1 Lecture 16: Near-field Scanning Optical Microscopy (NSOM) Background of NSOM; Basic principles and mechanisms of NSOM; Basic components of a NSOM; Different scanning modes and systems of NSOM; General applications and advantages of NSOM.
2 Scanning probe microscopies STM AFM NSOM Atomic resolution; Molecular bonding. Atomic Interaction: Contact mode Non-contact mode Tapping (intermittent) mode Other Interactions: Electrostatic mode (scanning electrostatic potential microscope) Magnetic mode Chemical Force mode Direct correlation between nanostructures and optical properties tunable SCM (scanning confocal microscope) Single-molecule spectroscopy
3 What can NSOM do? STM measures electric current, and AFM measures forces, neither deals with light; Light is a crucial excitation source in both scientific research and mother nature systems (e.g. photosynthetic system). Scientific research fields: absorption, fluorescence, photoinduced electron transfer, light-emitting devices, photovoltaic cells.
4 Why NSOM? Light diffraction limit of conventional optical microscopy: λ/2, ~ 250 nm. Actually, in real cases, the optical resolution ~ λ, 500 nm; in contrast, NSOM offers higher resolution around 50 nm (or even < 30 nm), depending on tip aperture size. NSOM provides simultaneous measurements of the topography and optical properties (fluorescence) --- direct correlation between surface nanofeatures and optical/electronic properties. This is especially useful for the studying the inhomogeneous materials or surfaces, like nanoparticles, polymer blends, porous silicon, biological systems. A 500 nm wide nanobelt appears wider than real size under an optical microscope.
5 NSOM imaging: Direct correlation between nanostructure features and optical/electronic properties TiO 2 particles wrapped in PPV film Fluorescence quenching by TiO 2 particles NESMI Lab data
6 NSOM Operation System
7 Major components of NSOM Optical: Light source (lasers: CW and pulsed), Fibers, Mirrors, Lenses, Objectives (oil, large NA) Photon detectors (Photon-Multiplier, Avalanche Diode) Probe (tip) Mechanical: Translation stage, Piezo scanner Anti-vibration optical table Electrical: Scanning drivers for piezo scanner z distance control (feedback system) Amplifiers, Signal processors Software and Computer
8 What is Near-Field? requires a nanometer sized aperture (much smaller than the light wavelength). A specimen is scanned very close to the aperture. As long as the specimen remains within a distance less than the aperture diameter, an image with sub-wavelength resolution (aperture size) can be generated. There is a tradeoff between resolution and sensitivity (light intensity) --- aperture size cannot be too small.
9 What is Near-Field? ~ 50 nm ~ 1 nm Near-Field λ/2 ~ 300 nm Near-field: For high spatial resolution, the probe must be close to the sample
10 What is Near-Field? ~ 50 nm ~ 1 nm Near-Field λ/2 ~ 300 nm For high spatial resolution, the probe must be close to the sample
11 Feedback Mechanism 1: Shear Force with Tuning Fork
12 Feedback Mechanism 1: Shear Force with Tuning Fork
13 NSOM based on shear force mode Dan Higgins, Acc. Chem. Res. 2005, 38,
14 NSOM imaging of cleaned glass NESMI Lab data
15 NSOM imaging of cleaned glass NESMI Lab data
16 Feedback Mechanism 2: AFM force sensor Compared to force sensor: Poor spatial resolution due to the fiber cantilever (reflection and spring constant); Damage to sample due to tapping scanning. Nanonics Imaging Ltd.
17 Feedback Mechanism 2: AFM force sensor
18 Structure of a NSOM Tip?
19 Different Operation Modes Illumination by the tip is probably the easiest to operate and interpret, and gives the most signal. It requires a transparent sample, so is limited for application in many samples like silicons and bio-species. Reflection modes give less light, and are more dependent on the details of the probe tip, but allow one to study opaque samples. The illumination/collection mode provides a complement to the reflection modes, but the signal contains a large background.
