2 What is mass spectrometry? A versatile analytical technique used to determine the composition of chemical samples either at the atomic or molecular level. Ionic forms of samples are separated according to differences in component massto-charge (m/z) ratio. Signals from each different isotopic composition Isotopes usually stable Keep in mind charge, z, in (+1, +2, +3, etc.); not knowing z can result in mass errors by a factor of 2, 3, etc. Dichloromethane CH 2 Cl 2 12 C 1 H 2 35 Cl 2 = C 1 H 2 35 Cl 2 = C 1 H 2 35 Cl 37 Cl = C 1 H 2 35 Cl 37 Cl = C 1 H 2 37 Cl 2 = 88 Isotope % Natural Abundance 12 C C 1.07 } 35 Cl Cl m/z Textbook value is a weighted average Cl At. wt. =
3 Basic instrumental setup for mass spectrometry Molecular mass spectrometry Chapter 20 Fig
4 Generating ions for molecular mass spectrometry Two major classes Gas phase: Sample vaporized then ionized Desorption: Solid or liquid sample is directly converted into gas phase ions * * * * *
5 Molecular mass spec ion source considerations Gas-phase Thermally stable compounds with bp less than 500 o C Limited to masses less than 1000 Da (atomic mass unit; amu) Desorption Does not require volatilization Analytes up to 100,000+ Da Hard vs. soft sources Hard sources leave molecule in excited energy states which relax via bond cleavage. Give daughter ion fragments at lower m/z. Soft sources minimize fragmentation. Resulting spectra has fewer peaks.
6 The electron impact source Gas phase source: classical method still used today Sample is thermally vaporized and bombarded with beam of electrons Electrons are expelled from molecule due to electrostatic repulsion Result: singly charged positive ions (primarily) Fig M + e- M + + 2e-
7 The electron impact source, continued Highly inefficient process and often gives complex spectra with many fragmented species Daughter ions But, good sensitivity Most common ion source in (small) molecular mass spec 1-pentanol Fig. 20-4
8 Chemical ionization sources Gas phase source; M = molecule of interest Gas phase atoms of the sample are collided with ions generated from electron bombardment of a reagent gas Reagent gas is often CH 4, which is ionized to CH 4+, CH 3+, and CH 2+. CH 4+ + CH 4 CH 5+ + CH 3 CH 3+ + CH 4 C 2 H 5+ + H 2 Ionization of sample typically occurs via proton transfer (H + ) or hydride transfer (hydride is H - ) CH 5+ + MH MH 2+ + CH 4 C 2 H 5+ + MH MH 2+ + C 2 H 4 C 2 H 5+ + MH M + + C 2 H 6
9 Comparison of electron impact and chemical ionization sources Electron impact is a much harder ionization method. Chemical ionization is softer but still classifies as a hard method. Fragmentation can be an advantage or disadvantage; can add certainty or uncertainty. Both are limited to volatile and thermallystable compounds Electron impact Both are MS of 1-decanol Chemical ionization -OH
10 Field desorption Usually a soft ionization method Multi-tipped electrode is coated with sample solution (solid or liquid phase, not gas) Electrode is inserted into compartment and a high voltage is applied, and maybe heat The sample ionizes and goes into the gas phase by electrostatic repulsion; it desorbs from the surface it was on
11 Hard vs. Soft ionization Mass spectra of glutamic acid via different ionization sources Electron impact Field desorption Fig. 20-6
12 Matrix-assisted laser desorption-ionization (MALDI) Demonstrated almost simultaneously in 1988 by German and Japanese research groups Hillenkamp & Karas (Germany) and Tanaka (Japan) Tanaka won the 2002 NP Sample is dispersed in matrix (usually a crystalline solid) and exposed to laser beam Matrix material absorbs radiation and explodes producing desorbed and ionized analyte molecules. Can analyze very large molecules without fragmenting; SOFT ionization Fig
13 Matrix Assisted Laser Desorption/Ionization Pulsed Nd:YAG Laser Microchannel plate detectors Matrix/Analyte Crystals Extraction Ion Optics Reflectron Dr. Eric Monroe Bruker
14 Electrospray ionization (ESI) Desorption method First described in 1984 John Fenn (Virgina Commonwealth University) awarded 2002 Nobel Prize in Chemistry With Koichi Tanaka (Shimadzu) for soft laser desorption (precursor to MALDI) Extremely useful for characterizing large biopolymers (oligonucleotides, peptides, proteins, etc.) having masses greater than 100,000 Da. Also very useful for inorganic and synthetic polymer research.
