# Case study: how to use cutoff conditions in a FRA frequency scan?

Save this PDF as:

Size: px
Start display at page:

Download "Case study: how to use cutoff conditions in a FRA frequency scan?"

## Transcription

1 NOVA Technical Note 8 Case study: how to use cutoff conditions in a FRA frequency scan? 1 Using cutoffs Cutoffs in FRA 1 The NOVA options can be used to test measured data points for a cutoff condition. Using cutoffs is quite straightforward for DC measurements (see Figure 1). Figure 1 The Edit Options window can be used to define cutoff conditions Note More information on the use of cutoffs can be found in the dedicated Cutoff tutorial, available from the Help menu in NOVA. On the other hand, FRA measurements in NOVA do not support the use of the Cutoff conditions as such. This technical note illustrates how to implement a cutoff condition during impedance measurements in NOVA by embedding a DC measurement into the FRA frequency scan. 1 This Technical Note applies to NOVA 1.10 or later. 1 Page

2 NOVA Technical note 8 Reminders: A cutoff condition can only be defined for a signal that is sampled 2. The cutoff condition must be met in order for the cutoff action to be triggered. 2 FRA impedance galvanostatic To illustrate this Case study, the default Autolab FRA impedance galvanostatic procedure will be used on the dummy cell (c) as a starting point (see Figure 2). In this example we assume that a cutoff condition which stops the complete procedure if a second signal, measured on the ADC164, drops below 0.6 V, is required during the FRA measurement. Figure 2 The Autolab FRA impedance galvanostatic procedure 2 Cutoffs can be defined on any signal available in the Signal Sampler. 2 Page

3 NOVA Technical Note Defining the DC current This procedure will be used with a DC current of 1 ma. Set the current value of the Set current command to (see Figure 3). Figure 3 Defining the DC current 2.2 Adding a Record signals to the FRA frequency scan Since a cutoff option can only be applied on a measured signal, it is necessary to add a Record signals (> 1 ms) command to the frequency scan. This command is used to measure the necessary signals and check for the cutoff condition. Locate the Record signals (> 1 ms) command, available in the Measurement chrono methods group and add it to the FRA frequency scan command, as shown in Figure 4. 3 Page

4 NOVA Technical note 8 Figure 4 Adding the Record signals (> 1 ms) to the FRA frequency scan For this example, the second signal, measured through the ADC164, needs to be sampled during 1 second, with an interval time of 0.1 s (see Figure 5). 4 Page

5 NOVA Technical Note 8 Figure 5 Changing the duration and the interval time for the Record signals (> 1 ms) command Since we only want to sample the second signal through the ADC164 in this example, the Signal sampler must also be adjusted (see Figure 6). 5 Page

6 NOVA Technical note 8 Figure 6 Adding the External(1).External 1 signal to the signal sampler After each single frequency defined in the frequency scan, the External(1).External 1 signal and Time will be sampled 10 times, using an interval time of 0.1 s. Plotting of the i vs t data obtained during this measurement is not required, and the i vs t plot located in the Record signal (> 1 ms) command can be deleted (see Figure 7). 6 Page

7 NOVA Technical Note 8 Figure 7 It is possible delete the i vs t plot generated by the Record signals command 2.3 Defining the cutoff condition The procedure is now ready to be used in combination with a cutoff condition on the External(1).External 1 signal. As described in the Cutoff tutorial, click the button next to the Options item in the Record signal (> 1 ms) command to open the Edit Options window (see Figure 8). 7 Page

8 NOVA Technical note 8 Figure 8 Opening the Options editor window for the Record signals (> 1 ms) command Note It is also possible to edit the options by clicking the Record signals (> 1 ms) command and clicking the icon in the quick access toolbar. 8 Page

9 NOVA Technical Note 8 Figure 9 Editing the options using the quick access toolbar The Edit Options window will be displayed. Click on the Cutoff tab to show the edit the cutoff conditions (see Figure 10). 9 Page

