1 Catalog Number Hydras 3 LT Quick Start SOFTWARE MANUAL December 2005, Edition 2
2 Hach Company, All rights reserved. Printed in the U.S.A.
3 Catalog Number Hydras 3 LT SOFTWARE MANUAL December 2005, Edition 2 Hach Company, All rights reserved. Printed in the U.S.A. jk/dk
4 Visit us at
5 Table of Contents Section 1 General Information... 5 Section 2 Startup Downloading Data Operating a Sonde System Tab Online Monitoring Tab Time Series Vertical Profile Manual Profile Stability Check Real-time Graph Functions Log Files Tab Creating a Log File Storing a Template File Recalling Template Files Parameter Setup Tab Calibration Tab Settings Tab Define SDI Parameter Order Software Tab Hydras 3 Options (F2) Graphics Tab DataSonde Tab Section 3 Contact Information Appendix A Troubleshooting
6 Visit us at
7 Section 1 General Information Hydras 3LT is software used to connect Hydrolab sondes to a PC. It automatically scans for connected sondes and recognizes any log files held in memory. Up to 32 sondes can be connected at once, and all log files can be downloaded simultaneously. When operating the sonde, Hydras 3LT provides seven function tabs to help view real-time data, or to set up a sonde for operation: The System tab allows the user to setup basic sonde identification information. Also, passwords can be established to allow up to four different access levels, which can help to protect the data. The Online Monitoring tab allows viewing real-time data as either a time series graph, or as a vertical profile graph. Up to six parameters can be viewed at a time on either graph. Data can also be downloaded into an Excel or text file. The Log Files tab allows quick and easy setup of logging runs with a GUI interface. Choose the start and stop time, the logging interval, as well as the circulator and sensor warm-up time. Add or remove parameters and arrange them in order. When the logging run is complete, download all the files at once. The Parameter Setup tab allows the user to configure available parameters. The Calibration tab sets the calibration parameters for an individual sensor. Choose the sensor to calibrate and enter the calibration standards. The current value and temperature readings are shown for temperature sensitive calibrations. The Settings tab sets the communication settings between the Hydras 3LT and the sonde. The Software tab uploads software to the sonde and updates and removes software drivers. 5
8 Visit us at
9 Section 2 Startup 1. Connect the Data Cable from the computer and to the sonde. 2. Start Hydras 3 LT. The software will automatically scan for sondes. All detected sondes are displayed in the Connected sondes list in the Main window displayed below. If a sonde is not found, press RE-SCAN FOR SONDES. If a connection can not be established, refer to Appendix A Troubleshooting on page 29. 7
10 2.1 Downloading Data After a log file is created in the Log Files tab, the files can be downloaded by checking the appropriate Log File box and clicking DOWNLOAD SELECTED FILES. Multiple files can be downloaded at once. The downloaded log files are stored in the location indicated at the bottom of the screen (typically C:\Program Files\HYDRAS3LT\LogFiles ). The location can be changed by clicking EDIT and specifying a new directory. 2.2 Operating a Sonde Note: To enter a higher level security mode, click on the Level button from the System Tab and enter the password. 1. From the Main Hydras 3 LT menu, highlight the appropriate sonde and press OPERATE SONDE. The sonde window will appear. The window defaults to Security Level The sonde window contains 7 function tabs; System, Online Monitoring, Log Files, Parameter Setup, Calibration, Settings, and Software. 8
11 2.2.1 System Tab The System Tab sets up general information and configures the system components. # Function Description 1 Instrument ID Enter a description of the sonde. This will be used as an output on the data file. 2 Sonde Information Displays static information about the sonde. 3 Clock Sets the sonde date and time to match your PC clock or manually set the date and time (this is useful when monitoring in different time zones). 4 Circulator Select whether to start or stop the circulator. 5 Audio Turns on/off the beeping sound the sonde makes when taking measurements. 6 Security Level Click on LEVEL 0, LEVEL 1, LEVEL 2, or LEVEL 3 and enter the password to view that security mode. While in Level 3, click on PASSWORDS to set the passwords for Level 1, Level 2, and Level 3. The default Level 3 password from the factory is Hydrolab. 9
