# Three-axis optical force plate for studies in small animal locomotor mechanics

Save this PDF as:

Size: px
Start display at page:

Download "Three-axis optical force plate for studies in small animal locomotor mechanics"

## Transcription

4 S. Tonia Hsieh Rev. Sci. Instrum. 77, Signal-position calibration protocol Equidistant points were marked off in a grid on the force platform and a single known mass was placed at each point to obtain Z-axis channel responses. Since the total torque total acting on any static object is always zero, position calibration curves were derived by calculating the torques acting on each channel 1 4 for the Z axis, mass = =0, F L X = b 1 F 1 + b 2 F 2 + b 3 F 3 + b 4 F 4, F L Y = b 5 F 1 + b 6 F 2 + b 7 F 3 + b 8 F 4, 8 FIG. 2. Signal-force calibration results. a Schematic of the calibration setup. The thick, solid line is the tape and the dotted lines are string. The calibration mass is labeled as m. Individual channel calibration results were plotted against the actual calibration weight for the b X axis, c Y axis, and d Z axis. The dotted lines are the linear regressions. Slopes for all axes were 1.00, with R and solving for X and Y which are the positions on the force platform at which the mass is placed. Since cross talk between axes was low see Sec. III, the uncorrected force value for each channel, a i s i, was used as the value for F i. The constants b 1 8 are the perpendicular distances of the Z-axis sensors from the platform origin. If the weight is placed at p positions, the following system of matrices represents the calibration scheme and it is possible to solve for the vector of b constants using a least squares fit: X p = F1 /F L Y X F 1 /F L Y ] X F p p 1 /F L Y F 2 /F L F 3 /F L F 4 /F L F 1 /F L F 2 /F L F 3 /F L F b 4 /F 2 L F 2 /F L F 3 /F L F b 4 /F L F 1 /F L F 2 /F L F 3 /F L F b 4 4 /F L b 5 ] ] ] ] ] ] ] ] b F p p 2 /F L F p p 3 /F L F p p 6 4 /F L b F p p 1 /F L F p p 2 /F L F p p p b1 3 /F L F p 7 4 /F L b 8. 9 D. Data collection Fishes were induced to jump off the force plate in the X direction. Each jump was recorded with one high-resolution , high-speed camera Photron Ultima APX, San Diego, CA filming at 500 frames/s, zoomed in on the lateral view of the force platform and fish, and one camcorder Sony Handycam model DCR-TRV38 simultaneously filming the lateral and dorsal via a 45 mirror views of the entire length of the jump. A 0.5 cm grid was positioned in the background to permit video calibration and extraction of jump distance and height. The analog force plate response signals were electronically amplified, then converted to a digital signal using an ADInstruments Colorado Springs, CO analog-digital converter. Data were sampled at 1 10 khz and recorded on a laptop computer with POWERLAB CHART5.0 ADInstruments, Colorado Springs, CO software. Jump force during a trial was calculated using the a vector of constants determined during the signal-force calibration. E. Confirming force plate accuracy The blenny was modeled as a point-mass projectile initially subject to three forces in the X, Y, and Z directions. Time to peak height t p, maximum jump height z t p, and jump distance d jump were determined to be t p = ż 0 /g, 10 z t p = ż 2 0 t p /2g, and 11 d jump =2t p ẋ ẏ Calculated values for jump height and distance were then compared with those determined from the video images. The moment when the blenny s tail left the locomotor surface was set as t=0.

