v1
Health Care Job Information Sheet #2 Patient Care A. Occupations 1) Registered Nurse (RN) 2) Nurse Practitioner (NR) 3) Registered Practical Nurse (RPN) 4) Other positions in the Field B. Labour Market
Institute for Safe Medication Practices 1800 Byberry Road, Suite 810 Huntingdon Valley, PA 19006 FOR MORE INFORMATION, CONTACT : Michael A. Donio, MPA Marketing & Consumer Affairs 215-947-7797 Mdonio@ismp.org
Guide to the National Safety and Quality Health Service Standards for health service organisation boards April 2015 ISBN Print: 978-1-925224-10-8 Electronic: 978-1-925224-11-5 Suggested citation: Australian
Service Specification Template Department of Health, updated June 2015 Service Specification No. : 2 Service: Commissioner Lead: Provider Lead: Period: Anti-coagulation monitoring Date of Review: 31 st
Do we have a common understanding of medication errors? Editor Meyler s Side Effects of Drugs Jeffrey Aronson Co-editor: Stephens Detection and Evaluation of Adverse Drug Reactions Side Effects of Drugs
Commissioning Policy (EMSCGP005V2) Defining the boundaries between NHS and Private Healthcare Although Primary Care Trusts (PCTs) and East Midlands Specialised Commissioning Group (EMSCG) were abolished
CONNECTICUT Downloaded January 2011 19 13 D8T. CHRONIC AND CONVALESCENT NURSING HOMES AND REST HOMES WITH NURSING SUPERVISION (d) General Conditions. (6) All medications shall be administered only by licensed
Please note that the following document was created by the former Australian Council for Safety and Quality in Health Care. The former Council ceased its activities on 31 December 2005 and the Australian
In this place, it takes all the running you can do, to stay in the same place Lewis Carroll Alice Through The Looking Glass By 2010 we are promised (Direct Quotations) Patients will be admitted for treatment
Safety indicators for inpatient and outpatient oral anticoagulant care 1 Recommendations from the British Committee for Standards in Haematology (BCSH) & National Patient Safety Agency (NPSA) Address for
CLAIMS AND CARE MANAGEMENT SOLUTIONS Sedgwick manages injury claims from start to finish with proven strategies that produce better outcomes for clients and injured workers. As the leader in customized
High Integrity Systems in Health Environment Mark Nicholson Introduction National Patient Safety Agency (NPSA - 2003) has described patient safety as process by which an organisation makes patient care
Objectives Sidra Medical & Research Center Building a Data Driven Safety Culture October 6 th 2015 Describe a model Learning Healthcare System Review the role of Health Information Technology in Patient
Nurse Led systemic monitoring clinics guidance on setting up this service Introduction Nurse led systemic monitoring clinics are an innovative approach to improving care delivery and maintaining both a
Learning from Defects Problem Statement: Healthcare organizations could increase the extent to which they learn from defects. We define learning as reducing the probability that a future patient will be
NEW TECHNOLOGIES, NEW WORRIES Medical Malpractice and False Claims Act Liability in the Age Of Electronic Health Records (EHR) and Health Information Exchange (HIE) OVERVIEW MAJOR EHR FUNCTIONALITIES There
SOURCE: Ministry of Health DATE APPROVED: DATE EFFECTIVE: Date of Approval REPLACESPOLICY DATED: 1 POLICY TITLE: Incident/Accident Reporting REFERENCE NO. MOH/04 PAGE: 1 of 7 REVISION DATE(s): Ministry
Cost of medical injury in New Zealand: a retrospective cohort study Paul Brown, Colin McArthur 1, Lynette Newby 1, Roy Lay-Yee, Peter Davis 2, Robin Briant Department of Community Health, University of
FMEA Failure Risk Scoring Schemes 1-10 Scoring for Severity, Occurrence and Detection CATEGORY Severity 10 9 8 7 6 5 3 2 1 Hazardous, without warning Hazardous, with warning Very High High Moderate Low
1. Introduction Southern Health NHS Foundation Trust 1.1 Southern Health NHS Foundation Trust provides Mental Health, Learning Disability, Community and Social Care services in Hampshire, Oxford, Dorset
The State Claims Agency Clinical Indemnity Scheme Incident Notification Requirements. Preface The Clinical Indemnity Scheme (CIS) was established in July 2002, within the State Claims Agency (SCA), (S.I.