20 Different Operation Modes
21 Brief History of NSOM Ideas started in mid-1980 s; D.W. Pohl, W. Denk, and M. Lanz, Appl. Phys. Lett. 44, (1984). A. Lewis, M. Isaacson, A. Harootunian, and A. Murray, Ultramicroscopy 13, 227 (1984); Technology developed in 1990 s; Eric Betzig, et al. Science, 262, (1993). Eric Betzig, et al. Nature, 2369, (1994). Prototype commercial available since 2000 s;
22 Quick Looking back : Home Build NSOMs
23 New versions of NSOM 4 companies over the world produce good NSOM systems. The picture shows the model of Veeco Aurora III (DI, Thermomicro). The Aurora III installed in Zang Lab is the newest version (launched in January 2003). This one is based on shear force feedback.
26 Analytical Chemistry, 2003, vol.75, page A
28 Fabrication of NSOM Tip: chemical etching
29 Fabrication of NSOM Tip: mechanical pulling
30 Major applications of NSOM
31 Extended NSOM system spectral and optical imaging
32 Typical examples of NSOM research Quantum size effect for semiconductor nanocrystals; Self-organized nanostructures: thin film dewetting, phase separation; Self-assembly or self-alignment: nanospheres, nanorods, nanowires; Heterogeneous biological systems: cells, proteins, enzymes, membranes; Real optoelectronic devices: solar cells, optical switches, LEDs; Imaging single-molecules (research of SMS was initiated by NSOM); High resolution studies of charge transfer in DNA and polymer chains. To be discussed in the coming lectures
33 Nanoparticles: a unique manifestation Semiconductor nanospheres represent one of the most attractive nanostructures, and have a wide variety of applications in optoelectronics, magnetics, and biological applications. The size of nanoparticles can be tuned between individual molecules and the bulk counterparts. The nanosphere remains the same crystalline structure as the bulk crystal, but it shows unique size dependent physical and chemical properties, so called quantum size effect. See next slide. Conventional spectroscopy measurements of nanoparticles require uniform size distribution of the particle system, which is normally hard to attain. However, NSOM measurement removes such a need by focusing on only one particle a time.
34 Quantum Size Effect of Semiconductor Materials
35 Quantum size effect: an extensively research topic 1. As particle decreases in size, the bandgap increases, approaching the energy difference between LUOM and HOMO for the individual molecules; 2. For fluorescent semiconductor materials, like CdS, the different bandgap leads to different emission wavelength; 3. By making different sizes of the particles, people can tune the emission color across the whole visible region.
37 Organic Semiconductor Nanocrystals: Q-effect less papers on organic nanoparticles, while thousands on inorganic counterparts. Why? CdSe O N O O N O Non-fluorescent fluorescent fluorescent fluorescent Inorganic Nanoparticle Organic Nanoparticle
38 Emission shift of PTCDI molecules upon crystal formation O N O O N O Emission (a.u.) CH 3 OH 38:62 H 2 O:CH 3 OH Quantum size effect is expected for the transition state between the two. crystal Wavelength (nm) NESMI Lab data
39 Tuning Emission of PTCDI materials Free molecules crystals NESMI Lab data
40 Spatially resolving quantum wells GaAs/AlGaAs quantum well structure
43 Typical NSOM Examples: J-Aggregates of dyes
44 Typical NSOM Examples: Muscle Tissues
45 Typical NSOM Examples: Single-Molecules Embedded in Polymer Films
46 Revealing double strands of DNA: fluorescence dye YOYO-1 only combines with the double-strand
47 NSOM imaging in water: an approach to living cells
48 NSOM imaging of living cells
49 Aperturelss NSOM: tip-sample interaction in the polarizable sphere approximation model. when excited by an external radiation beam (wave vector k, fieldintensity E 0 ), the tip (radius of curvature a), gets polarized. The tip-dipole field p induces an image dipole field p located inside the sample at a distance 2r from the tip centre. The backreflected light contains information about the effective polarizability of the tip-sample system.