15 esi.html Electrospray ionization (ESI): How it works Sample (in solution) passed through needle biased at several kv Charged droplets pass through capillary where solvent evaporates and molecules begin to inherit the droplets charge Droplets shrink until their surface tension can no longer support the charge density Rayleigh limit Coulombic explosion occurs and the droplet bursts into smaller droplets that repel each other Process repeats until all of the solvent in gone and analytes are multiply charged. A very big deal because it couples flowing, liquid samples to mass spec!
16 ESI: A few more images
17 Electrospray ionization (ESI), continued Multiply charged analytes have low m/z ratios Typically less than 1500; interfaces with quadrupole mass analyzer; more later Molecular mass can be determined from isotopic peak distribution (back calculation) No fragmentation disadvantage in some applications Use MS/MS (mass spec followed by mass spec again) to gain more structural knowledge of the molecule Readily interfaces with solution-phase Takes place at atmospheric pressures and temps (RT) Liquid chromatography and capillary electrophoresis
18 ESI-MS Data Note multiple charges and low m/z ratios Large, complex, complete molecules MW = 1060 Da MW = 14,700 Da antipsychotic and antihypertensive g/mol
19 Desorption Electrospray Ionization (DESI) Method developed by Prof. Graham Cooks (Purdue) Mass Spectrometry Sampling Under Ambient Conditions with Desorption Electrospray Ionization, Science, 2004, 306, A fine spray of charged droplets hits the surface of interest, from which it picks up small organic molecules and large biomolecules, ionizes them, and delivers them as desolvated ions into the mass spectrometer Now electrospray can be applied to surfaces and interfaces!!! R.G. Cooks, Z. Ouyang, Z. Takats, and J.M. Wiseman, Science, 2006, 311,
20 After generating ions Sorting ions by mass Detection of ions Single-channel detectors Faraday cup Electron multiplier Array transducers for ions Microchannel plates Micro-Faraday array microfabricated array of Faraday cups
21 Faraday cup The charge from positive ions striking plate is neutralized by electrons supplied from ground by external circuit via a load resistor. The voltage dropped across the resistor can be amplified and is proportional to the number of ions that struck the collector. The Faraday cage eliminates reflections of ions or secondary electrons ejected from collector. Response is independent of energy, mass, chemistry, etc. A universal charge detector. Inexpensive and simple design. High impedance amplifier limits time resolution. But, not as sensitive as electron multiplier (next). cations in Fig e - in to neutralize entering cations
22 lectron Multipliers
23 Electron multipliers Similar to photomultiplier tubes used in optical detection Ions strike cathode generating a burst of electrons that are subsequently amplified by a series of dynodes. Multipliers with ~20 dynodes can give gains of Rugged, reliable, high-current gains with fast response times (nsec) Used for most routine mass spectrometry applications Continuous-dynode multiplier Discrete multiplier; individual dynodes Fig. 11-2
24 Microchannel Plate detector 18mm An MCP detector is like placing 1,000s of electron multipliers in parallel right next to each other. They are typically used in time-of-flight instruments because of their exceptional speed and sensitivity.
25 Microchannel plate an array detector for ions Electro-optical ion detector (electrons and light) Analogous to optical detectors, detector arrays allow multiple ion trajectories (different ions) to be monitored simultaneously The utility of this type of detector will be discussed later Think diode array versus single diode Fig Individual electron multipliers: 1000-fold!
26 Mass analyzers Sort ions based upon m/z ratio Exert forces upon ions in order to differentiate m/z ratios Ions can be manipulated by electric and magnetic fields. Ions that traverse a magnetic field have a bent trajectory Efficiency of ion sorting described by resolution: Common types Magnetic sector Double-focusing Time-of-flight Quadrupole Quadrupole ion trap FT-Ion cyclotron resonance Orbitrap R m m m is the mass difference between the two peaks m is the average mass of the peaks Needed resolution varies dramatically based upon application!!! Better R often means more $ or more complex instrumentation!