12 2.2.2 Online Monitoring Tab The Online Monitoring Tab has three modes for capturing data; Manual, Time Series, and Vertical Profile (sections , , and ). Criteria checks can be set before capturing the data (section ) and then the results can be graphed (section ). # Function Description 1 Monitoring Mode The monitoring mode must be set before starting Online Monitoring. Select Time Series for logging values at regular time intervals. Select Vertical Profile for logging values at user-defined depth levels. Or select Manually for User-defined logging. 2 Monitoring Interval Select the time interval or depth level between samples as well as the stability check for Vertical Profile and Manual settings (see section ). 3 Parameters Check the parameters to be monitored. 4 Arrow buttons Select a parameter and use the UP and DOWN arrow buttons to move the parameter in the list. 10
13 5 Data Buttons 1 (Monitoring must begin before the data button functions can be used) New Graph: Opens a graphic window with the selected parameters (max. 6) plotted as a time series. See Real-time Graph Functions on page 15 for more information on changing graph properties. New Depth Graph: Opens a graphic window with the selected parameters (max. 6) displayed at the corresponding depth. New Table: Displays the selected monitored data in a tabular view (real-time). Export Excel: Microsoft Excel must be installed on the computer for this function to work. The monitored data is transferred to an Excel application running in the background. If Excel is not running, it is automatically started and a new worksheet is created. After the data is exported, the Excel file is not updated with new real-time data. Hydras 3LT continues to log such data for graphing and storage purposes. Export Textfile: Writes the monitored data to a text file. After the data is exported, the file is not updated with new real-time data. Transfer to Database: Not available with Hydras 3 LT. 1 The data buttons become available when at least one parameter in the list is selected Time Series A time-based logging can be set by selecting Time Series as the Monitoring Mode. The user must select the time interval and parameters. Click START to begin monitoring data (see section for a screen shot). 11
14 Vertical Profile A vertical profile can be set by selecting Vertical Profile as the Monitoring Mode. The user must select the depth increment for the vertical profile, the units of measure for the profile, and whether the profile is ascending (up arrow) or descending (down arrow). The user should also indicate whether or not a stability check is to be used and how many samples per measurement point are preferred. 12
15 Manual Profile A manual profile can be set by selecting Manually as the Monitoring Mode. The user should also indicate whether or not a stability check is to be used and how many samples per measurement point are preferred. The data is captured by either pressing the space bar or by clicking on the capture button 13
16 Stability Check When monitoring in Depth Mode or Manually, the user can set up stability criteria which must be met prior to Hydras 3LT recording a measurement. The user needs to select a given parameter such as SpCond, the maximum delta (± this number from an average value), the units of measurement, and the number of samples that must meet this stability criteria. To select a parameter of interest, click on the parameter name in the Stability Criteria window. The check box in front of the parameter name is used to indicate whether that parameter will be used as part of the stability criteria check. When more than one stability criteria is checked, all conditions must be met for data to be collected. Click on CONFIGURATION in the Online Monitoring screen to enter the stability criteria screen. 14
17 Real-time Graph Functions Up to six selected parameters can be displayed with different colors and an individual axis range for every sensor. In the graphic options (F2), the user can select if every parameter has its own vertical axis displayed or that only 2 vertical axis are used (one left, one right). To change the scale on the y-axis, click on the minimum and maximum values on the graph. A dialog box will appear, enter the minimum and maximum y-axis values. Press Ctrl-A to autoscale the y-axis at any time while the data is plotted on the graph. Right-click on the graph to display a context menu. See the table below for context menu options. Function Span Grid Overlay Statistics Table Ruler Description Select the time span that is displayed until the current value (1 min, 5 min, 10 min, 30 min, 1 h, 3 h, 6 h, 12 h, or 24 h). Turn the horizontal and vertical grid on or off. Select overlay statistics to show simple statistics on the graph such as max, min, and average. Select table to view the graph in a tabular form. Turn the two rulers on or off. 15