50 Some limitations (disadvantages) of NSOM Practically zero working distance (for objective) and an extremely small depth of field (for tip). Extremely long scan times for high resolution images or large specimen areas. Very low transmissivity of apertures smaller than the incident light wavelength --- low intensity of incident light for excitation, a problem for weak fluorescent molecules. Only surface features can be imaged and studied. Fiber optic probes are somewhat problematic for imaging soft materials due to their high spring constants, especially in shear-force mode
5. Scanning Near-Field Optical Microscopy 5.1. Resolution of conventional optical microscopy Resolution of optical microscope is limited by diffraction. Light going through an aperture makes diffraction
Adviser: dr. Maja Remškar Institut Jožef Stefan January 2010 1 2 3 4 5 6 Fluorescence Raman and surface enhanced Raman 7 Conventional optical microscopy-limited resolution Two broad classes of techniques
Vol. 93 (1997) A CTA PHYSICA POLONICA A No. 2 Proceedings of the 1st International Symposium on Scanning Probe Spectroscopy and Related Methods, Poznań 1997 NEAR FIELD OPTICAL MICROSCOPY AND SPECTROSCOPY
OSE6650 - Optical Properties of Nanostructured Materials Optical Properties of Nanostructured Materials Fall 2013 Class 3 slide 1 Challenge: excite and detect the near field Thus far: Nanostructured materials
Scanning Near-Field Optical Microscopy for Measuring Materials Properties at the Nanoscale Outline Background Research Design Detection of Near-Field Signal Submonolayer Chemical Sensitivity Conclusions
Scanning Near Field Optical Microscopy: Principle, Instrumentation and Applications Saulius Marcinkevičius Optics, ICT, KTH 1 Outline Optical near field. Principle of scanning near field optical microscope
FEMTOSECOND MEASUREMENTS COMBINED WITH NEAR FIELD OPTICAL MICROSCOPY Artyom A. Astafiev, Semyonov Institute of Chemical Physics, Moscow, Russian Federation. Keywords: diffraction limit nearfield scanning
Lecture 4 Scanning Probe Microscopy (SPM) General components of SPM; Tip --- the probe; Cantilever --- the indicator of the tip; Tip-sample interaction --- the feedback system; Scanner --- piezoelectric
Optical Microscopy Beyond the Diffraction Limit: Imaging Guided and Propagating Fields M. Selim Ünlü, Bennett B. Goldberg, and Stephen B. Ippolito Boston University Department of Electrical and Computer
Near-Field Scanning Optical Microscopy: a Brief Overview Serge HUANT Laboratoire de Spectrométrie Physique (SPECTRO) Université Joseph Fourier Grenoble et CNRS Thanks to my former & present collaborators
Nano-optical Imaging using Scattering Scanning Near-field Optical Microscopy Fehmi Yasin, Advisor: Dr. Markus Raschke, Post-doc: Dr. Gregory Andreev, Graduate Student: Benjamin Pollard Department of Physics,
Nano-Spectroscopy Solutions AFM-Raman, TERS, NSOM Chemical imaging at the nanoscale Since its introduction in the early 80 s, Scanning Probe Microscopy (SPM) has quickly made nanoscale imaging an affordable
Scanning tunneling microscopy STM and atomic force microscopy AFM Content The components of a scanning probe microscope SPM The scanner Measurement of the distance between surface and tip The cantilever
DOE Solar Energy Technologies Program Peer Review Evaluation of Nanocrystalline Silicon Thin Film by Near-Field Scanning Optical Microscopy AAT-2-31605-05 Magnus Wagener and George Rozgonyi North Carolina
Chemical Synthesis Spontaneous organization of molecules into stable, structurally well-defined aggregates at the nanometer length scale. Overview The 1-100 nm nanoscale length is in between traditional
Lecture 6 Scanning Tunneling Microscopy (STM) General components of STM; Tunneling current; Feedback system; Tip --- the probe. Brief Overview of STM Inventors of STM The Nobel Prize in Physics 1986 Nobel
International Journal of Arts and Sciences 3(1): 18-26 (2009) CD-ROM. ISSN: 1944-6934 InternationalJournal.org Usage of Carbon Nanotubes in Scanning Probe Microscopes as Probe Bedri Onur Kucukyildirim,
Microscopy: Principles and Advances Chandrashekhar V. Kulkarni University of Central Lancashire, Preston, United kingdom May, 2014 University of Ljubljana Academic Background 2005-2008: PhD-Chemical Biology
A Microsc. Microanal. Microstruct. 5 (1994) 399 AUGUST/OCTOBER/DECEMBER 1994, PAGE 399 Classification Physics Abstracts 42.30. d On the way to a multitask near field optical microscope: Simultaneous STM/SNOM