The diagnosis of dementia for people living in care homes Frequently Asked Questions by GPs A discussion document jointly prepared by Maggie Keeble, GP with special interest in palliative care and older
promoting access to White Rose research papers Universities of Leeds, Sheffield and York http://eprints.whiterose.ac.uk/ This is an author produced version of a paper published in Journal of Clinical Nursing
Commissioning Policy: Defining the boundaries between NHS and Private Healthcare April 2013 Reference : NHSCB/CP/12 NHS Commissioning Board Commissioning Policy: Defining the Boundaries between NHS and
Reducing Medical Errors with an Electronic Medical Records System A recent report by the Institute of Medicine estimated that as many as 98,000 people die in any given year from medical errors in hospitals
PROPOSAL FOR INTEGRATION OF ICU MEDICAL DEVICES WITH ELECTRONIC MEDICAL RECORD Presented on November 2011 to the Community Health Center Board of Directors Presented by: Jeana O Brien Sharon Perelman Maureen
Introduction of a Dedicated Admissions Nurse to Improve Access to Care for Surgical Patients Editor s Note: In Introduction of a Dedicated Admissions Nurse to Improve Access to Care for Surgical Patients
The Academy of Managed Care Pharmacy s Concepts in Managed Care Pharmacy Medication Errors Medication errors are among the most common medical errors, harming at least 1.5 million people every year. The
Subtitle THE EMR ADOPTION MODEL: RELATIONSHIPS BETWEEN IT ADOPTION, PATIENT SAFETY AND PERFORMANCE PRESENTED BY: UWE BUDDRUS, MANAGING DIRECTOR HIMSS ANALYTICS EUROPE HIMSS HIMSS PARALLEL Analytics SESSION
ISMP Canada Dangerous abbreviations: U can make a difference! By Christine Koczmara, RN, BScPsy, Valentina Jelincic, RPh, BScPhm, and Carol Dueck, RN, BScN Abstract Dangerous abbreviations are also known
RFID in the Health Care Industry RFID in the Hospital Environment Speaker: Daniel Engels, Ph.D. Director of Research MIT Auto-ID Labs I. Mun, Ph.D. Director, Bio-Medical Research Aventura Hospital & Medical
Strategies and Tools to Enhance Performance and Patient Safety Ice Breaker Mod 1 05.2 06.2 Page 2 2 Do No Harm Jess Story Do no Harm Jess' Story Mod 1 05.2 06.2 Page 3 3 Medical Error Have you been affected
Leeds Humulin R (U500) insulin: Prescribing Guidance Amber Drug Level 2 We have started your patient on Humulin R (U500) insulin for the treatment of diabetic patients with marked insulin resistance requiring
NUMBER 19 JANUARY 2005 Nursing Staff Mix: A Key Link to Patient Safety The statistics are startling. Of patients 1 admitted to Canadian acute care hospitals in 2000, an estimated 7.5 per cent experienced
Trust Board 8 May 2014 Title of the Paper: Quarter 4 (1 st January 2014 31 st March 2014) CLIPS Report Agenda item: 205/17 Author: Jackie Ardley, Interim Chief Nurse Trust Objective: 1) Achieving continuous
Health Service Circular Series Number: HSC 2002/009 Issue Date: 04 July 2002 Review Date: 04 July 2005 Category: Public Health Status: Action sets out a specific action on the part of the recipient with
Your consent to our cookies if you continue to use this website.