5 Optical force plate design Rev. Sci. Instrum. 77, FIG. 3. Signal-position calibration results for real vs. calculated a X position and b Y position. In both panels, the fitted linear regression dotted lines have a slope of 1.00, R 2 =0.999 X and Y, and p c Ball-rolling experiment result. X-Y position plots of the ball position digitized from video footage and the ball position calculated from individual force plate Z-axis channel output. To assess accuracy of center of pressure determination, a 1.06 g steel ball bearing was rolled across the force platform and filmed in dorsal view. The motion of the steel ball was then digitized and the calculated position data were then compared with the actual, digitized data for confirmation of center of pressure accuracy. III. RESULTS AND DISCUSSION A. Force plate calibration results Force calibration and position calibration curves were linear, all with R Figs. 2 b 2 d, 3 a, and 3 b. Force plate natural frequency was much lower than desired Hz because the moving elements of the force plate were heavy relative to the blades spring stiffness. Mean cross talk between channels were 4.3% from X to Y, 0.6% from X to Z, 0.6% from Y to X, 8.2% from Y to Z, 8.1% from Z to X, and 4.4% from Z to Y. B. Experimental results When rolling a ball across the surface of the force plate, calculation of the location of center of pressure matched up well Fig. 3 c, with maximum errors of 0.56 and 0.41 cm at the start and end of the roll when the ball was moving most rapidly. Mean errors were 0.07 cm along the X axis and 0.13 cm along the Y axis. The apparent mismatch between calibration accuracy along the Y axis as shown in Fig. 3 b with the calculated results in panel c may be caused by two factors. First, the system was not fully modeled and therefore some characteristics of force plate response, such as damping elements, were not accounted for. A more rigorous model of FIG. 4. Jumping blenny data. a A single high-speed video frame of a Pacific leaping blenny jumping off a horizontal surface on land. Representative unfiltered gray line and filtered black line force traces of a 0.6 g blenny jumping off the force plate for the b X axis, c Y axis, and d Z axis. The black triangles represent the location of peak positive and negative forces used to calculate signal/noise ratios presented in Table I. The dotted line in panel d is the fish s body weight in mn. The blenny s tail loses contact with the surface at t=0 s. the dynamic components of the system would increase its positional accuracy. Second, the positional calibration as described here assumes the cross talk between channels is negligible when determining the center of pressure location. Although cross talk was shown to be less than 10% among all axes, this assumption likely introduced some error into the final center of pressure localization calculations. The forces generated along each axis during an actual blenny jump are shown in Fig. 4. All data were filtered with a ninth-order Butterworth low-pass filter with a cut-off frequency of 20 Hz. A high order for the filter was chosen to increase the steepness of signal drop-off. For the trial shown

7 Optical force plate design Rev. Sci. Instrum. 77, I. K. Bartol, G. Morteza, D. Weihs, P. W. Webb, J. R. Hove, and M. S. Gordon, J. Exp. Biol. 206, E. G. Drucker and G. V. Lauder, J. Exp. Biol. 203, F. E. Fish, J. Hurley, and D. P. Costa, J. Exp. Biol. 206, J. E. Harris, J. Exp. Biol. 13, Y.-C. Pai and K. Iqbal, J. Biomech. 32, M. G. McCay, J. Exp. Biol. 204, D. L. Jindrich and R. J. Full, J. Exp. Biol. 205, J. Schmitt, M. Garcia, R. C. Razo, P. Holmes, and R. J. Full, Biol. Cybern. 86, M. Sun and Y. Xiong, J. Exp. Biol. 208, L. H. Ting, R. Blickhan, and R. J. Full, J. Exp. Biol. 197, G. A. Cavagna, J. Appl. Physiol. 39, A. A. Biewener and R. J. Full, in Biomechanics Structures and Systems: A Practical Approach, edited by A. A. Biewener Oxford University Press, Oxford, 1992, pp M. Bhikajee and J. M. Green, African Zoology 37, R. Blickhan and R. J. Full, in Biomechanics Structures and Systems: A Practical Approach, edited by A. A. Biewener Oxford University Press, Oxford, 1992, pp N. C. Heglund, J. Exp. Biol. 93, F. P. Beer and E. R. Johnston, Jr., Mechanics of Materials McGraw-Hill, New York, 1992, p. 740.

### 22.302 Experiment 5. Strain Gage Measurements

22.302 Experiment 5 Strain Gage Measurements Introduction The design of components for many engineering systems is based on the application of theoretical models. The accuracy of these models can be verified

### Chapter 5: Tool Dynamometers

Chapter 5: Tool Dynamometers LEARNING OBJECTIVES Different types of transducers used in Dynamometers Design Requirements Types of Dynamometers ---------------------------------------------------------------------------------------------------------------------

### Calibration and Use of a Strain-Gage-Instrumented Beam: Density Determination and Weight-Flow-Rate Measurement

Chapter 2 Calibration and Use of a Strain-Gage-Instrumented Beam: Density Determination and Weight-Flow-Rate Measurement 2.1 Introduction and Objectives This laboratory exercise involves the static calibration

w Technical Product Information Precision Miniature Load Cell with Overload Protection 1. Introduction The load cells in the model 8431 and 8432 series are primarily designed for the measurement of force

### EDUMECH Mechatronic Instructional Systems. Ball on Beam System

EDUMECH Mechatronic Instructional Systems Ball on Beam System Product of Shandor Motion Systems Written by Robert Hirsch Ph.D. 998-9 All Rights Reserved. 999 Shandor Motion Systems, Ball on Beam Instructional