10 NOVA Technical note 8 Figure 10 The Edit Options window Click the button to add a cutoff condition (see Figure 10). Since we want to stop the complete procedure whenever the External(1).External 1 drops below 0.6 V, a cutoff on this signal should be defined, in the properties frame on the right-hand side, as shown in Figure Page

11 NOVA Technical Note 8 Figure 11 Setting the properties for the Cutoff Note The Nr. of detections property defines the number of consecutive detections required to trigger the cutoff condition. In this example, this value is set to 1. See the Cutoff tutorial for more information. Click the button to close the Edit Options window. The cutoff option will now be added to the Record signals (> 1 ms) command (see Figure 12). 11 Page

12 NOVA Technical note 8 Figure 12 The cutoff condition is now added to the Record signals command The procedure is now ready to be used. After each individual frequency measurement, the second signal will be sampled ten consecutive times, using an interval time of 0.1 s. If during this measurement the measured value drops below 0.6 V the whole procedure will be stopped. If the cutoff condition is not met, the measurement will proceed with the next frequency. 3 Testing the procedure on the dummy cell In order to simulate the cutoff condition during the FRA measurement, the procedure provided with this technical note has been slightly adjusted. The cutoff condition is simulated during the FRA frequency scan by changing the value generated by the DAC164. This value will be recorded on the ADC164. To simulate this condition, a special BNC connection must be established: For a PGSTAT fitted with a FRA module, connect a BNC cable between DAC164 #1 and ADC164 #1 on the front panel of the instrument. For a µautolabiii/fra2, connect a BNC cable between the V out and the V in on the back panel of the instrument. 12 Page

13 NOVA Technical Note 8 For the M101 module installed in a Multi Autolab Cabinet, connect a BNC cable between the V out and the V in connection on the optional monitor cable 3. Load the procedure Cutoff in FRA special [NOVA 1.10].nox provided with this technical note. This procedure is identical to the procedure detailed in this case study. However, an additional Set DAC command has been added to the procedure (see Figure 13). This command is used to set the DAC164 value to given voltage. A calculate signal is used to multiply the frequency index by 50 mv after each repetition. Figure 13 The modified procedure used in to simulate the cutoff condition 3 Contact your Autolab distributor for more information. 13 Page

14 NOVA Technical note 8 Connect dummy cell (c) and press the start button. The procedure will be started with the following experimental parameters: DC current: 1 ma AC amplitude: 100 µa (rms) Highest frequency: 10 khz Lowest frequency: 0.1 Hz Number of frequencies and distribution: 50, logarithmic Initial value of the DAC164: 1 V Since the value of the DAC164 decreases with 50 mv after each frequency, the cutoff condition will be automatically triggered after the 13 th frequency (see Figure 14). Figure 14 The cutoff condition is met after the 13 th frequency in the scan Note Switch to the analysis view to analyze the measured data points. When the cutoff condition is met, the measurement is immediately stopped and a message will be displayed in User log. 14 Page

### The Autolab control command is located in the Measurement General group of commands.

Version 1.11.0 NOVA Autolab control tutorial 1 The Autolab control command The Autolab control command is one of the most important commands in the NOVA software. Through this command, all of the hardware

### Increase the measurement speed with NOVA

Increase the measurement speed with NOVA Case study: how to increase the measurement speed by optimizing the USB communication with the Autolab? 1 USB 1.1 vs USB 2.0 Although the Autolab uses a USB 1.1

### NOVA. Getting started

NOVA Getting started Table of contents Introduction... 7 The philosophy of Nova... 8 1 Nova installation... 11 1.1 Requirements... 11 1.2 Software installation... 11 1.2.1.NET 4.0 framework installation...

### Event counters in NOVA

Case study: how to use the event counters in NOVA? 1 Event counter support NOVA Technical Note 22 Event counters in NOVA Most of the measurement commands, like CV staircase or Record signals (> 1 ms) provide

### Designing the Solution for Electrochemistry Potentiostat/Galvanostat І Battery Cycler І Fuel Cell Test Station +82-2-578-6516 І sales@wonatech.