Ernst Meyer Hans Josef Hug Roland Bennewitz Scanning Probe Microscopy The Lab on a Tip With 117 Figures Mß Springer Contents 1 Introduction to Scanning Probe Microscopy f f.1 Overview 2 f.2 Basic Concepts
Imaging Methods: Breath Patterns Breath / condensation pattern: By cooling a substrate below the condensation temperature H 2 O will condense in different rates on the substrate with the nucleation rate
Scanning probe microscopy AFM, STM Near field Scanning Optical Microscopy(NSOM) Scanning probe fabrication Scanning Probe Microscopy 1986 Binning and Rohrer shared Nobel Prize in Physics for invention.stm
Raman spectroscopy Lecture Licentiate course in measurement science and technology Spring 2008 10.04.2008 Antti Kivioja Contents - Introduction - What is Raman spectroscopy? - The theory of Raman spectroscopy
Fast-scanning near-field scanning optical microscopy using a high-frequency dithering probe Yongho Seo and Wonho Jhe * Center for Near-field Atom-photon Technology and School of Physics, Seoul National
5684 J. Phys. Chem. B 1997, 101, 5684-5691 A Reflection Near-Field Scanning Optical Microscope Technique for Subwavelength Resolution Imaging of Thin Organic Films Kenneth D. Weston and Steven K. Buratto*
Large scale scanning probe microscope: Making the shear-force scanning visible E. Bosma, a H. L. Offerhaus, J. T. van der Veen, F. B. Segerink, and I. M. van Wessel Optical Sciences Group, Faculty of Science
Review of NSOM Microscopy for Materials Yannick DE WILDE (firstname.lastname@example.org) ESPCI Laboratoire d Optique Physique UPR A0005-CNRS, PARIS, FRANCE Outline : Introduction : concept of near-field scanning
Physics 441/2: Transmission Electron Microscope Introduction In this experiment we will explore the use of transmission electron microscopy (TEM) to take us into the world of ultrasmall structures. This
BIODIVERSITY I BIOL1051 Microscopy Professor Marc C. Lavoie email@example.com MAJOR FUNCTIONS OF MICROSCOPES MAGNIFY RESOLVE: => INCREASE CONTRAST Microscopy 1. Eyepieces 2. Diopter adjustment
Chapter 4 Microscopy, Staining, and Classification 2012 Pearson Education Inc. Lecture prepared by Mindy Miller-Kittrell North Carolina State University Microscopy and Staining 2012 Pearson Education Inc.
Problem Set 3 Solutions CH332 (SP 06) 1. Skoog problem 15-1 (omit terms (j), (k) and (m)). Draw diagrams as necessary. a) fluorescence Relaxation of an excited state by emission of a photon without a change
Chapter 3 Measurement Techniques This chapter describes the techniques that were used to measure the contact angles and the surface texture of the prepared surfaces, as well as the high-speed imaging techniques.
CHAPTER ONE: INTRODUCTION... 1 STUDY BACKGROUND... 1 STUDY GOALS AND OBJECTIVES... 1 INTENDED AUDIENCE... 1 SCOPE AND FORMAT OF REPORT... 2 METHODOLOGY AND INFORMATION SOURCES... 2 ANALYST CREDENTIALS...
Observing Microorganisms through a Microscope Biol 3400 I Microscopy Magnification = how much larger an object is made to appear compared to its real size. Resolving power = The minimum separation distance
Activity 17 Electromagnetic Radiation Why? Electromagnetic radiation, which also is called light, is an amazing phenomenon. It carries energy and has characteristics of both particles and waves. We can
Laboratory #3 Guide: Optical and Electrical Properties of Transparent Conductors -- September 23, 2014 Introduction Following our previous lab exercises, you now have the skills and understanding to control
From apertureless near-field optical microscopy to infrared near-field night vision Yannick DE WILDE ESPCI Laboratoire d Optique Physique UPR A0005-CNRS, PARIS firstname.lastname@example.org From apertureless
Confocal Microscopy and Atomic Force Microscopy (AFM) of biofilms A very brief primer... Fundamentals of Confocal Microscopy Based on a conventional fluorescence microscope Fluorescent Microscope Confocal
Atomic Force Microscopy Long Phan Nanotechnology Summer Series May 15, 2013 1 World s Smallest Movie 2 Outline What is AFM? How does AFM Work? 3 Modes: Contact mode Non contact mode Tapping mode Imaging
(Nano)materials characterization MTX9100 Nanomaterials Lecture 8 OUTLINE 1 -What SEM and AFM are good for? - What is the Atomic Force Microscopes Contribution to Nanotechnology? - What is Spectroscopy?