27 Magnetic sector analyzers Ions are injected into a circular path (electro)magnet and only those of a certain m/z ratio are able to balance the centripetal (F C ) and magnetic forces (F M ) in successfully traverse the path; others hit the sides. Mass selection is achieved by sweeping the field, B (more typical) or V. Resolution mv KE = zev = ½ mv 2 FM Bzev Fc r 2 mv Bzer Bzev v r m 1 2 KE zev mv 2 Fig Assumption: all ions have same KE after acceleration through Slit B m z 2 2 B r e 2V B=magnetic field strength e=electron charge v=velocity V=accelerating voltage r = radius m = mass
28 Limitations of magnetic sector Not all ions of same m/z have identical kinetic energy A Boltzmann distribution emerges from ion source Causes broadening in m/z transmitted to detector Solution: Use a second focusing element before magnetic sector double focusing Kind of like a pre-filter Two bent, electrostatic plates; at constant dc potential
29 Double focusing analyzer Ions are first passed through electrostatic energy filter to isolate a single kinetic energy Resolution 100,000! Neir-Johnson design Single channel measurements Early design Mattauch-Herzog configuration Better for array detectors that capture entire mass spectra
30 Time-of-flight mass analyzer (TOF) Ions are generated in pulses and accelerated (also in pulses, but with a lag) in a V field into a field-free drift tube All ions (theoretically) have the same kinetic energy so higher mass species will traverse the drift tube more slowly (big mass = slow v) 1 2 KE mv 2 Resolution better than, but sensitivity (quantitation) not as good as quadrupoles (next). Advantages: simplicity, ruggedness, virtually unlimited range, and fast data acquisition (entire mass spectrum collected simultaneously) t F L m 2Vez t F =flight time L=distance from source to detector V=accelerating voltage
31 Reflectron configuration: a variation on standard TOF Faster ions penetrate more deeply into electrostatic reflector and thus travel a longer distance. Correcting for KE distribution from ion source Result: Faster and slower ions of same mass arrive at the detector simultaneously. Improved resolution but lower sensitivity: more components = more lost molecules
32 Quadrupole mass analyzer Mass isolation is achieved with alternating applied electric fields DC voltage is applied to rods opposite one another and AC voltage is superimposed. The same thing, but opposite sign of applied voltage on the other two rods Important: The DC and AC voltages are both essential and are always controlled and scanned together. The repulsion (or not) of an ion is dependent upon its velocity which tracks with m/z. Limited, but tunable, mass range passes. A ions focused toward z-axis during + half of AC cycle B ions attracted to x rods during half of AC cycle
33 Quadrupole mass analyzer, continued Only ions having a specific m/z ratio are confined within and traverse the entire quadrupole without becoming unstable and colliding with a rod. Most commonly used mass analyzers today for a range of applications More compact, less expensive, and more rugged than TOF instruments better quantitation! Good resolution (R ~ 1 amu) Much less than TOF Fast! 100-ms scans, full mass spectra Nominal mass range is up to 1500 m/z Best go up to m/z. Four parallel electrode rods
34 Putting it all together: Helpful video Small molecule mass spectrometry Ionization, quadrupole mass analyzer, and detector
35 Quadrupole ion trap Can store ions of particular m/z ranges and selectively eject particular m/z ions for up to 15 minutes, for some species Rugged, compact, and cheap, but limited dynamic range and accuracy in m/z measurement. Useful for MS/MS and other applications where one wants to store or accumulate ions for subsequent analysis.
36 Linear ion trap Rods are segmented to allow independent electrical contact. Potential well created in middle traps ions Higher capacity than standard quadrupole ion trap
37 Tandem mass spectrometry selection A pre-selected ion is fragmented and the pieces analyzed again by mass spectrometry. Parent ion (M n+ ) fragmented into product ions Fragmented via: Electron capture dissociation Collisions with gas (often CH 4 ) Photo-induced dissociation (w/laser) Valuable tool for identification of species with the same parent ion mass because the fragments will differ Soft ion sources preferable for MS step 1 Harder sources preferable for MS step 2 usually soft M + m/z = 279 M + m/z = 279 usually hard(er) Fig Fig
38 Types of MS/MS instrumentation Tandem-in-space MS/MS Often multiple quadrupoles that can act both as mass analyzers and collision chambers Image below Tandem-in-time MS/MS Collision is done inside of mass analyzer FT-ICR and quadrupole ion traps are ideally suited Fig A triple quadrupole mass spec!