### Piezoelectric Polymer Speakers Application Note Aug 98 Rev A

1 Aug 98 Rev A INTRODUCTION A loudspeaker is a transducer which transforms electrical signals into acoustic sound. Conventional speakers usually consist of a vibrating surface (called a diaphragm) and

### Position Sensors MicroSense, LLC

Understanding Capacitive Position Sensors 2011 MicroSense, LLC Table of Contents 1. Introduction to Capacitive Sensors... 3 1.1 Characteristics of Capacitive Sensors... 3 1.1.1 Non Contact... 3 1.1.2 High

### Active Vibration Isolation of an Unbalanced Machine Spindle

UCRL-CONF-206108 Active Vibration Isolation of an Unbalanced Machine Spindle D. J. Hopkins, P. Geraghty August 18, 2004 American Society of Precision Engineering Annual Conference Orlando, FL, United States

### International Journal of Engineering Research-Online A Peer Reviewed International Journal Articles available online http://www.ijoer.

RESEARCH ARTICLE ISSN: 2321-7758 DESIGN AND DEVELOPMENT OF A DYNAMOMETER FOR MEASURING THRUST AND TORQUE IN DRILLING APPLICATION SREEJITH C 1,MANU RAJ K R 2 1 PG Scholar, M.Tech Machine Design, Nehru College

### Trace Gas Exchange Measurements with Standard Infrared Analyzers

Practical Environmental Measurement Methods Trace Gas Exchange Measurements with Standard Infrared Analyzers Last change of document: February 23, 2007 Supervisor: Charles Robert Room no: S 4381 ph: 4352

### PUMPED Nd:YAG LASER. Last Revision: August 21, 2007

PUMPED Nd:YAG LASER Last Revision: August 21, 2007 QUESTION TO BE INVESTIGATED: How can an efficient atomic transition laser be constructed and characterized? INTRODUCTION: This lab exercise will allow

Blackbody Radiation References 1) R.A. Serway, R.J. Beichner: Physics for Scientists and Engineers with Modern Physics, 5 th Edition, Vol. 2, Ch.40, Saunders College Publishing (A Division of Harcourt

### Force. Net Force Mass. Acceleration = Section 1: Weight. Equipment Needed Qty Equipment Needed Qty Force Sensor 1 Mass and Hanger Set 1 Balance 1

Department of Physics and Geology Background orce Physical Science 1421 A force is a vector quantity capable of producing motion or a change in motion. In the SI unit system, the unit of force is the Newton

### There are many different type of accelerometers and each has unique characteristics, advantages and disadvantages. The different types include:

How does a piezo-electric accelerometer work? Piezo-electric crystals are man-made or naturally occurring crystals that produce a charge output when they are compressed, flexed or subjected to shear forces.

### 8.2 Elastic Strain Energy

Section 8. 8. Elastic Strain Energy The strain energy stored in an elastic material upon deformation is calculated below for a number of different geometries and loading conditions. These expressions for

### Data: 2.2 Does an inductive sensor react to non-ferromagnetic materials? 2.3 Does an inductive sensor react to non-conductive materials?

Lab-exercises Actuators and EMC/Sensory Analysis Faculty of Mechatronics and Electrical Engineering Laboratory of Electrical Drives and Installations Lab-exercise EA - 4 Inductive Position Sensor Data:

### Force measurement. Forces VECTORIAL ISSUES ACTION ET RÉACTION ISOSTATISM

Force measurement Forces VECTORIAL ISSUES In classical mechanics, a force is defined as "an action capable of modifying the quantity of movement of a material point". Therefore, a force has the attributes

### Rotational Motion. Symbol Units Symbol Units Position x (m) θ (rad) (m/s) " = d# Source of Parameter Symbol Units Parameter Symbol Units

Introduction Rotational Motion There are many similarities between straight-line motion (translation) in one dimension and angular motion (rotation) of a rigid object that is spinning around some rotation

### Concept Review. Physics 1

Concept Review Physics 1 Speed and Velocity Speed is a measure of how much distance is covered divided by the time it takes. Sometimes it is referred to as the rate of motion. Common units for speed or

### 4 SENSORS. Example. A force of 1 N is exerted on a PZT5A disc of diameter 10 mm and thickness 1 mm. The resulting mechanical stress is:

4 SENSORS The modern technical world demands the availability of sensors to measure and convert a variety of physical quantities into electrical signals. These signals can then be fed into data processing