18 2.2.3 Log Files Tab Creating a Log File Users frequently like to use the same Log File settings for multiple sondes or for redeploying a sonde after retrieving a stored log file. The Log Files tab allows the user to share and retrieve template Log File settings using a right-click while the cursor is in the Log Files screen. Note: A log file must be created and then enabled before data can be collected. 1. Click the CREATE button. 2. Enter the name for the new log file. The empty log file is now created. Note: To delete a log file, select the log file in the Log File drop-down menu and click the DELETE button. 3. Enter the start and end time of the log, the logging interval, the sensor warm-up time before logging, and how long before logging the circulator will be turned on, and if audio signals will be used while logging. 4. Select the parameters in the Parameter in sonde list and click the ADD button to place them into the Parameters in log file list. Change the order of the parameters using the UP and DOWN arrow buttons. 5. Click SAVE SETTINGS to send the configuration to the sonde. 6. Click ENABLE to start collecting data. Click DISABLE to stop collecting data during logging. A fully completed logging run will automatically disable at the end of the run. 7. Click DOWNLOAD to download and display the log file. Select Printable or Spreadsheet format. 16
19 Storing a Template File Note: In order to create a stored template file, a sonde must be connected to Hydras 3LT. 1. Open an existing log file by using the Log File pull-down menu or create a new log file by following steps 1 4 in section While the cursor is in the Log Files screen, right-click to open the template selection screen. 3. Select STORE TEMPLATE. 4. Choose a name for the template. It will be stored automatically in the template folder. Reusing a template name will overwrite the old template. 5. Select SAVE. 6. The log file template is now saved and can be used to speed up the creation of future log files. 17
20 Recalling Template Files Note: In order to use a template file, the sonde used to create the template and the sonde used when retrieving the template should be configured in a compatible manner. That is, all parameters used in the template file should be available in the sonde to which the template file is being applied. 1. Click the CREATE button to create a new log file or open an existing log file by using the Log File pull-down menu. 2. Enter a name for the new log file, if necessary. An empty log file is created. 3. While the cursor is in the Log Files screen, right-click to open the template selection screen. 4. Select Load Template. 5. The template will load onto the Log Files screen. It will be identical to the stored template except the Start Logging start time will be the current PC time. 6. Verify that all items are prepared as needed, including the Start Logging and Stop Logging times. 7. Click SAVE SETTINGS to send the configuration to the sonde. 8. Click ENABLE to start collecting data. 18
21 2.2.4 Parameter Setup Tab Click on the appropriate parameter and enter the setup information for each parameter. Press SAVE SETTINGS when all parameter setup information is complete. Refer to the instrument manual for more information for each parameter. 19
22 2.2.5 Calibration Tab Click on the appropriate parameter and enter the calibration information for each parameter. Press CALIBRATE when all the calibration information is complete. Refer to the instrument manual for more calibration information. 20