ZIVE Electrochemical Workstations ZIVE SP2 Designing the Solution for Electrochemistry Potentiostat/Galvanostat І Battery Cycler І Fuel Cell Test Station +82-2-578-6516 І sales@wonatech.com www.wonatech.com

### Case study: using the Autolab SDK to develop NOVA applications in LabVIEW

Autolab SDK Case study: using the Autolab SDK to develop NOVA applications in LabVIEW 1 What is the Autolab SDK? The Autolab Software Development Kit (Autolab SDK) is designed to control the Autolab instrument

### Building a Simulink model for real-time analysis V1.15.00. Copyright g.tec medical engineering GmbH

g.tec medical engineering GmbH Sierningstrasse 14, A-4521 Schiedlberg Austria - Europe Tel.: (43)-7251-22240-0 Fax: (43)-7251-22240-39 office@gtec.at, http://www.gtec.at Building a Simulink model for real-time

### Lab 3: Introduction to Data Acquisition Cards

Lab 3: Introduction to Data Acquisition Cards INTRODUCTION: In this lab, you will be building a VI to display the input measured on a channel. However, within your own VI you will use LabVIEW supplied

### Lab 4 Op Amp Filters

Lab 4 Op Amp Filters Figure 4.0. Frequency Characteristics of a BandPass Filter Adding a few capacitors and resistors to the basic operational amplifier (op amp) circuit can yield many interesting analog

### Rx Medical. SMD Utility. Task Scheduler Configuration

Rx Medical SMD Utility Task Scheduler Configuration This document contains important information for all Rx Medical users. Please ensure this document is circulated amongst all your staff. We suggest these

### How to Attach Files in Blackboard Learn Email, Messages, Discussions, and Assignments

E-Mail 1. Launch the Email by clicking the icon from the Menu frame found on the left side of the screen. 2. From within the Send Email page, click the "Select Users" link. If you click any links above

### Cadence Tutorial A: Schematic Entry and Functional Simulation Created for the MSU VLSI program by Professor A. Mason and the AMSaC lab group.

Cadence Tutorial A: Schematic Entry and Functional Simulation Created for the MSU VLSI program by Professor A. Mason and the AMSaC lab group. Revision Notes: July 2004 Created tutorial for analog simulation

### Automatic data recovery

Case study: storing data during long measurements? 1 Storing data in NOVA Automatic data recovery In NOVA, all the measured data points are stored in a single.nox file at the end of the measurement. The.nox

### MATERIALS. Multisim screen shots sent to TA.

Page 1/8 Revision 0 9-Jun-10 OBJECTIVES Learn new Multisim components and instruments. Conduct a Multisim transient analysis. Gain proficiency in the function generator and oscilloscope. MATERIALS Multisim

### NEW FEATURES IN VERSION 1.4

NEW FEATURES IN VERSION 1.4 Some significant improvements and new capabilities have been introduced to the Goldeneagle HD with software/firmware version 1.4, released in February 2007 and FREE available

### ESE 216 LTSpice Quick Guide

ESE 216 LTSpice Quick Guide Install the LTSpice from the webpage http://www.linear.com/ltspice and install it on your PC. It is free simulation software which has the SPICE simulator core and has many

### Tutorial #2: RC Circuit

Tutorial #2: RC Circuit In this tutorial, we will build and simulate the RC circuit shown in Figure 1. For simplicity, wires are referred to as w1-w5. Figure 1: Circuit to be simulated Step 1. Open the

### Manual. EN Appendix. Blue Solar PWM Pro setup and monitoring software

Manual EN Appendix Blue Solar PWM Pro setup and monitoring software EN Appendix Contents 1. Region and language settings 2 2. Download the software 2 3. UnRAR the software 2 4. Install the software 3

### Interactive Logging with FlukeView Forms

FlukeView Forms Technical Note Fluke developed an Event Logging function allowing the Fluke 89-IV and the Fluke 189 models to profile the behavior of a signal over time without requiring a great deal of

### SainSmart DDS Series User Manual (Software)

SainSmart DDS Series User Manual (Software) For Oscilloscope, Signal Generator and Logic Analyzer Updated software and user manual can be downloaded from http://www.sainsmart.com/tools-equipments/oscilloscope-dso/sainsmart-dds.html

