1 Corrective and Preventive Action Background & Examples Presented by: Kimberly Lewandowski-Walker Food and Drug Administration Division of Domestic Field Investigations Office of Regulatory Affairs
2 Overview Background on Corrective and Preventive Action (CAPA) Requirements Quality System (QS) CAPA (21 CFR ) Link Between CAPA and Other QS Regulation Requirements Examples on CAPA Guidance and Other Resources
3 Purpose of the CAPA Subsystem Collect and Analyze Information based on appropriate Statistical Methodology as necessary to detect recurring quality problems Identify and Investigate Existing and Potential Product and Quality Problems Take Appropriate, Effective, and Comprehensive Corrective and/or Preventive Actions
4 CAPA Subsystem in Context Design Controls Material Controls Management Controls Production & Process Controls Corrective and Preventive Actions Records, Documents, & Change Controls Service Reports Equipment & Facility Controls
5 When Does FDA Review CAPA? Inspections Quality System Inspection Technique (QSIT) Corrective and Preventive Action (CAPA) Subsystem Compliance Program ( ) Inspection of Medical Device Manufacturers Premarket Approval Applications (PMAs) Original PMAs Some PMA supplements (Site changes, 30-Day Notices) CAPA is NOT Reviewed in 510(k) applications Recalls (corrections and removals)
6 Why is CAPA Important? Linked to many other requirements Complaint Files Nonconforming Product Acceptance Activities Servicing Audits 803 Medical Device Reporting (MDR) 806 Reports of Corrections and Removals ( Recalls )... And many more Ensures problems are detected AND resolved.
7 Definitions Correction: repair, rework, or adjustment and relates to the disposition of an existing nonconformity. Corrective Action: the action taken to eliminate the causes of an existing nonconformity, defect or other undesirable situation in order to prevent recurrence. Preventive Action: action taken to eliminate the cause of a potential nonconformity, defect, or other undesirable situation in order to prevent occurrence. Nonconformity: non-fulfillment of a specified requirement.
8 Correction, Corrective Action, or Preventive Action? Replacing the label on a device that had the wrong label applied? Revising process parameters in response to complaints? Rewelding a contact that does not meet visual inspection requirements? Auditing all vendors of a key component after quality issues with only one vendor are identified? Revising equipment maintenance procedures to reduce drift in process specifications?
9 CAPA Procedures 21 CFR (a) Each manufacturer shall establish and maintain procedures for implementing corrective and preventive action. Procedures must ensure all requirements of CAPA subsystem are met Establish: define, document (in writing or electronically), and implement. Number and complexity of procedures vary based on the organization
10 The CAPA Process INPUTS Internal/ External Sources ELEMENTS Analyze Data, (a)(1) Investigate Cause, (a)(2) Identify Action, (a)(3) Verify/Validate Effectiveness, (a)(4) OUTPUTS Implement Changes, (a)(5) Disseminating Information, (a)(6) Submit for Management Review, (a)(7) Document, (b)
11 CAPA Procedures 21 CFR (a) What is your firm s process for events that trigger a CAPA Not every complaint is a CAPA Not every nonconformance is a CAPA
12 CAPA Data Analysis 21 CFR (a)(1) Analyzing processes, work operations, concessions, quality audit reports, quality records, service records, complaints, returned product, and other sources of quality data to identify existing and potential causes of nonconforming product, or other quality problems. Appropriate statistical methodology shall be employed where necessary to detect recurring quality problems.
13 Data Sources INTERNAL SOURCES Inspection/Test Data Nonconforming Material Reports Equipment Data Scrap/Yield Data Rework Data Returned Product Internal Audits Process Control Data Acceptance Activities EXTERNAL SOURCES Complaints Field Service Reports Legal Claims Warranty Claims External Audits Medical Device Reports (MDRs)
14 CAPA and Statistical Analysis FDA emphasizes that the appropriate statistical tools must be employed when it is necessary to utilize statistical methodology. FDA has seen far too often the misuse of statistics by manufacturers in an effort to minimize instead of address the problem. Such misuse of statistics would be a violation of this section. 61 Fed. Reg. at , Comment 159
15 Common Statistical Techniques Pareto charts Run charts Control charts Mean and standard deviation T tests for comparisons Experimental design (DOE) Graphical methods (fishbone diagrams, histograms, scatter plots, spreadsheets, etc.)