39 A few other helpful videos LC-interfaced triple quad MS/MS LC-interfaced qtof
41 Applications of molecular mass spectrometry Versatile detection technology Not only detects, but provides additional detail in the detection process. Qualitative and quantitative applications Identification of compounds and determination of purity Hyphenated techniques GC/MS, LC/MS, CE/MS, GC/MS/MS, LC/MS/MS, and LC/MS n
42 Traditional MS/MS-based proteomics Proteomics is the large-scale study of proteins, their structure and function 1) Sample collection and 1-D PAGE for rough separation 2) Tryptic digest to yield peptides (more manageable). Whole proteins cannot be unequivocally identified by mass alone. 3) LC purification and ESI injection 4) MS spectrum obtained and peptide composition can be identified, but not sequenced 5) MS/MS fragment analysis allows the linear sequence to be reconstructed Replaced by LC after digestion Nature, 2003, 422,
43 Types of peptide fragments
44 Common non-proteolytic fragmentation methods Collision activated dissociation (CAD) Collision-induced dissociation (CID) Fragments at amide bonds giving b and y ions Infrared multiphoton dissociation (IRMPD) Fragments at amide bonds giving b and y ions Electron capture dissociation (ECD) Low energy electrons used to fragment peptide often at backbone giving c and z ions Mainly used on FT- instruments Electron transfer dissociation (ETD) Radical anions (antracene and azobenzene) used to fragment molecular ions (not direct fragmentation) giving c and z ions Also applicable to simpler quadrupole instruments Both are very useful for studying posttranslational protein modifications because they only cleave the backbone of the polypeptide
45 Proteomics: Protein and peptide sequencing using MS/MS Single-letter amino acid designations Fragments obtained by proteolytic digestion b n is the left fragment y m is the right fragment MS/MS b n + y m = 775 = MW From Introduction to Proteomics: Tools for the New Biology, by Daniel C. Liebler, 2002, Humana Press
46 Proteomics: Protein and peptide sequencing using MS/MS The bottom-up approach to analysis: analyze the proteolytic pieces with MS/MS Problem: Often only partial peptide sequence coverage is achieved From Introduction to Proteomics: Tools for the New Biology, by Daniel C. Liebler, 2002, Humana Press
Aiping Lu Key Laboratory of System Biology Chinese Academic Society APLV@sibs.ac.cn Proteome and Proteomics PROTEin complement expressed by genome Marc Wilkins Electrophoresis. 1995. 16(7):1090-4. proteomics
13 C NMR Spectroscopy Introduction Nuclear magnetic resonance spectroscopy (NMR) is the most powerful tool available for structural determination. A nucleus with an odd number of protons, an odd number
Mass Spectrometry Based Proteomics Proteomics Shared Research Oregon Health & Science University Portland, Oregon This document is designed to give a brief overview of Mass Spectrometry Based Proteomics
Electrospray Ion Source Electrospray Ion Trap Mass Spectrometry Introduction The key to using MS for solutions is the ability to transfer your analytes into the vacuum of the mass spectrometer as ionic
Advantages of Using Triple Quadrupole over Single Quadrupole Mass Spectrometry to Quantify and Identify the Presence of Pesticides in Water and Soil Samples André Schreiber AB SCIEX Concord, Ontario (Canada)
What is ICP-MS? and more importantly, what can it do? Inductively Coupled Plasma Mass Spectrometry or ICP-MS is an analytical technique used for elemental determinations. The technique was commercially
AMD Analysis & Technology AG Application Note 120419 Author: Karl-Heinz Maurer APCI-MS Trace Analysis of volatile organic compounds in ambient air A) Introduction Trace analysis of volatile organic compounds
Laboration 1 Identifiering av proteiner med Mass Spektrometri Klinisk Kemisk Diagnostik Sven Kjellström 2014 firstname.lastname@example.org 0702-935060 Laboration 1 Klinisk Kemisk Diagnostik Identifiering av
Mass Spectrometry Overview Mass Spectrometry is an analytic technique that utilizes the degree of deflection of charged particles by a magnetic field to find the relative masses of molecular ions and fragments.2