### 10.3.1.8 Lab - Configure a Windows 7 Firewall

5.0 10.3.1.8 Lab - Configure a Windows 7 Firewall Print and complete this lab. In this lab, you will explore the Windows 7 Firewall and configure some advanced settings. Recommended Equipment Step 1 Two

### In this note, you will learn the basic applications of McAfee VirusScan Enterprise (hereafter McAfee ). Six topics will be covered as below:

ANTIVIRUS SOFTWARE In this note, you will learn the basic applications of McAfee VirusScan Enterprise (hereafter McAfee ). Six topics will be covered as below: 1. Start McAfee VirusScan Enterprise 2. Change

### A Primer for the ESyst Analog Systems ilab James Hardison - Document version 1.0

A Primer for the ESyst Analog Systems ilab James Hardison - hardison@mit.edu Document version 1.0 This document is intended as an introduction to the ESyst Analog Systems ilab. The functionality of this

### Efficiency of a Motor and a Generator

Efficiency of a Motor and a Generator 6EM Object: Apparatus: The purpose of this laboratory activity is to determine the efficiency of an electric motor and the efficiency of an electric generator. Electric

### Technical Note T1. Batch Mode to Sequence Experiments. 1. Introduction

Technical Note T1 Batch Mode to Sequence Experiments 1. Introduction Ivium Potentiostat/Galvanostats are controlled by IviumSoft Control and Data Analysis Software. There are three ways that IviumSoft

### Replacing the USB DIO Decoder

Metrohm Autolab Service Document # 008 Table of Contents Replacing the USB DIO Decoder 1. Scope of delivery... 2 2. Replacing the USB DIO Decoder on Autolab Series 7 instruments... 3 2.1. Removal of the

Taskbar and Start Menu Options This area is used to make changes as to how items appear on the Taskbar or the Start Menu. Click the Start Button. Click Control Panel. The Control Panel window will display

### Create a PDF File. Tip. In this lesson, you will learn how to:

Create a PDF File Now that you ve seen what an ETD looks like and how to browse the contents, it s time to learn how to convert your own thesis or dissertation into a PDF file. There are several different

### CHAPTER 6: SEARCHING AN ONLINE DATABASE

CHAPTER 6: SEARCHING AN ONLINE DATABASE WHAT S INSIDE Searching an Online Database... 6-1 Selecting a Display Mode... 6-1 Searching a Database... 6-1 Reviewing References... 6-2 Finding Full Text for a

### Universal Simple Control, USC-1

Universal Simple Control, USC-1 Data and Event Logging with the USB Flash Drive DATA-PAK The USC-1 universal simple voltage regulator control uses a flash drive to store data. Then a propriety Data and

### Quick Start Guide. Agilent Technologies AC Source/ Power Analyzer Graphical User Interface for Windows and Windows NT 4.0

Quick Start Guide Agilent Technologies AC Source/ Power Analyzer Graphical User Interface for Windows 95-98 and Windows NT 4.0 Contents Description...3 System Requirements...3 Installing and Running the

### In this lab you will explore the Windows XP Firewall and configure some advanced settings.

16.3.2 Lab: Configure Windows XP Firewall Print and complete this lab. In this lab you will explore the Windows XP Firewall and configure some advanced settings. Recommended Equipment Two computers directly

### Exercise 4 - Broadband transistor amplifier

ANALOG ELECTRONIC CIRCUITS Laboratory work Exercise 4 - Broadband transistor amplifier Task 1: Design a broadband amplifier using a bipolar NPN transistor in a common emitter orientation. For input signal

### 10.3.1.9 Lab - Configure a Windows Vista Firewall

5.0 10.3.1.9 Lab - Configure a Windows Vista Firewall Print and complete this lab. In this lab, you will explore the Windows Vista Firewall and configure some advanced settings. Recommended Equipment Step

### Teacher s Guide Physics Labs with Computers, Vol C P52: LRC Circuit. Teacher s Guide - Activity P52: LRC Circuit (Voltage Sensor)