16 Other Analysis Techniques Management reviews Quality review boards Material review boards Other internal reviews
17 CAPA Data Analysis 21 CFR (a)(1) Ensure all quality data sources are defined and analyzed to identify existing product and quality problems How is the data is captured and maintained?
18 CAPA Data Analysis 21 CFR (a)(1) How does your firm categorize and group data and perform the analysis? Expect FDA to verify your firm is using appropriate analysis techniques Analysis of data should also include a comparison of the same problem type across different data sources
19 CAPA Data Analysis 21 CFR (a)(1) Is the data received by the CAPA system is complete, accurate, and timely? Trend analysis is one type of data analysis
20 CAPA Investigation 21 CFR (a)(2) Investigating the cause of nonconformities relating to product, process, and the quality system. 21 CFR , Complaint Handling, also requires investigations for the device involved, but the CAPA requirement is broader to cover the process and the quality system.
21 Typical Investigation Steps Identify problem and characterize. Determine scope and impact. Investigate data, process, operations and other sources of information. Determine root cause, if possible.
22 Possible Root Causes Training Design Manufacturing Management Change Control Purchasing/Supplier Quality Testing Documentation Maintenance Many manufacturers tend to overuse training as a corrective action and do not adequately address the systemic corrective action.
23 Root cause analysis tools Commonly used tools Fishbone diagrams 5 whys Fault-tree analysis Among others
24 CAPA Investigation 21 CFR (a)(2) Is the depth of the investigation sufficient? Expect FDA to evaluate the adequacy of your firm s rationale for determining if a corrective or preventive action is necessary Decision process may be linked to risk analysis
25 CAPA Investigation 21 CFR (a)(2) What controls does your firm have over devices suspected of having potential nonconformities? Justification for concessions should be well documented and appropriate to the product risk
26 Identify Required Actions 21 CFR (a)(3) Identify the action(s) needed to correct and prevent recurrence of non-conforming product and other quality problems. Beware of terminology! A Preventive Action is NOT required for all situations; however, a Corrective Action to prevent recurrence is required.
27 Other Quality Problems?... The objective of is to correct and prevent poor practices, not simply bad product.... Correction and prevention of unacceptable quality system practices should result in fewer nonconformities related to product.... For example, it [CAPA] should identify and correct improper personnel training, the failure to follow procedures, and inadequate procedures, among other things. 61 Fed. Reg. at , Comment 162
28 Taking Action Identify solutions. Develop action plan for corrective action and/or preventive action. Should consider the risk posed by the problem. Not all problems require the same level of investigation and action. It is appropriate to elevate some issues at the expense of others
29 CAPA and Risk Management FDA agrees that the degree of corrective and preventive action taken to eliminate or minimize actual or potential nonconformities must be appropriate to the magnitude of the problem and commensurate with the risks encountered FDA does expect the manufacturer to develop procedures for assessing the risk, the actions that need to be taken for different levels of risk, and how to correct or prevent the problem from recurring, depending on that risk assessment. 61 Fed. Reg. at , Comment 159
30 CAPA and Risk Management Risk analysis allows a manufacturer to: Determine priorities Assign resources Determine the severity of impact Determine the depth of investigation Common tools Hazard analysis Used early for potential problems Failure Mode Effects Analysis (FMEA) Bottom up Fault Tree Analysis (FTA) Top down
31 Identify Required Actions 21 CFR (a)(3) Expect FDA to review the actions taken Be prepared to discuss the appropriateness of the action taken Why was corrective action taken? Does the corrective action extend to include any additional actions (component suppliers, training, acceptance activities, field actions) if necessary?