The CHROMacademy Essential Guide Understanding GC-MS Analysis Part 1 Speakers John Hinshaw GC Dept. Dean CHROMacademy Tony Taylor Technical Director Crawford Scientific Moderator M ( g ) e M ( g ) 2e Peter
Summary of the characteristics of different mass analyzers All mass spectrometers combine ion formation, mass analysis, and ion detection. This discussion is concerned with how various mass analyzers are
Reiner Westermeier, Torn Naven Hans-Rudolf Höpker Proteomics in Practice A Guide to Successful Experimental Design 2008 Wiley-VCH Verlag- Weinheim 978-3-527-31941-1 Preface Foreword XI XIII Abbreviations,
TOF FUNDAMENTALS TUTORIAL Presented By: JORDAN TOF PRODUCTS, INC. 990 Golden Gate Terrace Grass Valley, CA 95945 530-272-4580 / 530-272-2955 [fax] www.rmjordan.com [web] email@example.com [e-mail] This
Pesticide Analysis by Mass Spectrometry Purpose: The purpose of this assignment is to introduce concepts of mass spectrometry (MS) as they pertain to the qualitative and quantitative analysis of organochlorine
THE BASICS OF MASS SPECTROMETRY IN THE TWENTY- FIRST CENTURY Gary L. Glish* and Richard W. Vachet Enormous advances in our understanding of the chemistry underlying life processes have identified new targets
Application Note # LCMS-81 Introducing New Proteomics Acquisiton Strategies with the compact Towards the Universal Proteomics Acquisition Method Introduction During the last decade, the complexity of samples
LC-MS/MS for Chromatographers An introduction to the use of LC-MS/MS, with an emphasis on the analysis of drugs in biological matrices LC-MS/MS for Chromatographers An introduction to the use of LC-MS/MS,
AB SCIEX TOF/TOF 4800 PLUS SYSTEM Cost effective flexibility for your core needs AB SCIEX TOF/TOF 4800 PLUS SYSTEM It s just what you expect from the industry leader. The AB SCIEX 4800 Plus MALDI TOF/TOF
Electrospray Mass Spectrometry of Tris-acetylacetonatochromium(III) Purpose: Verify the synthesis of the synthesis product, Cr(acac) 3. The presence of any urea containing byproducts will also be determined.
Accurate calibration of on-line Time of Flight Mass Spectrometer (TOF-MS) for high molecular weight combustion product analysis B. Apicella*, M. Passaro**, X. Wang***, N. Spinelli**** firstname.lastname@example.org
1 Gas Chromatography/Mass Spectroscopy (GC/MS/MS) Background Information Instructions for the Operation of the Varian CP-3800 Gas Chromatograph/ Varian Saturn 2200 GC/MS/MS See the Cary Eclipse Software
Introduction to mass spectrometry (MS) based proteomics and metabolomics Tianwei Yu Department of Biostatistics and Bioinformatics Rollins School of Public Health Emory University September 10, 2015 Background
GENERAL UNKNOWN SCREENING FOR DRUGS IN BIOLOGICAL SAMPLES BY LC/MS Luc Humbert, Michel Lhermitte, Frederic Grisel Laboratoire de Toxicologie & Génopathologie, CHRU Lille, France Waters Corporation, Guyancourt,
In-Depth Qualitative Analysis of Complex Proteomic Samples Using High Quality MS/MS at Fast Acquisition Rates Using the Explore Workflow on the AB SCIEX TripleTOF 5600 System A major challenge in proteomics
20 Principles and Applications of LC-MS/MS for the Quantitative Bioanalysis of Analytes in Various Biological Samples Ju-Seop Kang Department of Pharmacology & Clinical Pharmacology Laboratory, College
ProteinScape Proteomics Data Analysis & Management Innovation with Integrity Mass Spectrometry ProteinScape a Virtual Environment for Successful Proteomics To overcome the growing complexity of proteomics
Time-of-Flight Mass Spectrometry Technical Overview Introduction Time-of-flight mass spectrometry (TOF MS) was developed in the late 1940 s, but until the 1990 s its popularity was limited. Recent improvements
CHE334 Identification of an Unknown Compound By NMR/IR/MS Purpose The object of this experiment is to determine the structure of an unknown compound using IR, 1 H-NMR, 13 C-NMR and Mass spectroscopy. Infrared
Sputtering Vacuum Evaporation Recap Use high temperatures at high vacuum to evaporate (eject) atoms or molecules off a material surface. Use ballistic flow to transport them to a substrate and deposit.