23 2.2.6 Settings Tab # Function Description 1 Baud Rate 2 SDI 3 MODBUS Select the instrument baud rate (1200, 2400, 4800, 9600, or 19200). Note: If the automatic baud rate scan is enabled (section DataSonde Tab on page 25), Hydras3LT will only scan for 9600 and baud rates. Enable SDI when multiple sondes or other SDI-12 enabled sensors are connected to a single SDI-12 controller. Set the delay to allow the sensors to warm-up and stabilize for accurate measurements. The transmitter factory default SDI-12 address is 0 for all parameters. If continuous mode is enabled, the unit will never enter sleep mode and measurements will be available immediately upon receiving an SDI-12 data request. The Modbus address default is 1. When using multiple sondes, assign a unique address (1 to 247) for each sonde. The Modbus works with even-parity, 8 data bit, and 1 stop bit. 4 Log Files When Auto Log is selected, it works as a back-up logging file. Auto Log captures a reading of all available parameters, battery voltages, and turns on the audio and circulator with a two-minute warm-up (if installed) once every hour. No log files must be selected in the Log Files tab for Autolog to run. 5 Statistics Press the STATISTICS button to view a summary of the sonde. 6 TTY Mode Use this option only when backwards compatibility is required. TTY mode provides data and limited menu access for external devices that interface with earlier generations of sondes. If TTY mode is entered, a user may exit TTY mode by opening a terminal screen and then pressing the space bar followed by a Q or q to quit. 7 Options See Hydras 3 Options (F2) on page
24 Define SDI Parameter Order The user can define the order in which parameters are reported for an SDI 12 data request. In the HydrolabsondeParameterOrderForm screen, the user can add parameters to be reported using SDI 12 data requests. The user can also order the parameters by using the UP and DOWN arrows to arrange the parameters in the desired order. Note: The parameter order set in this screen will also be utilized by TTY mode for reporting data values. 22
25 2.2.7 Software Tab Important Note: Level 3 Security Required. Software version 3.35 or higher is required to use the features on the Software screen. If the sonde has an older version of software, upgrade to the newest software version using terminal mode. Refer to the instrument manual for more information on updating the sonde software using terminal mode. # Function Description 1 Software Version Drivers/Remove To update the firmware, click on the UPDATE button in the Main group box. A prompt to select a file name will appear and then the selected file is uploaded to the sonde. After the upload, the sonde is reset to default values and the sonde window is created again. To remove a driver from a slot, select the driver and click the REMOVE button. After the driver is removed the sonde is reset and the sonde window is created again. 2 To update a driver, select the slot of the driver and click the UPDATE button in the Drivers group box. Select a file name when prompted. The selected file is then uploaded to the selected slot in Drivers/Update the sonde. If Reset Driver After Update is checked, all driver parameters are set to default values. 3 Reset Driver Resets all driver calibration parameters to default values. 4 Reset Sonde Resets all system parameters to default values. 23
26 2.2.8 Hydras 3 Options (F2) Graphics Tab To set the sonde Options select File>Options or press F2 from the sonde startup screen. The options menu contains two tabs, Graphics and DataSonde (sections and ). The Graphics tab adjusts the settings and appearance of the graph (ruler size, labeling, type of grid, and multiple graph functions). Up to six selected parameters can be displayed, with different colors and an individual axis range for every sensor. In the graphic options (F2), the user can select if every parameter has its own vertical axis displayed or that only 2 vertical axis are used (one left, one right). 24
27 DataSonde Tab The user is able to define which serial ports to scan for sondes using the Options screen (F2). Hydras 3LT can be set to scan either all ports or only those ports indicated in the ports list. A default PC baud rate can also be set. An extended timeout can be used when the user is connecting to older Series 4a sondes, typically those built before If a user is having difficulty connecting to a sonde, it is recommended that they set the Communication screen to Auto Scan (all serial ports), check the automatic baud rate scan, and use the extended timeout. The DataSonde tab is used to select the PC baud rate (1200, 2400, 4800, 9600, or 19200), measurement units (Celsius, Fahrenheit, Kelvin, depth, and battery voltage), log file formats (spreadsheet or printable), and format options such as date order, digits for year, date separator, time separator, and radix (decimal point or comma). 25