Teacher s Guide Physics Labs with Computers, Vol. 2 012-06101C P52: LRC Circuit Teacher s Guide - Activity P52: LRC Circuit (Voltage Sensor) Concept DataStudio ScienceWorkshop (Mac) ScienceWorkshop (Win)

### Database Program Instructions

Database Program Instructions 1) Start your Comparative Rating software by double-clicking the icon on your desktop. 2) Click on the button on the Comparative Rating Software Main Menu. 3) A message Loading

### Kramer Electronics, Ltd. Site-CTRL and Web Access Online User Guide (Documentation Revision 2)

Kramer Electronics, Ltd. Site-CTRL and Web Access Online User Guide (Documentation Revision 2) Software Version 1.9. 2902 To check that you have the latest version, go to the DOWNLOADS section of our Web

### 10.3.1.10 Lab - Configure a Windows XP Firewall

5.0 10.3.1.10 Lab - Configure a Windows XP Firewall Print and complete this lab. In this lab, you will explore the Windows XP Firewall and configure some advanced settings. Recommended Equipment Step 1

### Chronicle USER MANUAL

Chronicle USER MANUAL 1st Edition 2 IN THIS MANUAL Part One The Chronicle Interface The Overview Screen The Bill Detail Screen Part Two Creating, Editing and Viewing Bills Creating Your First Bill Editing

### Instruction Manual SU

DIGITAL SOUND LEVEL METER Off between 1 to 9 minutes or disable Auto Power Off. * The instrument can memorize 30 measurement results and measuring conditions for later use or download to PC * Can communicate

ACTIVE TRADING WITH TRADESTATION AND THE STOCKSCORES STRATEGIES WORKSPACE/DESKTOP LAYOUT There are infinite ways to set up your Tradestation layout; how you do it is a personal preference. The process

### Current Monitoring Kit QUICK START GUIDE

Current Monitoring Kit QUICK START GUIDE Pico Technology EL040 Current Monitoring Kit Quick Start Guide CONTENTS 1 Introduction... 1 2 Kit contents... 3 3 Installation... 3 3.1 Connecting the system together...

### Teacher References archived classes and resources

Archived Classes At the end of each school year, the past year s academic classes are archived, meaning they re still kept in finalsite, but are put in an inactive state and are not accessible by students.

### User Guide: Vodafone Mobile Broadband

User Guide: Vodafone Mobile Broadband Contents Introduction... 3 Vodafone Mobile Broadband... 4 Application Start... 4 Main View... 4 Pre-Pay Account Option... 4 My Account tab... 5 Customer Messaging...

### Current Monitoring Kit

Current Monitoring Kit QUICK START GUIDE DO090-6 CONTENTS Issues: 1) 2.10.02 WP A4 format 2) 2.10.02 Added safety warning 3) 17.3.06 Word A5 format. S1: Removed relay modules. S2: Added MI010. S4.1: added

### Spectrum and Power Measurements Using the E6474A Wireless Network Optimization Platform

Application Note Spectrum and Power Measurements Using the E6474A Wireless Network Optimization Platform By: Richard Komar Introduction With the rapid development of wireless technologies, it has become

### Lab 5 Getting started with analog-digital conversion

Lab 5 Getting started with analog-digital conversion Achievements in this experiment Practical knowledge of coding of an analog signal into a train of digital codewords in binary format using pulse code

### Quick Start Guide. Highly customizable automated trading Automate your trades according to rules and models you create.

POWER E*TRADE PRO EXCEL MANAGER Quick Start Guide We are pleased to announce the launch of Excel Manager, an exciting new feature in Power E*TRADE Pro that leverages the flexibility of Microsoft Excel

### Ticket Validation Application System User s Guide. Created by:

Ticket Validation Application System User s Guide Created by: 12-21-2011 Table of Contents Ticket Validation Application System Desktop Icon... 3 Prerequisites... 4 Ticket Validation Reporting... 5 Merchant

### CRM Knowledge Base. Contents

Contents Overview:... 2 The Article Record:... 3 Searching for Articles... 3 Quick Search... 3 Article Groups... 5 Using Favorites... 5 Adding New Articles... 6 Maintaining Articles... 8 Groups... 9 Keywords...