32 Verify and Validate 21 CFR (a)(4) Verifying or validating the corrective and preventive action to ensure that such action is effective and does not adversely affect the finished device. Effectiveness: Did my solution work? Did it create other potential nonconformances?
33 Verify and Validate 21 CFR (a)(4) Verify that verification/validation protocols were established Review data associated with verification or validation activities Review the effectiveness of the corrective and preventive actions by reviewing data to determine if similar quality problems exist after implementation
34 Implement and Record Changes 21 CFR (a)(5) Implementing and recording changes in methods and procedures needed to correct and prevent identified quality problems.
35 Implementing Changes Tie CAPA implementation to: Document control for products and processes (DMR ) Change control (820.40) Ensure that controlled documents are reviewed and approved if changes are made.
36 Implementing Changes Expect FDA to verify implementation of changes by viewing actual processes and equipment Implemented changes may directly link to design or production and process controls
37 Disseminate Information 21 CFR (a)(6) Ensuring that information related to quality problems or nonconforming product is disseminated to those directly responsible for assuring the quality of such product or the prevention of such problems
38 Management Review 21 CFR (a)(7) Submitting relevant information on identified quality problems, as well as corrective and preventive actions, for management review The significance of the problem impacts the level of management review. Need management awareness and buy-in so that resources are allocated, etc.
39 Documentation 21 CFR (b) All activities required under this section, and their results, shall be documented.
41 What is Nonconforming Product? Nonconformity = the nonfulfillment of a specified requirement. [21CFR820.3(q)] + Product = components, manufacturing materials, in-process devices, finished devices, and returned devices. [21CFR820.3(r)]
42 Examples of Nonconforming Product Received components/material that fail incoming inspection. Products/components that fail inspection or test steps during manufacturing. Product returned to the manufacturer with defects.
43 Links Between Nonconforming Product and CAPA Identification and monitoring of nonconforming product often triggers CAPA activities. Nonconforming product investigations can also be leveraged during CAPA investigations. Not every nonconformance is a CAPA.
44 What is a complaint? Any written, electronic, or oral communication that alleges deficiencies related to the identity, quality, durability, reliability, safety, effectiveness, or performance of a device after it is released for distribution. 21 CFR 820.3(b)
45 Link Between Complaints and CAPA Complaints are a required data source to CAPA and may trigger CAPA activity. Remember, not every complaint is a CAPA Complaint investigations can be leveraged during CAPA investigations. Recurrent complaints that involve a health risk may be evaluated by CAPA process and determined to be recalls.
46 Key Word #1: Nonconformance CAPA, nonconforming product, and complaints are all related to identified or potential nonconformances. Nonconformances are: A data source for the CAPA system The basis of the nonconforming product requirements A potential cause of complaints
47 Key Word #2: Investigation Investigation is used in CAPA, nonconforming product, and complaints sections of the QS Regulation Investigations are: Required for CAPA Done, or justified not done, for complaints May/may not be done for nonconformances, no justification required.
48 Closing Remarks CAPA is a pulse check for FDA on how well a firm s Quality System is operating Strong CAPA systems are usually indicative of strong Quality Systems Feedback Loop between CAPA, Complaints, and Nonconforming Product is essential CAPA Subsystem is all about identifying and resolving problems that can or have result[ed] in nonconforming product
49 Examples Related to CAPA
50 Example #1 Warning Letter Citation Your firm failed to establish, maintain, and implement a corrective and preventive action procedure, as required by (a). For example, Your firm has no CAPA procedures as defined in the QS regulation including: failure investigation, procedures to analyze quality data procedures to verify/validate corrections, procedures that ensure that information related to quality problems is disseminated and for submitting relevant information on identified quality problems to management for review.
51 Example #2 Warning Letter Citation The procedures addressing verification or validation of corrective and preventive actions were not implemented as required by 21 CFR (a)(4). Specifically, CAPA Procedure , Rev 4, requires documentation of verification and validation, but this was not completed for CAPA #0225.