28 Visit us at
29 Section 3 Contact Information U.S.A. Customers By Telephone: 6:30 a.m. to 5:00 p.m. MST Monday through Friday (800) or (970) By Fax: (970) Ordering Information Required By Mail: Hach Environmental P.O. Box 389 Loveland, Colorado U.S.A. Ordering information by Account number (if available) Billing address Your name and phone number Shipping address Purchase order number Catalog number Brief description or model number Quantity International Customers To locate the representative nearest you, send an to: or contact: Hach Environmental World Headquarters; Loveland, Colorado, U.S.A. Telephone: (970) ; Fax: (970) Technical and Customer Service (U.S.A. only) Call or 27
30 Visit us at
31 Appendix A Troubleshooting If Hydras 3LT does not automatically detect the Sonde when launched, the following screen will appear after a period of time: Note: For some users with more than one active serial port, it will take some time for Hydras 3LT to try to auto-detect sondes on all ports. It may take up to 30 seconds per active serial port to do the auto-scan. If the sondes is not automatically detected, press the RE-SCAN FOR SONDES button. If communication is still not established after several attempts, try the following: Verify the Hardware. 1. Check power cables and connections: a. Verify that your PC and multiprobe are properly connected to the wall outlet or external battery if used. 2. Verify that the input voltage to the multiprobe is between 7V and 14V. 3. If your multiprobe is equipped with an internal battery pack, check the batteries' polarity and voltages. 29
32 Troubleshooting Verify the Hydras3LT Communication Settings. 1. Select File>Options from the Hydras 3LT Connection Screen. 2. If the Sondes/PC baud rate and COM port are known, disable the COM Port and baud rate auto scans and set Hydras3LT to use the known values. This requires Hydras 3LT to concentrate scan activities on only the serial ports of interest. 30
33 Troubleshooting 3. In some cases, extending the connection timeout from 10 to 15 seconds, will improve the connection procedure. This option allows time for additional retries. It is normally disabled to decrease the time taken to scan for a sonde. 31
User Manual Thermo Scientific Orion Orion Star Com Software Program 68X637901 Revision A April 2013 Contents Chapter 1... 4 Introduction... 4 Star Com Functions... 5 Chapter 2... 6 Software Installation
Ovation Operator Workstation for Microsoft Windows Operating System Features Delivers full multi-tasking operation Accesses up to 200,000 dynamic points Secure standard operating desktop environment Intuitive
File Management Utility T u t o r i a l Contents System Requirements... 2 Preparing Files for Transfer to GlobalMark... 2 Application Launch... 2 Printer Setup... 2 Communication Status... 4 Communication
Rubbermaid Medical Solutions Cart Interface Installation & Control Board Manual for M38 XP and RX Carts and Legacy M39 Carts REVISED FEBRUARY 25, 2010 Filename: Cart Interface Installation & Control Board
SubCue Analyzer Software - Introduction The SubCue Analyzer is a tool to initialize and download the temperature data from implantable SubCue Dataloggers. To begin, open the SubCue Analyzer program from
SA-9600 Surface Area Software Manual Version 4.0 Introduction The operation and data Presentation of the SA-9600 Surface Area analyzer is performed using a Microsoft Windows based software package. The
Real Time Monitor A Real-Time Windows Operator Interface DDE Compliant (for remote data display) TABLE OF CONTENTS 1. INTRODUCTION...1 1.1 INSTALLATION...2 1.2 FIRST START UP - DDE CONFIGURE...2 1.3 AUTO-STARTUP...2
Quick Start Using DASYLab with your Measurement Computing USB device Thank you for purchasing a USB data acquisition device from Measurement Computing Corporation (MCC). This Quick Start document contains
WatchPower User s Manual Management Software for Inverter Table of Contents 1. WatchPower Overview... 2 1.1. Introduction... 2 1.2. Features... 2 2. WatchPower Install and Uninstall... 2 2.1. System Requirement...
Installation and Operation Manual version version About this document This document is intended as a guide for installation, maintenance and troubleshooting of Portable Device Manager (PDM) and is relevant
Global Water Instrumentation, Inc. 151 Graham Road P.O. Box 9010 College Station, TX 77842-9010 Telephone : 800-876-1172 International : (979) 690-5560, Fax : (979) 690-0440 e-mail : email@example.com