Università degli Studi di Bari Corso di Laurea Magistrale in Scienze Chimiche Corso di SPETTROMETRIA DI MASSA II semestre, 6 + 1 crediti (48 + 15 ore) Prof. Ilario Losito E-mail: email@example.com Sito
CHEMISTRY STANDARDS BASED RUBRIC ATOMIC STRUCTURE AND BONDING Essential Standard: STUDENTS WILL UNDERSTAND THAT THE PROPERTIES OF MATTER AND THEIR INTERACTIONS ARE A CONSEQUENCE OF THE STRUCTURE OF MATTER,
Functional Data Analysis of MALDI TOF Protein Spectra Dean Billheimer firstname.lastname@example.org. Department of Biostatistics Vanderbilt University Vanderbilt Ingram Cancer Center FDA for MALDI TOF
Mass spectrometry What are the principles behind MS? What do all MS instruments have in common? What are the different types of MS? Lecture outline: 1) Introduction to mass spectrometry 2) sample introduction
Application Notes # LCMS-35 esquire series Application of LC/APCI Ion Trap Tandem Mass Spectrometry for the Multiresidue Analysis of Pesticides in Water An LC-APCI-MS/MS method using an ion trap system
NUVISAN Pharma Services CESI MS Now available! 1st CRO in Europe! At the highest levels of quality. LABORATORY SERVICES Equipment update STATE OF THE ART AT NUVISAN CESI MS Now available! 1st CRO in Europe!
Guide to Reverse Phase SpinColumns Chromatography for Sample Prep www.harvardapparatus.com Contents Introduction...2-3 Modes of Separation...4-6 Spin Column Efficiency...7-8 Fast Protein Analysis...9 Specifications...10
MultiQuant Software 2.0 for Targeted Protein / Peptide Quantification Gold Standard for Quantitative Data Processing Because of the sensitivity, selectivity, speed and throughput at which MRM assays can
NANOCOMPOSIX'S GUIDE TO ICP-MS MEASUREMENT AND ANALYSIS SEPTEMBER 2012, V 1.1 4878 RONSON CT STE K SAN DIEGO, CA 92111 858-565 - 4227 NANOCOMPOSIX.COM Note to the Reader: We at nanocomposix have published
Protons, neutrons and electrons Nuclear Structure particle relative charge relative mass proton 1 1 atomic mass unit neutron 0 1 atomic mass unit electron -1 negligible mass Protons and neutrons make up
V Contents List of Contributors XIII Introduction 1 Christopher Barner-Kowollik, Jana Falkenhagen, Till Gruendling, and Steffen Weidner References 4 1 Mass Analysis 5 Gene Hart-Smith and Stephen J. Blanksby
The Scheduled MRM Algorithm Enables Intelligent Use of Retention Time During Multiple Reaction Monitoring Delivering up to 2500 MRM Transitions per LC Run Christie Hunter 1, Brigitte Simons 2 1 AB SCIEX,
Ma B. Challenges in computational analysis of mass spectrometry data for proteomics. SCIENCE AND TECHNOLOGY 25(1): 1 Jan. 2010 JOURNAL OF COMPUTER Challenges in Computational Analysis of Mass Spectrometry
Application Note # LCMS-62 Walk-Up Ion Trap Mass Spectrometer System in a Multi-User Environment Using Compass OpenAccess Software Abstract Presented here is a case study of a walk-up liquid chromatography
Increasing the Multiplexing of High Resolution Targeted Peptide Quantification Assays Scheduled MRM HR Workflow on the TripleTOF Systems Jenny Albanese, Christie Hunter AB SCIEX, USA Targeted quantitative
Tutorial for Proteomics Data Submission Katalin F. Medzihradszky Robert J. Chalkley UCSF Why Have Guidelines? Large-scale proteomics studies create huge amounts of data. It is impossible/impractical to
Using Natural Products Application Solution with UNIFI for the Identification of Chemical Ingredients of Green Tea Extract Lirui Qiao, 1 Rob Lewis, 2 Alex Hooper, 2 James Morphet, 2 Xiaojie Tan, 1 Kate
Atoms, Ions and Molecules The Building Blocks of Matter Chapter 2 1 Chapter Outline 2.1 The Rutherford Model of Atomic Structure 2.2 Nuclides and Their Symbols 2.3 Navigating the Periodic Table 2.4 The
Ionization Methods in Organic Mass Spectrometry Contents:. Introduction. Gas-Phase Ionization. Electron Ionization (EI). Chemical Ionization (CI). Desorption Chemical Ionization (DCI). Negative-ion chemical