52 Example #3 Warning Letter Citation Failure to implement procedures addressing documentation of corrective and preventive action activities as required by 21 CFR (b). Specifically: The Corrective and Preventive Action procedure (GP.1401, Ver. 6.0) establishes requirements to (i.) " *** Design and implement corrective or preventive action, including verification or validation that such action does not adversely affect the finished device or system" and (ii.) "Demonstrate and document the effectiveness of corrective and/or preventive action(s)." The procedure also states, (iii.) "The corrective action and preventive action changes shall follow design, process, or Quality System change process requirements. However, the validation of patient management recommendations (implemented in CAPA 644 as part of the lead fracture corrective action and October 15, 2007, recall) was not reviewed and documented appropriately.
53 Injunction Example #1 SOP Revs. C, D, and E, Corrective and Preventive Action Procedure, 4.4 states one of the requirements of a CAPA is Verifying or validating that the corrective action and/or preventive action is effective and does not adversely affect the finished device. Form D, CAPA Form, dated 3/13/06 describes a CAPA initiated because of a complaint associated with an aortic valve that had a mislabeled size. The firm s CAPA included a revision of SOP , Label Control, to include an additional inspection. The verification/validation activities were stated as The additional inspection procedure has no bearing on the product quality and will only help to prevent a future occurrence. As a part of this CAPA, the firm did not ensure that the procedure had been correctly implemented and that employees were retrained on this revised procedure.
54 References 21 CFR Part ch.cfm?cfrpart=820 Preamble to the QS Regulation Final Rule andguidance/postmarketrequirements/qualitysystemsregulati ons/medicaldevicequalitysystemsmanual/ucm pdf Compliance Program ( ) Inspection of Medical Device Manufacturers ce/guidancedocuments/ucm htm
55 References, cont. Quality System Inspection Technique (QSIT) htm Quality System Information for Certain Premarket Application Reviews: Guidance for Industry and FDA Staff ce/guidancedocuments/ucm htm Global Harmonization Task Force (GHTF) Guidance Document on Corrective and Preventive Action
Corrective and Preventive Actions A Five Step Approach Topics to Be Covered What is CAPA? Governing authority Five steps to a good CAPA process Where companies have difficulty Example citations Recap What
FINAL DOCUMENT Global Harmonization Task Force Title: Quality management system Medical Devices Guidance on corrective action and preventive action and related QMS processes Authoring Group: Study Group
environmental failure analysis & prevention health technology development How to Use the Design Process to Manage Risk: Elements of Design Controls and Why It Matters Kevin L. Ong, Ph.D., P.E. Managing
TRAINING TITLE: CAPA System Expert Certification (CERT-003) OVERVIEW: Medical devices, biopharmaceutical, and traditional drug manufacturing companies devote an important part of their resources dealing
Setting Up A Complaint Handling System Presented by Sue Jacobs Principal Consultant, QMS Consulting, Inc. January 2008 ~ Orlando, Florida QMS Consulting, Inc. 2008 1 Complaint Systems Written Procedures
IVD Regulation Overview Requirements to Assure Quality & Effectiveness CLIAC Jan. 2002 Statutory and Regulatory Requirements Statute: Food, Drug, and Cosmetic Act Food and Drugs Act of 1906 Food and Drug
CAPA Issues and Pitfalls Timothy Barash, BSI April 3, 2014 2 CAPA: Corrective Action, Preventive Action It does NOT mean JUST Corrective Action! A CAPA Story 3 4 Overview A CAPA Story Core CAPA Steps CAPA
GxP Process Management Software : Ten Most Common Reasons for FDA 483 Observations and Warning Letter Citations Most FDA violations involve one of the following: Not having procedures in a regulated area