Journal of Microscopy, Vol. 202, Pt 1, April 2001, pp. 2±6. Received 28 August 2000; accepted 11 October 2000 Diamond colour centres as a nanoscopic light source for scanning near-field optical microscopy
contamination analysis for compound semiconductors ANALYTICAL SERVICES B u r i e d d e f e c t s, E v a n s A n a l y t i c a l g r o u p h e l p s y o u C O N T R O L C O N T A M I N A T I O N Contamination
Introduction to Atomic Force Microscope Introduction to Scanning Force Microscope Not that kind of atomic Tien Ming Chuang ( 莊 天 明 ) Institute of Physics, Academia Sinica Tien Ming Chuang ( 莊 天 明 ) Institute
UCRL-ID-142178 Defect studies of optical materials using near-field scanning optical microscopy and spectroscopy M. Yan, J. McWhirter, T. Huser, W. Siekhaus January, 2001 U.S. Department of Energy Laboratory
j1 1 Introduction 1.1 Historical Perspective The invention of scanning probe microscopy is considered one of the major advances in materials science since 1950 [1, 2]. Scanning probe microscopy includes
Raman spectroscoopy Matti Hotokka Department of Physical Chemistry Åbo Akademi University The phenomenon Electronically excited state Virtual level, light is not absorbed Resonance Raman Vibrational levels
Hard Condensed Matter WZI Tom Gregorkiewicz University of Amsterdam VU-LaserLab Dec 10, 2015 Hard Condensed Matter Cluster Quantum Matter Optoelectronic Materials Quantum Matter Amsterdam Mark Golden Anne
CSCI 4974 / 6974 Hardware Reverse Engineering Lecture 8: Microscopy and Imaging Data Acquisition for RE Microscopy Imaging Registration and stitching Microscopy Optical Electron Scanning Transmission Scanning
Nanoparticle Technologies 1 2 NANOPARTICLE SYNTHESIS FROM BATCH TO CONTINUOUS The importance and unique advantages of nano- and microparticles made from different types of inorganic and organic/polymeric
LabRAM HR Research Raman Made Easy! Raman Spectroscopy Systems Powered by: Spectroscopy Suite Cutting-Edge Applications with the LabRAM HR Deeply involved in Raman spectroscopy for decades, HORIBA Scientific
Feedback Control in an Atomic Force Microscope Used as a Nano-Manipulator M. Hrouzek This paper offers a concise survey of the most commonly used feedback loops for atomic force microscopes. In addition
Nanoelectronics 09 Atsufumi Hirohata Department of Electronics 12:00 Wednesday, 4/February/2015 (P/L 006) Quick Review over the Last Lecture ( Field effect transistor (FET) ): ( Drain ) current increases
Fluorescence Workshop UMN Physics June 8-10, 2006 Fluorescence Microscopy and Fluorescence Correlation Spectroscopy Joachim Mueller Fluorescence Microscopy Use a microscope as a fluorometer Advantages:
Notice TPA (AFM) Scanning Probe Microscopy Résumé R. Sanjinés Date: 09.09.2014 SCANNING PROBE MICROSCOPY: Study of surfaces by 3- Dimentional images I Introduction Atomic Force Microscopy (AFM) The Atomic
Polarized light microscopy Polarized light microscopy uses plane-polarized light to analyze structures that are birefringent; unisotropic structures that have two different refractive indices, and capable
1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 Single Defect Center Scanning Near-Field Optical Microscopy on Graphene J. Tisler, T. Oeckinghaus, R. Stöhr, R. Kolesov, F. Reinhard and J. Wrachtrup 3. Institute
UNIVERSITY OF SOUTHAMPTON Scanning Near-Field Optical Microscope Characterisation of Microstructured Optical Fibre Devices. Christopher Wyndham John Hillman Submitted for the degree of Doctor of Philosophy
Near-field scanning optical microscopy using a super-resolution cover glass slip Yu-Hsuan Lin 1,2 and Din Ping Tsai 1,3,4,* 1 Instrument Technology Research Center, National Applied Research Laboratories,