Wireless Asset Monitoring - Local Food Safety Software BA/WAM-L-F Software Installation and User Guide System Overview The BAPI Wireless Asset Monitoring Local (WAM Local) Software receives temperature
Navigator Software Contents 1. NAVIGATOR SOFTWARE 2. INSTALLATION 3. USING NAVIGATOR SOFTWARE 3.1 STARTING THE PROGRAM 3.2 SYSTEM SET UP 3.3 LOAD DATA FILE 3.3.1 LOADING PARTIAL FILES 3.4 DATA TABLE WINDOW
AZI P/N 700-0042-B Last update May 2009 USER MANUAL JEROME COMMUNICATIONS SOFTWARE (JCS) INSTALLATION and OPERATION MANUAL V1.02 for Windows XP May 2009 ARIZONA INSTRUMENT LLC 3375 N Delaware Street Chandler,
Guide to Installing BBL Crystal MIND on Windows 7 Introduction The BBL Crystal MIND software can not be directly installed on the Microsoft Windows 7 platform, however it can be installed and run via XP
Using Avaya Flare Experience for Windows Release 9.0 Issue 02.01 September 2013 Contents Chapter 1: About Flare Experience... 5 About Flare Experience... 5 Main window... 6 Button descriptions... 10 Chapter
TRAKPRO Data Analysis Software User s Manual November 1996 P/N 1980200 Rev. D TSI Incorporated TRAKPRO Data Analysis Software User s Manual November 1996 P/N 1980200 Rev. D SHIP TO: TSI Incorporated 500
Network Management Interface Catalog VPi 32.15E For use with these ATS products: 7000 & 4000 Series (Group 5 Controller) Series 300 (Group 1 Controller), and 940, 962, 436, 434, 447, 448 (Group 6A/7A Control
ViewPower User s Manual Management Software for Uninterruptible Power Supply Systems Table of Contents 1. ViewPower Overview... 2 1.1. Introduction...2 1.2. Structure...2 1.3. Applications...2 1.4. Features...2
HP Point of Sale (POS) Peripherals Configuration Guide 2D Imaging / Linear / Presentation Scanner Document Version 3.42 December 2011 1 Copyright 2007-2011 Hewlett-Packard Development Company, L.P. The
Gigabyte Management Console User s Guide (For ASPEED AST 2400 Chipset) Version: 1.4 Table of Contents Using Your Gigabyte Management Console... 3 Gigabyte Management Console Key Features and Functions...
TABLE OF CONTENTS CHAPTER 1 ESTABLISHING A CONNECTION TO TIME CLOCK 1.1 CONNECT TIME CLOCK TO PC 2 1.2 VIA USB CABLE CONNECTION & TCP/IP 3-6 1.3 VIA TCP/IP NETWORK CONNECTION ON TIMECLOCK 7-8 CHAPTER 2
INF-USB2 and SI-USB Quick Start Guide Please follow these instructions carefully. DO NOT connect the INF-USB2 or SI-USB module to your computer before running the setup program. After running Setup and
BMV-602 Data Link Manual 1 Contents of the kit 1 x BMV-602 data link cable. 1 x Installation CD 2 System Requirements 2.1 Hardware requirements A BMV-602 with software version 2.03 or higher. A spare RS-232
Accounting Manager User Guide A31003-P1030-U114-2-7619 Our Quality and Environmental Management Systems are implemented according to the requirements of the ISO9001 and ISO14001 standards and are certified
Veriteq > > > > > vlog 4.4 User s Guide w w w. v e r i t e q. c o m w w w. v a i s a l a. c o m Copyright Copyright 2010 Veriteq Instruments, Inc. All rights reserved. No copying, distribution, publication,
PowerShield NetGuard User s Manual Management Software for Uninterruptible Power Supply Systems Table of Contents 1. NetGuard Overview...2 1.1. Introduction...2 1.2. Structure...2 1.3. Applications...3
High Resolution Temperature Sensing Strip Online Monitoring User Guide 888.637.3282 www.nerdata.com Page 1 of 26 Contents The Aurora Online Monitoring System... 3 1. Creating an Account on the Aurora Online
Legal Notes Unauthorized reproduction of all or part of this guide is prohibited. The information in this guide is subject to change without notice. We cannot be held liable for any problems arising from
SPC-1 Single Point Controller Operation Manual NOTICE: PPM Technology Limited shall not be liable for errors that may appear herein or for incidental or consequential damages in connection with or arising
Water Quality Monitor WQM Host Software User s Guide The user s guide is an evolving document. If you find sections that are unclear, or missing information, please let us know. Please check our website
SNMP Web card User s Manual Management Software for Uninterruptible Power Supply Systems Table of Contents 1. Overview... 3 1.1 Introduction... 3 1.2 Features... 3 1.3 Overlook... 3 1.4 Installation and