### RealSimPLE: Soundcard. Testing Your Soundcard

RealSimPLE: Soundcard Testing Your Soundcard An experiment that helps you assess the actual performance of your sound card and determine if it is good enough RealSimPLE 2007 Testing Your Sound Card Page

### Introduction to Microsoft Access 2003

Introduction to Microsoft Access 2003 Zhi Liu School of Information Fall/2006 Introduction and Objectives Microsoft Access 2003 is a powerful, yet easy to learn, relational database application for Microsoft

### Using an Automatic Back Up for Outlook 2003 and Outlook 2007 Personal Folders

Using an Automatic Back Up for Outlook 2003 and Outlook 2007 Personal Folders Part 1 Install the Personal Folder Backup Utility Note: You should close Outlook and any other applications before installing

### WinTask x64 Scheduler for Windows 7 64 bit, Windows 8/8.1 64 bit and Windows 2008 R2 64 bit. Scheduler Quick Start Guide

WinTask x64 Scheduler for Windows 7 64 bit, Windows 8/8.1 64 bit and Windows 2008 R2 64 bit Scheduler Quick Start Guide 2 INTRODUCTION 5 CHAPTER I : INSTALLATION 7 CHAPTER II : SET UP YOUR FIRST SCHEDULED

### Personal Training. Chapter 27. FrontDesk v4.1.25. PT Specific Program Options. Contract Templates. Set Member s Secure Check In Password

Chapter 27 Personal Training FrontDesk v4.1.25 PT Specific Program Options Empl oyees Contract Templates Personal Training Window in Contract Wizard Contract Manager / Personal Training Information Clear

### Microsoft Office 2010

Access Tutorial 1 Creating a Database Microsoft Office 2010 Objectives Learn basic database concepts and terms Explore the Microsoft Access window and Backstage view Create a blank database Create and

### Distribution List Manager User s Manual

Distribution List Manager User s Manual Welcome... 2 Configuring the Add-in... 3 Settings Tab... 3 Advanced Tab... 4 Troubleshooting... 5 Distribution List Manager Copyright 2008 Sperry Software Inc. P

### Cell Phone Vibration Experiment

Objective Cell Phone Vibration Experiment Most cell phones are designed to vibrate. But at what frequency do they vibrate? With an accelerometer, data acquisition and signal analysis the vibration frequency

### Administrator User Guide OhioHealth University learning management system (LMS)

Administrator User Guide OhioHealth University learning management system (LMS) Table of Contents Introduction Page 3 Student and Group Transcript Report Pages 4-5 Assignment Completion Report Pages 6-0

### Windows 8.1 Tips and Tricks

Windows 8.1 Tips and Tricks Table of Contents Tiles... 2 Removing, Resizing and Moving Existing Tiles... 2 Adding New Tiles... 2 Returning to the Start Screen (Charms)... 3 The Search Feature... 3 Switching

### Tutorial. Patch Management

Tutorial Patch Management 2011 DeskCenter Solutions AG Table of Contents 1. Introduction...3 2. Software deployer...4 2.1 Preparing the software deployer...4 2.2 Configuring software deployers...4 3. Central

### Microsoft Outlook 2007 Calendar Features

Microsoft Outlook 2007 Calendar Features Participant Guide HR Training and Development For technical assistance, please call 257-1300 Copyright 2007 Microsoft Outlook 2007 Calendar Objectives After completing

### LABORATORY 2 THE DIFFERENTIAL AMPLIFIER

LABORATORY 2 THE DIFFERENTIAL AMPLIFIER OBJECTIVES 1. To understand how to amplify weak (small) signals in the presence of noise. 1. To understand how a differential amplifier rejects noise and common

### LEARNING MANAGEMENT SYSTEM (LMS) SUBSCRIBING TO A REPORT

LEARNING MANAGEMENT SYSTEM (LMS) SUBSCRIBING TO A REPORT This QRG provides instructions for subscribing to a report in the LMS. Report subscriptions in the LMS allow reports to be emailed at scheduled