Quality System: Design Control Procedure - Appendix Page 1 of 10 Quality System: Design Control Procedure - Appendix CORP Medical Products Various details have been removed, indicated by [ ] 1. Overview
Effective Root Cause Analysis For Corrective and Preventive Action Manuel Marco Understanding Key Principles Requirement need or expectation that is stated, generally implied, or obligatory Generally implied
Supplier Training 8D Problem Solving Approach What is the 8D method? 8D stands for the 8 disciplines or the 8 critical steps for solving problems. It is a highly disciplined and effective scientific approach
Page 1 of 7 Home Inspections, Compliance, Enforcement, and Criminal Investigations Compliance Actions and Activities Warning Letters 2014 Inspections, Compliance, Enforcement, and Criminal Investigations
Guidance for Industry Q10 Pharmaceutical Quality System U.S. Department of Health and Human Services Food and Drug Administration Center for Drug Evaluation and Research (CDER) Center for Biologics Evaluation
2014 Annual FDA Medical Device Quality System Data FDA Form 483 Observations and Warning Letter Citations Why is FDA making these data available? In support of the FDA s Transparency and Case for Quality
1. QUALITY MANAGEMENT SYSTEM QUALITY MANAGEMENT SYSTEM (QMS) ASSESSMENT CHECKLIST 1.1 Quality Management System General 1.1.1 Is objective evidence available to demonstrate that the MDSAP site has defined,
Getting a Handle on Complaints Industry continues to be deficient in complaint-handling procedures despite FDA having made those a priority for a long time. A new survey aims to help change that. By: Erik
FAILURE INVESTIGATION AND ROOT CAUSE ANALYSIS Presented By Clay Anselmo, R.A.C. President and C.O.O. Reglera L.L.C. Denver, CO Learning Objectives Understand the Definitions of Failure Investigation and
FINE LOGISTICS Quality Manual Document No.: 20008 Revision: A 20008 Rev. A FINE LOGISTICS, Quality Manual Page 1 of 24 Quality Manual: Table of contents Number Section Page 1. GENERAL 3 1.1 Index and revision
CAPA Common Pitfalls Presented by: Sue Jacobs QMS Consulting, Inc. 1 847 359 4456 email@example.com 1 Topics CAPA Process Flow Elements of an Effective CAPA System Common Pitfalls 2 CAPA Process Flow
September 2015 EMA/CHMP/ICH/214732/2007 Committee for Human Medicinal Products Step 5 Transmission to CHMP May 2007 Transmission to interested parties May 2007 Deadline for comments November 2007 Final
Today s Best Practices in Global Product Surveillance Systems Presented by: Nancy Singer Compliance-Alliance Richard J. DeRisio, M.S. Kinetic Concepts, Inc. 1 Management of Complaint and Service Experience
GHTF/SG3/N15R8 FINAL DOCUMENT Title: Implementation of risk management principles and activities within a Quality Management System Authoring Group: GHTF Study Group 3 Endorsed by: The Global Harmonization
ISO 9001 Quality Management Systems Professional Professional Certifications Sample Questions Sample Questions 1. The non-fulfillment of a specified requirement is called a: A. Concession B. Nonconformity
Building an Effective Supplier Control Program: A review of key program elements & their implementation. Jonathan Lee VP RQCS Medtronic Surgical Technologies Building an Effective Supplier Control Program
Soft Computer Consultants, Inc. 4/30/15 Department of Health and Human Services Public Health Service Food and Drug Administration Florida District 555 Winderley Place, Suite 200 Maitland, Florida 32751
Page 1 of 12 4 Quality Management System 4.1 General Requirements The Peerless Pump Quality Management System shall include: Documented statements of a quality policy and of quality objectives; A quality
ISO 9001 Quality Systems Manual Revision: D Issue Date: March 10, 2004 Introduction Micro Memory Bank, Inc. developed and implemented a Quality Management System in order to document the company s best
CLOVER MACHINE AND MFG. 800 MATHEW ST. #101 SANTA CLARA, CA 95050 727-3380 727-7015 fax REVISION: DATE: PAGE 1 OF 45 QUALITY POLICY MANUAL DISTRIBUTION LIST: President Purchasing Manager Vice President