Capture Pro Software FTP Server System Output Overview The Capture Pro Software FTP server will transfer batches and index data (that have been scanned and output to the local PC) to an FTP location accessible
O P E R A T I O N S A N D P R O C E D U R E S F O R T H E P R O D U C T I V I T Y S O F T W A R E Productivity Software Features Remote CS-230 calibration and set-up on a personal computer. CS-230 calibration
Chapter III. Data Retrieval In This Chapter This chapter gives you instructions for setting up and using TradingExpert Pro s automatic data retrieval system which is used for downloading end-of-day data
Operating manual 42/49-27 EN Engineer IT Device configurator DRC200 R&C Process Data Management Software Impressum Device configurator DRC200 Operating manual Document No. 42/49-27 EN Date of issue: 11.02
Allworx OfficeSafe Operations Guide Release 6.0 No part of this publication may be reproduced, stored in a retrieval system, or transmitted, in any form or by any means, electronic, mechanical, photocopy,
SOFTWARE GUIDE PM1405 DATA PROCESSING SOFTWARE Equipment Software Manufacturer Survey meter РМ1405 "РМ1405 Software" Polimaster Copyright Polimaster 2015. All rights reserved. According Copyright protection
1 TABLE OF CONTENTS INTRODUCTION CHAPTER 1 ESTABLISHING A CONNECTION TO TIME CLOCK 1.1 INSTALLATION OF SOFTWARE 1.2 CONNECT TIME CLOCK TO PC 2 1.2.A..VIA USB CABLE CONNECTION 2-5 1.2.B... VIA TCP/IP NETWORK
EigenLight S500 Power Monitor Software Manual Software Installation... 1 Installing S500 Power Monitor Software and LabVIEW Run-time Engine... 1 Install Drivers for Windows XP... 4 Install VISA run-time...
Eaton Innovative Technology PEM PC Software Installation and Operation Instruction Manual: IM01005006E Rev. 2 March 2007 Modbus RTU (RS-485) TCP/IP (Ethernet) Effective Date 3/2007 For more information
PC Configuration Requirements: Configuration Requirements for ipc Operation The hardware and operating system of the PC must comply with a list of minimum requirements for proper operation with the ipc
Monitoring Network DMN User Manual Table of contents Table of contents... 2 1. Product features and capabilities... 3 2. System requirements... 5 3. Getting started with the software... 5 3-1 Installation...
STANDARD OPERATING PROCEDURE FOR USE OF LOG TAG ANALYZERS AS PART OF THE INTER HOSPITAL TRANSFER PROGRAM WITHIN THE PROVINCE OF NEWFOUNDLAND AND LABRADOR USING THE GOLDEN HOUR 24 / 2 SHIPPING CONTAINER
Power Log PC Application Software Users Manual May 2007 Rev. 1, 5/08 2007-2008 Fluke Corporation. All rights reserved. Specifications are subject to change without notice. All product names are trademarks
VIP-102B IP Solutions Setup Tool Reference Manual Table of Contents Overview...5 System Requirements for Managed VoIP Products...6 Term Definitions...7 Main Application Window...8 Legend Description...9
LESSON 13 Managing Devices OBJECTIVES After completing this lesson, you will be able to: 1. Open System Properties. 2. Use Device Manager. 3. Understand hardware profiles. 4. Set performance options. Estimated
Gigabyte Content Management System Console User s Guide Version: 0.1 Table of Contents Using Your Gigabyte Content Management System Console... 2 Gigabyte Content Management System Key Features and Functions...
DESKTOP CLIENT CONFIGURATION GUIDE BUSINESS EMAIL Version 2.0 Updated: March 2011 Contents 1. Mac Email Clients... 3 1.1 Configuring Microsoft Outlook 2011... 3 1.2 Configuring Entourage 2008... 4 1.3.
Software User Guide For the safe use of your camera, be sure to read the Safety Precautions thoroughly before use. Types of software installed on your computer varies depending on the method of installation
DT-CONFIG SOFTWARE Config software for D2 systems USER S MANUAL CONTENTS 1. Introductions ------------------------------------------------------------------- 3 2. System Requirement and Connection ----------------------------------------
Temperature Controller Support Software CX-Thermo Ver.4.0 Online Help Contents Chapter 1 Introduction 1.1 What is CX-Thermo? 1.2 Compatible Device Models and Computers 1.3 Hardware Setups 1.4 Basic Use
User's Guide Using RFDBManager For 433 MHz / 2.4 GHz RF Version 1.23.01 Copyright Notice Copyright 2005 Syntech Information Company Limited. All rights reserved The software contains proprietary information
V2.0 Congratulations on your purchase of the AORUS Notebook! This Manual will help you to get started with setting up your notebook. For more detailed information, please visit our website at http://www.aorus.com.