### E-Map Application CHAPTER. The E-Map Editor

CHAPTER 7 E-Map Application E-Map displays the monitoring area on an electronic map, by which the operator can easily locate the cameras, sensors and alarms triggered by motion or I/O devices. Topics discussed

### Impedance 50 (75 connectors via adapters)

VECTOR NETWORK ANALYZER PLANAR TR1300/1 DATA SHEET Frequency range: 300 khz to 1.3 GHz Measured parameters: S11, S21 Dynamic range of transmission measurement magnitude: 130 db Measurement time per point:

### NVMS-1200. User Manual

NVMS-1200 User Manual Contents 1 Software Introduction... 1 1.1 Summary... 1 1.2 Install and Uninstall... 1 1.2.1 Install the Software... 1 2 Login Software... 3 2.1 Login... 3 2.2 Control Panel Instruction...

Archiving An archive is a copy of a database you can create to store information no longer in use. Like a replica, an archive contains all documents and design elements in the original database, but unlike

### The Oscilloscope, the Signal Generator and Your Filter s Test Setup SGM 5/29/2013

The Oscilloscope, the Signal Generator and Your Filter s Test Setup SGM 5/29/2013 1. Oscilloscope A multimeter is an appropriate device to measure DC voltages, however, when a signal alternates at relatively

### NOVA MULTI-MONITORING SOFTWARE. Instruction Manual

NOVA MULTI-MONITORING SOFTWARE REMOTE OPERATION Precision Digital Corporation s Nova Multi-Monitoring software allows for the operation of common Nova control features remotely from a PC. Up to 30 Nova

### Figure 1. Front Panel of Function Generator

Equipment Introduction: Part I - Introduction to the Function Generator Overview: The function generator is used to generate a wide range of alternating-current (AC) signals. A diagram of the Leader LFG-1300S

### A Lead is a record of a short or long term sale possibility. Note: In Selltis, a Lead is an Activity Log with the Purpose set as Lead

Learning Objectives A Lead is a record of a short or long term sale possibility. Note: In Selltis, a Lead is an Activity Log with the Purpose set as Lead Leads are very important Selltis forms because,

### OrCAD Capture with PSpice and Allegro DE CIS with AMS Simulator. Describes how to create a PSpice Archive File with Capture

Title: Product: Summary: Creating a Project Archive OrCAD Capture with PSpice and Allegro DE CIS with AMS Simulator Describes how to create a PSpice Archive File with Capture Author/Date: Wei Ling / 03.08.2009

### Scanning Documents into OneSite The Preiss Company

Scanning Documents into OneSite The Preiss Company Scanning Single Documents (i.e. lease only) For a tutorial on scanning multiple documents at once (i.e. lease & application) skip to the section titled

### CANON FAX L360 SOFTWARE MANUAL

CANON FAX L360 SOFTWARE MANUAL Before You Begin the Installation: a Checklist 2 To ensure a smooth and successful installation, take some time before you begin to plan and prepare for the installation

### Optomate Training Compendium Appointment Book Page 2

2. Appointment Book The Optomate Premier Series Appointment Book has been developed based on the Microsoft Outlook calendar. It offers a quick and easy method for finding and entering new appointments,

### Check out our website!

Check out our website! www.nvcc.edu/woodbr idge/computer-lab Contact Us Location: Open Computer Lab Seefeldt Building #336 NOVA Woodbridge Campus Hussna Azamy (OCL Supervisor) Phone: 703-878-5714 E-mail:

### Fax and SMS Quickguide

INTRODUCTION U2 Fax and SMS allow faxes and SMS to be sent directly from a user s workstation via the Internet using U2 Webservices. This improves staff efficiency by removing the need to physically fax

### Welcome to. NukeWorker.com s Employer Services Job Management Tutorial

Welcome to NukeWorker.com s Employer Services Job Management Tutorial In this module, you will learn how to manage jobs using your Employer Services account. Page 1 of 5 Table of Contents 1. ACCOUNT SUMMARY...