### Course 1 Laboratory. Second Semester. Experiment: Young s Modulus

Course 1 Laboratory Second Semester Experiment: Young s Modulus 1 Elasticity Measurements: Young Modulus Of Brass 1 Aims of the Experiment The aim of this experiment is to measure the elastic modulus with

### Force Measurement Systems. Force Measurement Systems

Force Measurement Systems Force Measurement Systems Force Measurement Systems from IMS Your Guarantee of Quality Modern production equipment is designed for high productivity and quality. To achieve these

### Experiment P007: Acceleration due to Gravity (Free Fall Adapter)

Experiment P007: Acceleration due to Gravity (Free Fall Adapter) EQUIPMENT NEEDED Science Workshop Interface Clamp, right angle Base and support rod Free fall adapter Balls, 13 mm and 19 mm Meter stick

### Response to Harmonic Excitation Part 2: Damped Systems

Response to Harmonic Excitation Part 2: Damped Systems Part 1 covered the response of a single degree of freedom system to harmonic excitation without considering the effects of damping. However, almost

### Simple Harmonic Motion

Simple Harmonic Motion 1 Object To determine the period of motion of objects that are executing simple harmonic motion and to check the theoretical prediction of such periods. 2 Apparatus Assorted weights

### Subminiature Load Cell Model 8417

w Technical Product Information Subminiature Load Cell 1. Introduction... 2 2. Preparing for use... 2 2.1 Unpacking... 2 2.2 Using the instrument for the first time... 2 2.3 Grounding and potential connection...

### Solving Simultaneous Equations and Matrices

Solving Simultaneous Equations and Matrices The following represents a systematic investigation for the steps used to solve two simultaneous linear equations in two unknowns. The motivation for considering

### Physics 221 Experiment 5: Magnetic Fields

Physics 221 Experiment 5: Magnetic Fields August 25, 2007 ntroduction This experiment will examine the properties of magnetic fields. Magnetic fields can be created in a variety of ways, and are also found

### Diffraction and Interference of Light

Diffraction and Interference of Light Theory: When light encounters an opaque barrier with an opening that is not too large relative to the wavelength, it will bend around the edges to illuminate the space

### Enhancing the SNR of the Fiber Optic Rotation Sensor using the LMS Algorithm

1 Enhancing the SNR of the Fiber Optic Rotation Sensor using the LMS Algorithm Hani Mehrpouyan, Student Member, IEEE, Department of Electrical and Computer Engineering Queen s University, Kingston, Ontario,

### Chapter 21 Band-Pass Filters and Resonance

Chapter 21 Band-Pass Filters and Resonance In Chapter 20, we discussed low-pass and high-pass filters. The simplest such filters use RC components resistors and capacitors. It is also possible to use resistors

### IMU Components An IMU is typically composed of the following components:

APN-064 IMU Errors and Their Effects Rev A Introduction An Inertial Navigation System (INS) uses the output from an Inertial Measurement Unit (IMU), and combines the information on acceleration and rotation

### Spring Force Constant Determination as a Learning Tool for Graphing and Modeling

NCSU PHYSICS 205 SECTION 11 LAB II 9 FEBRUARY 2002 Spring Force Constant Determination as a Learning Tool for Graphing and Modeling Newton, I. 1*, Galilei, G. 1, & Einstein, A. 1 (1. PY205_011 Group 4C;

### Chapter 8. Low energy ion scattering study of Fe 4 N on Cu(100)

Low energy ion scattering study of 4 on Cu(1) Chapter 8. Low energy ion scattering study of 4 on Cu(1) 8.1. Introduction For a better understanding of the reconstructed 4 surfaces one would like to know

### ACTIVITY SIX CONSERVATION OF MOMENTUM ELASTIC COLLISIONS

1 PURPOSE ACTIVITY SIX CONSERVATION OF MOMENTUM ELASTIC COLLISIONS For this experiment, the Motion Visualizer (MV) is used to capture the motion of two frictionless carts moving along a flat, horizontal

### Optics. Determining the velocity of light by means of the rotating-mirror method according to Foucault and Michelson. LD Physics Leaflets P5.6.1.

Optics Velocity of light Measurement according to Foucault/Michelson LD Physics Leaflets P5.6.1.1 Determining the velocity of light by means of the rotating-mirror method according to Foucault and Michelson

### MENG 302L Lab 4: Modulus of Elasticity and Poisson s Ratio

MENG 302L Lab 4: Modulus of Elasticity and Poisson s Ratio Introduction: In Lab 4 we will measure the two fundamental elastic constants relating stress to strain: Modulus of Elasticity and Poisson s Ratio.