Exam Master course KEMM03 Principles of Mass Spectrometric Protein Characterization 2010-10-29 kl 08.15-13.00 Use a new paper for answering each question! Write your name on each paper! Aids: Mini calculator,
High resolution mass spectrometry (HRMS*) in Graz 10.2.2012 * Instruments with high resolution where exact mass measurements can be performed 1 HRMS in Graz s and Ionization Techniques 2 6 3 5 4 1 TU Graz
12.3.2 Types of ions Adduct ion An ion formed by interaction of two species, usually an ion and a molecule, and often within an ion source, to form an ion containing all the constituent atoms of one species
Week 9: MS in Space and Proteomics 1 Last Time Detectors Small Molecule Applications, Environmental: (e.g. TWQC) 2 Mass Spectrometry in Space Possibly the coolest application of small molecule MS is in
Investigating the use of the AB SCIEX TripleTOF 4600 LC/MS/MS System for High Throughput Screening of Synthetic Cannabinoids/Metabolites in Human Urine AB SCIEX MPX -2 High Throughput TripleTOF 4600 LC/MS/MS
Mass Spectrometry @ the Organic hemistry Department (A guide for novel users) Peter M. van Galen Research Assistant Mass Spectrometry Organic hemistry Department Nijmegen University September 2005 0 Introduction...
API 3200 LC/MS/MS SYSTEM Performance, productivity and value combined API 3200 LC/MS/MS SYSTEM Reliability, reproducibility, and confidence in your data. With an unmatched heritage of technological innovation
A Reference Measurement System for C-reactive Protein David M. Bunk, Ph.D. Chemical Science and Technology Laboratory National Institute of Standards and Technology Definition of the Measurand: Human C-reactive
Classical and special AMD Magnetic Sector mass spectrometers (extracted from power point presentations) High Resolution Mass Spectromter AMD 402/403 S 6 High Resolution Mass Spectrometer AMD 604 S 8 Version
JOURNAL OF MASS SPECTROMETRY, VOL. 32, 263È276 (1997) SPECIAL FEATURE: TUTORIAL High-resolution Mass Spectrometry and Accurate Mass Measurements with Emphasis on the Characterization of Peptides and Proteins
Increasing Quality While Maintaining Efficiency in Drug Chemistry with DART-TOF MS Screening Application Note Forensics Author Erin Shonsey Director of Research Alabama Department of Forensic Science Abstract
OVERVIEW Exact mass LC/MS analysis using an orthogonal acceleration time of flight (oa-tof) mass spectrometer is a well-established technique with a broad range of applications. These include elemental
DISSERTATION ZUR ERLANGUNG DES DOKTORGRADES DER FAKULTÄT FÜR CHEMIE UND PHARMAZIE DER LUDWIG-MAXIMILIANS-UNIVERSITÄT MÜNCHEN Overcoming Challenges of Shotgun Proteomics von Annette Michalski aus Hildesheim
Cambridge Isotope Laboratories, Inc. www.isotope.com Proteomics Mass Spectrometry Signal Calibration for Protein Quantitation Michael J. MacCoss, PhD Associate Professor of Genome Sciences University of
SimGlycan Software*: A New Predictive Carbohydrate Analysis Tool for MS/MS Data Automated Data Interpretation for Glycan Characterization Jenny Albanese 1, Matthias Glueckmann 2 and Christof Lenz 2 1 AB
Agilent 6000 Series LC/MS Solutions CONFIDENCE IN QUANTITATIVE AND QUALITATIVE ANALYSIS AGILENT 6000 SERIES LC/MS SOLUTIONS UNRIVALED ACHIEVEMENT AT EVERY LEVEL The Agilent 6000 Series of LC/MS instruments
Annu. Rev. Biochem. 2001. 70:437 73 Copyright c 2001 by Annual Reviews. All rights reserved ANALYSIS OF PROTEINS AND PROTEOMES BY MASS SPECTROMETRY Matthias Mann, 1,2 Ronald C. Hendrickson, 2 and Akhilesh
World Leader in MALDI-QIT-TOF MS Technology Matrix Assisted Laser Desorption/Ionization (MALDI) technology achieves high efficiency ionization by mixing the sample to be analyzed with a laser energy-absorbing
2015 Page 2 Waters Core Chromatography Training (2 Days) The learning objective of this two day course is to teach the user core chromatography, system and software fundamentals in an outcomes based approach.