### Application Model: EPM for SAP GUI Scripting

Welcome to this review guide for Configuration and Using ZyALERT This review guide will guide you through the steps necessary to configure ZyALERT, maintain users and alerts to be able to send email alerts

### 1. Installing The Monitoring Software

SD7000 Digital Microphone Monitor Software manual Table of Contents 1. Installing The Monitor Software 1.1 Setting Up Receivers For Monitoring 1.2 Running The Application 1.3 Shutdown 2. The Detail Monitoring

### Charge and Discharge of a Capacitor

Charge and Discharge of a Capacitor INTRODUCTION Capacitors 1 are devices that can store electric charge and energy. Capacitors have several uses, such as filters in DC power supplies and as energy storage

### Quick start guide to using Attendant

Quick start guide to using Attendant Attendant is an easy way to create and take registers for the classes you teach. These can be lectures, tutorials, studios, labs or any class that you would like to

### Manual. OIRE Escuela de Profesiones de la Salud. Power Point 2007

Power Point 2007 Manual OIRE Escuela de Profesiones de la Salud Power Point 2007 2008 The New Power Point Interface PowerPoint is currently the most common software used for making visual aids for presentations.

### ShoreTel Communicator User Guide Professional

ShoreTel Communicator User Guide Professional Using Communicator for Basic Call Handling Features and Functions Making Calls Type Name of desired party or enter Extension Number in Quick Dialer When populated

### Active Directory Syncing

User Guide Page 1 of 7 This document, in whole or in part, must not be reproduced in any form without the express written permission of 1.0 Introduction The Add-On Tool makes it easy to create users and

### Presents. And. Base documentation

Presents And Base documentation Par: Internal doc Title: Internal doc Date production: September 12 th 2011 Date revised : November 17 th 2011 www.scopserv.com Page 1 / 10 Direct Login page You can log

### INTRODUCTION TO ARBITRARY/FUNCTION GENERATOR

Page 1 of 7 INTRODUCTION TO ARBITRARY/FUNCTION GENERATOR BEFORE YOU BEGIN PREREQUISITE LABS Introduction to MATLAB Introduction to Oscilloscope EXPECTED KNOWLEDGE Ohm s law & Kirchhoff s laws Operation

### Function Generator. Instruction Manual A PI Function Generator. AC to 15 V DC, 1.6 A power adapter (not shown)

Instruction Manual 012-10425A Function Generator PI-8127 Frequency Adjust knob and Range Selection Waveform Selection Menu Voltage Adjust knob and Menu Selection Liquid Crystal Display (LCD) Power Output

### AMETEK s Elgar SW and California Instruments CSW Form, Fit and Functions Comparison

AMETEK s Elgar SW and California Instruments CSW Form, Fit and Functions Comparison Introduction The purpose of this technical note is to compare the differences between the discontinued Elgar SW Series

### Tutorial: Configuring GOOSE in MiCOM S1 Studio 1. Requirements

Tutorial: Configuring GOOSE in MiCOM S1 Studio 1. Requirements - Two (2) MiCOM Px4x IEDs with Version 2 implementation of IEC 61850 - Two (2) Cat 5E Ethernet cable - An Ethernet switch 10/100 Mbps - MiCOM

### Fluke 89-IV & 189 Event Logging

Fluke 89-IV & 189 Event Logging FlukeView Forms Technical Note One of the major features of the Fluke 89-IV & 189 meters are their ability to "do logging". This technical note explains what kind of logging

### Date: 9/15/2014. MR3 Signal Processing

Date: 9/15/2014 MR3 Signal Processing Real Time Processing Mean Value Recording Designed for long measurements, where any raw data recording would be too much data or processing time would be too long.

### GA-200 Gas Analyzer. Technical Note. Overview. Front Panel. iworx Systems, Inc. GA-200 Gas Analyzer

Technical Note Overview The uses sensors to measure and display the concentrations of oxygen and carbon dioxide in a sample as the percentage of a gas in the sample by volume. This method of expressing