Quality & Compliance Associates, LLC Best Practice In A Change Management System President Quality & Compliance Associates, LLC Change Control and Its Role in a Continuous Improvement Environment 3 Benefits
Corporate Quality Manual Preface The following Corporate Quality Manual is written within the framework of ISO 9001:2008 Quality System by the employees of CyberOptics. CyberOptics recognizes the importance
g GE Power & Water ISO 9001:2008 Audit Checklist Organization Auditor Date Page 1 Std. 4.1 General s a. Are processes identified b. Sequence & interaction of processes determined? c. Criteria for operation
Page 1 of 20 QUALITY MANUAL ISO 9001 Quality Management System Printed copies are not controlled unless marked "CONTROLLED". Upon receipt of this document, discard all previous copies. Page 2 of 20 Approval
WEIGH-TRONIX CANADA ULC NORTH AMERICA OPERATIONS (Fairmont and Montreal Facilities) QUALITY MANUAL Prepared to comply with the requirements of ISO 9001:2008 Meets or exceeds the requirements for design,
Complaints, Servicing, and FDA Reporting Dan O Leary CBA, CQA, CQE, CRE, SSBB, CIRM President Ombu Enterprises, LLC Dan@OmbuEnterprises.com www.ombuenterprises.com 603-358-3082 1 Outline Setting the Stage
Guidance for Industry Quality Systems Approach to Pharmaceutical CGMP Regulations U.S. Department of Health and Human Services Food and Drug Administration Center for Drug Evaluation and Research (CDER)
MAIL-IN / ONSITE This report is intended to furnish data relative to the Suppliers capability to control the quality of supplies and services furnished to Ontic UK. Please complete this Survey and return
Company Name: Date: Company Address: Quality Assurance Mgr: President: Number of employees: Is your company receptive to source inspection? List all standards the quality system is based on Web Page: Company
QUALITY ASSURANCE MANUAL JPM OF MISSISSIPPI, INC. Hattiesburg, MS Revision E 01/19/11 Revised to ISO 9001:2008 on July 9, 2009 JPM OF MISSISSIPPI, INC. MANAGEMENT QUALITY POLICY It is the goal of JPM of
Far West Technology, Inc. ISO 9001 Quality Manual Document No.: QM Revision: 0 Issue Date: 27 August 1997 Approval Signatures President/CEO Executive Vice President Vice President/CFO Change Record Rev
ISO 9001:2000 Gap Analysis Checklist Type: Assessor: ISO 9001 REQUIREMENTS STATUS ACTION/COMMENTS 4 Quality Management System 4.1 General Requirements Processes needed for the quality management system
Specialties Manufacturing Talladega Castings & Machine Co., Inc. ISO 9001:2008 This document is the property of TMS and may not be reproduced, wholly, or in part, without the express consent of TMS. Rev.
Company Name Street Address City, State, Zip code Phone Number Fax Company Website Email Address ORGANIZATION NAME PHONE NUMBER EMAIL ADDRESS President/CEO General Manager Engineering Manager Production
Optimizing Quality Control / Quality Assurance Agents of a Global Sourcing / Procurement Strategy Global Pharma Sourcing Conference December 6-7, 2011 Philadelphia, USA Nigel J. Smart, Ph.D. Smart Consulting
Title: Quality Management System Manual Page: 1 of 20 INTRODUCTION Inductors Inc., founded in 1991, specializes in the distribution of inductive components, including but not limited to standard and custom
Eagle Machining, Inc. Quality Management System 1 of 10310 Antoine Drive Bldg D, Houston, Texas 77086 BUSINESS OPERATING MANUAL (QUALITY MANUAL) Revision Date: 08/01/2014 Approved By: Joseph Vu Date: 08/01/2014
FDA Medical Device Industry Coalition ISO 14971: Overview of the standard Risk Management Through Product Life Cycle: An Educational Forum William A. Hyman Department of Biomedical Engineering Texas A&M
Row Manufacturing Inc. Quality Manual ISO 9001:2008 Row Manufacturing 210 Durham Drive Athens, Alabama 35611 Phone:256.232.4151 Fax:256.232.4133 Page 2 of 33 This Page intentionally left Blank Page 3 of
Quality Management Plan Procedure Check the Capital Project Delivery website to ensure this is the current version. Table of Contents Quality Management Plan Overview... 1 Introduction... 1 Objectives...