Ion/Molecule Reactions for Detecting Ammonia Using Miniature Cylindrical Ion Trap Mass Spectrometers Jonell N. Smith, a Adam D. Keil, b Robert J. Noll,* a and R. Graham Cooks a a Department of Chemistry,
Atoms, Molecules and Stoichiometry I. The Atomic Structure A. The Nucleus (a) Rutherford Bombardment Experiment This experiment leaded to the proposal of nucleus in atom. RESULTS (1)Most of the particles
Chapter 21 Principles of Chromatography and Mass Spectrometry Problems 1,2,3,4,8,9,11,13 21-1 What is Chromatography Process to separate and purify compounds in a mixture by passing them trough a material
Chapter 13 Spectroscopy NMR, IR, MS, UV-Vis Main points of the chapter 1. Hydrogen Nuclear Magnetic Resonance a. Splitting or coupling (what s next to what) b. Chemical shifts (what type is it) c. Integration
Modern approaches to determination of toxic metals in marine environmental objects. Atomic absorption and inductively coupled plasma, advantages and disadvantages Atomic spectroscopy Atomic spectroscopy
1 Why? Chapter 1 Intermolecular Forces and Liquids Why is water usually a liquid and not a gas? Why does liquid water boil at such a high temperature for such a small molecule? Why does ice float on water?
Molecular Cell Biology Prof. D. Karunagaran Department of Biotechnology Indian Institute of Technology Madras Module 5 Methods in Cell Biology (Methods to Manipulate Protein, DNA and RNA and Methods to
Technical ote Simultaneous Metabolite Identification and Quantitation with UV Data Integration Using LightSight Software Version 2.2 Alek. Dooley, Carmai Seto, esham Ghobarah, and Elliott B. Jones verview:
Back to Basics Section D: Ion Optics CHAPTER D1 ION OPTICS OF MAGNETIC/ELECTRIC SECTOR MS TABLE OF CONTENTS QuickGuide...369 Summary...369 Preamble...371 MassAnalysisofIons...371 MagneticSector...371 ElectrostaticAnalyser(ElectricSector)...375
Introduction The Cathode Ray Tube or Braun s Tube was invented by the German physicist Karl Ferdinand Braun in 897 and is today used in computer monitors, TV sets and oscilloscope tubes. The path of the
Principles of Mass Spectrometry-Based Proteomics Steven Gygi Department of Cell Biology Harvard Medical School, Boston, MA The Flow of Information The Flow of Information Networks Complexes Proteins Genes
Characterization Techniques for Organic Compounds. When we run a reaction in the laboratory or when we isolate a compound from nature, one of our first tasks is to identify the compound that we have obtained.
Experiment #5: Qualitative Absorption Spectroscopy One of the most important areas in the field of analytical chemistry is that of spectroscopy. In general terms, spectroscopy deals with the interactions
Spectroscopy Biogeochemical Methods OCN 633 Rebecca Briggs Definitions of Spectrometry Defined by the method used to prepare the sample 1. Optical spectrometry Elements are converted to gaseous atoms or
Forensic Science Standards and Standard 1: Understands and applies principles of scientific inquiry Power : Identifies questions and concepts that guide science investigations Uses technology and mathematics
.1.1 Measure the motion of objects to understand.1.1 Develop graphical, the relationships among distance, velocity and mathematical, and pictorial acceleration. Develop deeper understanding through representations
Chapter 6 Photoionization Detectors The photoionization detector (PID) utilizes ultraviolet light to ionize gas molecules, and is commonly employed in the detection of volatile organic compounds (VOCs).
LC-MS: From history to the future From Kanalstrahlen to Top Down Identification of Biomolecules Gunnar Stenhagen Stenhagen Analyslab AB one of the main reasons for writing this book was the hope that it
Chemistry 321, Experiment 8: Quantitation of caffeine from a beverage using gas chromatography INTRODUCTION The analysis of soft drinks for caffeine was able to be performed using UV-Vis. The complex sample
3 - Atomic Absorption Spectroscopy Introduction Atomic-absorption (AA) spectroscopy uses the absorption of light to measure the concentration of gas-phase atoms. Since samples are usually liquids or solids,