OT Programmer OPTOTRONIC LED Power Supply Configuration Tool User Manual V2.4.1 March 2015 Page 2 Contents 1. Introduction... 4 1.1. Compatible OPTOTRONIC Programmable LED power supplies... 4 1.2. PC system
Help File Version 188.8.131.52 February, 2010 MetaDigger for PC How to Use the Sound Ideas MetaDigger for PC Program: The Sound Ideas MetaDigger for PC program will help you find and work with digital sound
DaTa LoGGerXPTM User s Manual C R Y S T A L engineering corporation DataLoggerXP User s Manual Page Contents Introduction........................................................ 1 How it works........................................................
Rapid Assessment Key User Manual Table of Contents Getting Started with the Rapid Assessment Key... 1 Welcome to the Print Audit Rapid Assessment Key...1 System Requirements...1 Network Requirements...1
Novell ZENworks Asset Management 7.5 w w w. n o v e l l. c o m October 2006 USING THE WEB CONSOLE Table Of Contents Getting Started with ZENworks Asset Management Web Console... 1 How to Get Started...
MULTIFUNCTIONAL DIGITAL SYSTEMS Operator s Manual for AddressBook Viewer 2008, 2009 TOSHIBA TEC CORPORATION All rights reserved Under the copyright laws, this manual cannot be reproduced in any form without
Animated Lighting Software Revision 1.0 August 29, 2003 Table of Contents SOFTWARE OVERVIEW 1) Dasher Pro and Animation Director overviews 2) Installing the software 3) Help 4) Configuring the software
LOREX CLIENT Remote Software 4.0 Instruction Manual English Version 2.0 MODEL: L20WD800 Series www.lorexcctv.com Copyright 2008 LOREX Technology Inc. Table of Contents Table of Contents Software Installation...
Centurion C4 Transfer Guide using C4 File Transfer Utility 1010536 02 23 11 Section 50 1.0 Background: 1.1 A valid configuration is required for the Centurion system. Configuration files are downloaded
Getting Started with IntelleView POS Administrator Software Administrator s Guide for Software Version 1.2 About this Guide This administrator s guide explains how to start using your IntelleView POS (IntelleView)
Embedded Web Server User Guide General usage guide lines to the ISONAS Embedded Web Server available on all ISONAS PowerNet reader-controllers 06/12/09 Contents Revision history... 4 Scope... 4 Introduction...
TFA KlimaLogg Pro User Manual Revision: 0.1 Table of Contents Overview... 2 Getting Started... 2 Setting up the KlimaLogg Pro device... 2 Installing the USB-Stick... 2 Installation... 2 System Requirements...
Diamante WiFi Wireless Communication User Guide Linksys E1200 Release: February 2012; August 2011; February 2011 Patent Pending. Copyright 2012, Stenograph, L.L.C. All Rights Reserved. Printed in U.S.A.
P34 V6.0 Congratulations on your purchase of the GIGABYTE Notebook. This manual will help you to get started with setting up your notebook. The final product configuration depends on the model at the point
Manual Software SPECTRO3-MONITORING V5.0 (PC software for Microsoft Windows XP, VISTA, 7) for color sensors of the SPECTRO-3 series This manual describes the installation and operation of the SPECTRO3-MONITORING
Universal Simple Control, USC-1 Data and Event Logging with the USB Flash Drive DATA-PAK The USC-1 universal simple voltage regulator control uses a flash drive to store data. Then a propriety Data and
SW43W FlukeView Power Quality Analyzer Software Version 3.20 onwards Users Manual 4822 872 30778 December 2005, Rev.1, 04/06 2005 Fluke Corporation, All rights reserved. All product names are trademarks
User s Manual for Fingerprint Door Control Software Foreword The naissance of F7 indicated that fingerprint reader enters into professional door control domain. That s why we developed this software to