Fiber Optics: Engineering from Global to Nanometer Dimensions Prof. Craig Armiento Fall 2003 1 Optical Fiber Communications What is it? Transmission of information using light over an optical fiber Why
From functional nanostructured surfaces to innovative optical biosensors Giacomo Dacarro Dipartimento di Fisica A.Volta Dipartimento di Chimica Generale Università degli Studi di Pavia Dalla scienza dei
1 of 5 10/24/2006 12:55 PM Transmission electron microscopy From Wikipedia, the free encyclopedia (Redirected from Transmission electron microscope) It has been suggested that Selected area diffraction
Nano water Demonstration: Light Interactions with Dye Molecules and Nanomaterials Key Concepts: In this module, students are expected to learn: How the light can be absorbed or scattered by small objects
Microscopy Optical Sectioning Stephen Ross Ph.D. UCSF Principles and Practice Of Light Microscopy May 3 rd, 2010 Confocal Microscopy In Principle Point Scanning Confocal Spinning Disk Confocal Swept Field
12. CONFOCAL MICROSCOPY Confocal microscopy can render depth-resolved slices through a 3D object by rejecting much of the out of focus light via a pinhole. The image is reconstructed serially, i.e. point
Zeiss 780 Training Notes 780 Start Up Sequence Do you need the argon laser, 458,488,514nm lines? No Turn on the Systems PC Switch Turn on Main Power Switch Yes Turn on the laser main power switch and turn
Electronic Supplementary Information (ESI) for Journal of Materials Chemistry A Facile, One-Pot Synthesis of Ultra-long Nanoparticle-chained Polyaniline Wires Supplementary Information Taekyeong Kim, Kwang
Two-photon FCS Tutorial Berland Lab Department of Physics Emory University What is FCS? FCS : Fluorescence Correlation Spectroscopy FCS is a technique for acquiring dynamical information from spontaneous
Spettroscopia Raman in campo prossimo Salvatore Patanè (1) and Pietro G. Gucciardi (2) (1) Dip. di Fisica della Materia e Tecnologie Fisiche Avanzate, Università di Messina, Salita Sperone 31, 98166 Messina,
Generating contrast in light microscopy Orion Weiner Principles & Practice of Light Microscopy 2013 Absorption is not the only way samples interact with light. (polarization, phase shift) Brightfield Phase
Part 5: Lasers Amplification Atomic (or molecular, or semiconductor) system has energy levels Some higher energy states are stable for a short time (ps to ms) Incident photon can trigger emission of an
Northeastern University, PHYS5318 Spring 2014, 1 1. Introduction Experiment 5. Lasers and laser mode structure The laser is a very important optical tool that has found widespread use in science and industry,
X-Rays and Magnetism From Fundamentals to Nanoscale Dynamics Joachim Stöhr Stanford Synchrotron Radiation Laboratory X-rays have come a long way 1895 1993 10 cm 10 µm 100 nm Collaborators: SSRL Stanford:
Fluorescence Microscopy for an NMR- Biosensor Project Ole Hirsch Physikalisch-Technische Bundesanstalt Medical Optics Abbestr. -1, 10587 Berlin, Germany Overview NMR Sensor Project Dimensions in biological
Nanoparticle Enhanced Thin Film Solar Cells Solar Cells Solar cells convert visible light to electricity. It is one of the clean sources of energy. In theory a 100 square mile area covered with solar panels
Raman Spectroscopy Basics Introduction Raman spectroscopy is a spectroscopic technique based on inelastic scattering of monochromatic light, usually from a laser source. Inelastic scattering means that
1.5 Light absorption by solids Bloch-Brilloin model L e + + + + + allowed energy bands band gaps p x In a unidimensional approximation, electrons in a solid experience a periodic potential due to the positively