### LABORATORY 9. Simple Harmonic Motion

LABORATORY 9 Simple Harmonic Motion Purpose In this experiment we will investigate two examples of simple harmonic motion: the mass-spring system and the simple pendulum. For the mass-spring system we

### Dynamic Measurement of Brake Pads Material Parameter

VDA Dynamic Measurement of Brake Pads Material Parameter 315 This non-binding recommendation by the German Association of the Automotive Industry (VDA) has the following objectives: This standard treats

### MCE380: Measurements and Instrumentation Lab. Chapter 9: Force, Torque and Strain Measurements

MCE380: Measurements and Instrumentation Lab Chapter 9: Force, Torque and Strain Measurements Topics: Elastic Elements for Force Measurement Dynamometers and Brakes Resistance Strain Gages Holman, Ch.

### Experiment 3 Modulus of Elasticity

Related topics Young s modulus, modulus of elasticity, stress, deformation, Poisson s ratio, Hooke s Law. Principle and task A flat bar is supported at two points. It is bent by the action a force acting

### ε: Voltage output of Signal Generator (also called the Source voltage or Applied

Experiment #10: LR & RC Circuits Frequency Response EQUIPMENT NEEDED Science Workshop Interface Power Amplifier (2) Voltage Sensor graph paper (optional) (3) Patch Cords Decade resistor, capacitor, and

### PHYS-2212 LAB Coulomb s Law and the Force between Charged Plates

PHYS-2212 LAB Coulomb s Law and the Force between Charged Plates Objectives To investigate the electrostatic force between charged metal plates and determine the electric permittivity of free space, ε

### A Determination of g, the Acceleration Due to Gravity, from Newton's Laws of Motion

A Determination of g, the Acceleration Due to Gravity, from Newton's Laws of Motion Objective In the experiment you will determine the cart acceleration, a, and the friction force, f, experimentally for

### Mounting instructions. Acceleration Transducer B12. B 26.B12.10 en

Mounting instructions Acceleration Transducer B12 B 26.B12.10 en B12 3 Contents Page Safety instructions.............................................. 4 1 Scope of supply..............................................

### Weight Measurement Technology

Kistler-Morse (KM) introduced bolt-on weight measuring systems three decades ago. These devices featured Walter Kistler s invention, the Microcell. Over the years, many improvements were made to the Microcell

### Physics-1 Recitation-7

Physics-1 Recitation-7 Rotation of a Rigid Object About a Fixed Axis 1. The angular position of a point on a wheel is described by. a) Determine angular position, angular speed, and angular acceleration

### Experiment 5. Lasers and laser mode structure

Northeastern University, PHYS5318 Spring 2014, 1 1. Introduction Experiment 5. Lasers and laser mode structure The laser is a very important optical tool that has found widespread use in science and industry,

### Lab 7: Rotational Motion

Lab 7: Rotational Motion Equipment: DataStudio, rotary motion sensor mounted on 80 cm rod and heavy duty bench clamp (PASCO ME-9472), string with loop at one end and small white bead at the other end (125

### ISC Dual Frequency WFS RFPD Test Procedure:

LASER INTERFEROMETER GRAVITATIONAL WAVE OBSERVATORY LIGO Laboratory / LIGO Scientific Collaboration LIGO 08 January 2013 ISC Dual Frequency WFS RFPD Test Procedure: Richard Abbott Distribution of this

### Time Domain and Frequency Domain Techniques For Multi Shaker Time Waveform Replication

Time Domain and Frequency Domain Techniques For Multi Shaker Time Waveform Replication Thomas Reilly Data Physics Corporation 1741 Technology Drive, Suite 260 San Jose, CA 95110 (408) 216-8440 This paper

### Response to Harmonic Excitation

Response to Harmonic Excitation Part 1 : Undamped Systems Harmonic excitation refers to a sinusoidal external force of a certain frequency applied to a system. The response of a system to harmonic excitation

### Series 6000 Torque measured metal bellow coupling

Properties Free of float metal bellow coupling with integrated torque measurement Non-contact measurement system, high robustness High torsional stiffness Limited torque of inertia Performance Measurement

### AP1 Oscillations. 1. Which of the following statements about a spring-block oscillator in simple harmonic motion about its equilibrium point is false?

1. Which of the following statements about a spring-block oscillator in simple harmonic motion about its equilibrium point is false? (A) The displacement is directly related to the acceleration. (B) The

### Structures and Stiffness

Structures and Stiffness ENGR 10 Introduction to Engineering Ken Youssefi/Thalia Anagnos Engineering 10, SJSU 1 Wind Turbine Structure The Goal The support structure should be optimized for weight and

### CALIBRATING LOW FREQUENCY DIGITAL HYDROPHONES

CALIBRATING LOW FREQUENCY DIGITAL HYDROPHONES D. Tom Dakin, Jeff Bosma, John Dorocicz, Nicolai Bailly University of Victoria, TEF-160, 2300 McKenzie Ave., PO Box 1700, Stn CSC, Victoria, BC, Canada, V8W

### Comparison of the Response of a Simple Structure to Single Axis and Multiple Axis Random Vibration Inputs

Comparison of the Response of a Simple Structure to Single Axis and Multiple Axis Random Vibration Inputs Dan Gregory Sandia National Laboratories Albuquerque NM 87185 (505) 844-9743 Fernando Bitsie Sandia

### Designing interface electronics for zirconium dioxide oxygen sensors of the XYA series

1 CIRCUIT DESIGN If not using one of First Sensors ZBXYA interface boards for sensor control and conditioning, this section describes the basic building blocks required to create an interface circuit Before

### Tire testing at real driving conditions and

Tire testing at real driving conditions and at tthe test tstand Intelligent Tire Technology 26 28 September 2011 Darmstadt Content Motivation Measurement Equipment Approach Test description Results Conclusions

### E190Q Lecture 5 Autonomous Robot Navigation

E190Q Lecture 5 Autonomous Robot Navigation Instructor: Chris Clark Semester: Spring 2014 1 Figures courtesy of Siegwart & Nourbakhsh Control Structures Planning Based Control Prior Knowledge Operator

### Pre-lab Quiz/PHYS 224 Magnetic Force and Current Balance. Your name Lab section

Pre-lab Quiz/PHYS 224 Magnetic Force and Current Balance Your name Lab section 1. What do you investigate in this lab? 2. Two straight wires are in parallel and carry electric currents in opposite directions

### Structural Axial, Shear and Bending Moments

Structural Axial, Shear and Bending Moments Positive Internal Forces Acting Recall from mechanics of materials that the internal forces P (generic axial), V (shear) and M (moment) represent resultants

Compression Test, METHOD OF STATEMENT FOR STATIC LOADING TEST Tension Test and Lateral Test According to the American Standards ASTM D1143 07, ASTM D3689 07, ASTM D3966 07 and Euro Codes EC7 Table of Contents

### Introduction to Beam. Area Moments of Inertia, Deflection, and Volumes of Beams

Introduction to Beam Theory Area Moments of Inertia, Deflection, and Volumes of Beams Horizontal structural member used to support horizontal loads such as floors, roofs, and decks. Types of beam loads

### From Figure 5.1, an rms displacement of 1 mm (1000 µm) would not cause wall damage at frequencies below 3.2 Hz.

5-1 Problems and Solutions Section 5.1 (5.1 through 5.5) 5.1 Using the nomograph of Figure 5.1, determine the frequency range of vibration for which a machine oscillation remains at a satisfactory level

### Bending, Forming and Flexing Printed Circuits

Bending, Forming and Flexing Printed Circuits John Coonrod Rogers Corporation Introduction: In the printed circuit board industry there are generally two main types of circuit boards; there are rigid printed

### Mechanics. Determining the gravitational constant with the gravitation torsion balance after Cavendish. LD Physics Leaflets P1.1.3.1.

Mechanics Measuring methods Determining the gravitational constant LD Physics Leaflets P1.1.3.1 Determining the gravitational constant with the gravitation torsion balance after Cavendish Measuring the

### The Electronic Scale

The Electronic Scale Learning Objectives By the end of this laboratory experiment, the experimenter should be able to: Explain what an operational amplifier is and how it can be used in amplifying signal

### 14.5GHZ 2.2KW CW GENERATOR. GKP 22KP 14.5GHz WR62 3x400V

14.5GHZ 2.2KW CW GENERATOR GKP 22KP 14.5GHz WR62 3x400V UTILIZATION OF GKP 22KP GENERATOR With its characteristics of power stability whatever the load, very fast response time at a pulse, low ripple,

### Plate waves in phononic crystals slabs

Acoustics 8 Paris Plate waves in phononic crystals slabs J.-J. Chen and B. Bonello CNRS and Paris VI University, INSP - 14 rue de Lourmel, 7515 Paris, France chen99nju@gmail.com 41 Acoustics 8 Paris We

### High Resolution Spatial Electroluminescence Imaging of Photovoltaic Modules

High Resolution Spatial Electroluminescence Imaging of Photovoltaic Modules Abstract J.L. Crozier, E.E. van Dyk, F.J. Vorster Nelson Mandela Metropolitan University Electroluminescence (EL) is a useful

### Laser Frequency Stabilization with Optical Cavities. Abstract

Laser Frequency Stabilization with Optical Cavities Anya M. Davis Walla Walla University University of Washington Quantum Computing with Trapped Ions Group INT REU 2009 (Dated: September 17, 2009) Abstract

### Strain gauge lab Part 1

Strain gauge lab Part 1 In this lab, you will make measurements of mechanical strain in small aluminum beams as you bend them. We will continue this lab next week, so you will want to be careful to maintain

### Lab 5: Conservation of Energy

Lab 5: Conservation of Energy Equipment SWS, 1-meter stick, 2-meter stick, heavy duty bench clamp, 90-cm rod, 40-cm rod, 2 double clamps, brass spring, 100-g mass, 500-g mass with 5-cm cardboard square

### Differential and bidirectional differential pressure sensors, taking AMS 5812 as an example

In pressure sensor technology there is a difference between various physical methods of measurement, depending on the application. These include the measurement of absolute pressure, relative pressure,

### Lab 8: Ballistic Pendulum

Lab 8: Ballistic Pendulum Equipment: Ballistic pendulum apparatus, 2 meter ruler, 30 cm ruler, blank paper, carbon paper, masking tape, scale. Caution In this experiment a steel ball is projected horizontally

### Project 2B Building a Solar Cell (2): Solar Cell Performance

April. 15, 2010 Due April. 29, 2010 Project 2B Building a Solar Cell (2): Solar Cell Performance Objective: In this project we are going to experimentally measure the I-V characteristics, energy conversion

### THERMAL ANEMOMETRY ELECTRONICS, SOFTWARE AND ACCESSORIES

TSI and TSI logo are registered trademarks of TSI Incorporated. SmartTune is a trademark of TSI Incorporated. THERMAL ANEMOMETRY ELECTRONICS, SOFTWARE AND ACCESSORIES IFA 300 Constant Temperature Anemometry

### UEI App Notes: Using Accelerometers in a Data Acquisition System

UEI App Notes: Using Accelerometers in a Data Acquisition System by Bob Judd Director of Marketing 1 Using Accelerometers in a Data Acquisition System Accelerometers are widely used in industry for measuring

### Physics 9e/Cutnell. correlated to the. College Board AP Physics 1 Course Objectives

Physics 9e/Cutnell correlated to the College Board AP Physics 1 Course Objectives Big Idea 1: Objects and systems have properties such as mass and charge. Systems may have internal structure. Enduring

### HLC2701 Encoder Detector

HLC2701 FEATURES Side-looking plastic package TTL/LSTTL/CMOS compatible Inverting logic output Linear or rotary encoder applications Resolution to 0.009 in.(.229 mm) Sensitivity versus temperature compensation

### The Influence of Magnetic Forces on the Stability Behavior of Large Electrical Machines

The Influence of Magnetic Forces on the Stability Behavior of Large Electrical Machines Dr. H. Sprysl, Dr. H. Vögele, ABB Kraftwerke AG, CH-5242 Birr Dr. G. Ebi, SENSOPLAN GmbH, D-79801 Hohentengen a.h.

### Module 13 : Measurements on Fiber Optic Systems

Module 13 : Measurements on Fiber Optic Systems Lecture : Measurements on Fiber Optic Systems Objectives In this lecture you will learn the following Measurements on Fiber Optic Systems Attenuation (Loss)

### EQUILIBRIUM AND ELASTICITY

Chapter 12: EQUILIBRIUM AND ELASTICITY 1 A net torque applied to a rigid object always tends to produce: A linear acceleration B rotational equilibrium C angular acceleration D rotational inertia E none

### Technical Datasheet Scalar Network Analyzer Model 8003-10 MHz to 40 GHz

Technical Datasheet Scalar Network Analyzer Model 8003-10 MHz to 40 GHz The Giga-tronics Model 8003 Precision Scalar Network Analyzer combines a 90 db wide dynamic range with the accuracy and linearity

### Holographically corrected telescope for high bandwidth optical communications (as appears in Applied Optics Vol. 38, No. 33, , 20 Nov.

Holographically corrected telescope for high bandwidth optical communications (as appears in Applied Optics Vol. 38, No. 33, 6833-6835, 20 Nov. 1999) Geoff Andersen and R. J. Knize Laser and Optics Research

### In-situ Load Testing to Evaluate New Repair Techniques

In-situ Load Testing to Evaluate New Repair Techniques W.J. Gold 1 and A. Nanni 2 1 Assistant Research Engineer, Univ. of Missouri Rolla, Dept. of Civil Engineering 2 V&M Jones Professor, Univ. of Missouri

Week 8 homework IMPORTANT NOTE ABOUT WEBASSIGN: In the WebAssign versions of these problems, various details have been changed, so that the answers will come out differently. The method to find the solution

### Thermal analysis and measurements of spiral inductors using built-in and IR sensors

Thermal analysis and measurements of spiral inductors using built-in and IR sensors by M. Kałuża*, M. Felczak, and B. Więcek * Institute of Electronics, Lodz University of Technology, Poland, marcin.kaluza@p.lodz.pl

### Electrical Resonance

Electrical Resonance (R-L-C series circuit) APPARATUS 1. R-L-C Circuit board 2. Signal generator 3. Oscilloscope Tektronix TDS1002 with two sets of leads (see Introduction to the Oscilloscope ) INTRODUCTION

### Rotation: Moment of Inertia and Torque

Rotation: Moment of Inertia and Torque Every time we push a door open or tighten a bolt using a wrench, we apply a force that results in a rotational motion about a fixed axis. Through experience we learn

### Physics 1020 Laboratory #6 Equilibrium of a Rigid Body. Equilibrium of a Rigid Body

Equilibrium of a Rigid Body Contents I. Introduction II. III. IV. Finding the center of gravity of the meter stick Calibrating the force probe Investigation of the angled meter stick V. Investigation of

### Prelab Exercises: Hooke's Law and the Behavior of Springs

59 Prelab Exercises: Hooke's Law and the Behavior of Springs Study the description of the experiment that follows and answer the following questions.. (3 marks) Explain why a mass suspended vertically

### Physics 2A, Sec B00: Mechanics -- Winter 2011 Instructor: B. Grinstein Final Exam

Physics 2A, Sec B00: Mechanics -- Winter 2011 Instructor: B. Grinstein Final Exam INSTRUCTIONS: Use a pencil #2 to fill your scantron. Write your code number and bubble it in under "EXAM NUMBER;" an entry

### INFRARED PARTS MANUAL

INFRARED PARTS MANUAL PIR325 FL65 GLOLAB CORPORATION Thank you for buying our Pyroelectric Infrared components. The goal of Glolab is to produce top quality electronic kits, products and components. All

### Transmission through the quadrupole mass spectrometer mass filter: The effect of aperture and harmonics

Transmission through the quadrupole mass spectrometer mass filter: The effect of aperture and harmonics A. C. C. Voo, R. Ng, a) J. J. Tunstall, b) and S. Taylor Department of Electrical and Electronic

### Detailed simulation of mass spectra for quadrupole mass spectrometer systems

Detailed simulation of mass spectra for quadrupole mass spectrometer systems J. R. Gibson, a) S. Taylor, and J. H. Leck Department of Electrical Engineering and Electronics, The University of Liverpool,

### Rotation, Rolling, Torque, Angular Momentum

Halliday, Resnick & Walker Chapter 10 & 11 Rotation, Rolling, Torque, Angular Momentum Physics 1A PHYS1121 Professor Michael Burton Rotation 10-1 Rotational Variables! The motion of rotation! The same

### IMPACT REGISTER, INC. PRECISION BUILT RECORDERS SINCE 1914

IMPACT REGISTER, INC. PRECISION BUILT RECORDERS SINCE 1914 G ENVIRONMENTAL MINI-MONITOR GEMM DESCRIPTION THE G ENVIRONMENTAL MINI-MONITOR GEMM, IS THE SMALLEST, LIGHTEST WEIGHT, COST EFFECTIVE, ELECTRONICALLY

### Tutorial: Calculation of a planetary gear train

Tutorial: Calculation of a planetary gear train This tutorial shows the usage of MESYS shaft calculation with shaft systems. A two-stage planetary gear stage is defined using the program. Please start