Edwin Lindsay Principal Consultant, Tel: + 44 (0) 7917134922 E-Mail: firstname.lastname@example.org Corrective and Preventative Action (CAPA) is a system of quality procedures required to eliminate the causes
ORACLE CONSULTING GROUP An Official United States Agent Firm for Foreign Establishments CONSULTING MEMORANDUM: DEALING WITH A MEDICAL DEVICE IN THE U.S. 5398 Golder Ranch Rd., Ste. 1 Tucson, Arizona 85739
Making Recall Determinations AdvaMed January 23, 2008 Orlando, FL Edward C. Wilson, Jr., Esq. Partner Hogan & Hartson LLP (202) 637-5839 email@example.com www.hhlaw.com Michael S. Heyl, Esq. Associate
Lotlikar et al Journal of Drug Delivery & Therapeutics; 2013, 3(2), 149-154 149 Available online at http://jddtonline.info REVIEW ARTICLE QUALITY RISK MANAGEMENT (QRM): A REVIEW Lotlikar MV Head Corporate
Solumina is designed for companies that manufacture or service complex products with multiple levels of sub-assemblies, and that are subject to frequent engineering changes. PROCESS PLANNING MANUFACTURING
Appendix 3 (normative) High level structure, identical core text, common terms and core definitions NOTE In the Identical text proposals, XXX = an MSS discipline specific qualifier (e.g. energy, road traffic
Deviation and Out of Specification Handling APV Training Course GMP Requirements June,10th to11th 2004 Istanbul, Turkey Dr. Jürgen Mählitz GMP Inspector Fronhof 10 D-86152 Augsburg Germany Dr. Jürgen Mählitz
ISO-9001:2000 Quality Management Systems REQUIREMENTS 10/10/2003 ISO-9001:2000 Requirements 1 Process Based Approach C U S MANAGEMENT RESPONSIBILITY RESOURCE MANAGEMENT C U S T O M Requirements PRODUCT
Improved Utilization of Self-Inspection Programs within the GMP Environment A Quality Risk Management Approach Barbara Jeroncic Self-inspection is a well-established and vital part of the pharmaceutical
Medical Product Development and FDA Regulations IEEE Orange County Computer Society March 27, 2006 Carl R. Wyrwa Medical Product Development and FDA Regulations Introduction Regulated FDA Overview Medical
PS/INF 1/2010 * * Quality Risk Management Quality Risk Management Implementation of ICH Q9 in the pharmaceutical field an example of methodology from PIC/S Document > Authors: L. Viornery (AFSSAPS) Ph.
Quality Risk Management - The Medical Device Experience Niamh Nolan Principal Design Assurance Engineer Boston Scientific Agenda Intent of Risk Management (RM) and Associated Regulations Overview of RM
ISO 9001:2008 QUALITY MANUAL Revision B Because we want you to achieve the highest levels of performance, we connect care Because with leading we want science you to achieve continuously the highest improve,
ISO 9001:2008 11 Industry Lane Flippin, Arkansas 72634 QM-001-2008-F Page 2 of 39 Introduction Micro Plastics, Inc. developed and implemented a Quality Management System in order to document the company
QUALITY MANAGEMENT SYSTEM (QMS) MANUAL This document and the data contained herein are the property of ISOLINK, and shall not, without prior written permission, be reproduced or used for any unintended
WCC PDA Dinner Meeting Jan 2012 ICH Q10 - Pharmaceutical Quality System Neil Wilkinson NSF-DBA www.nsf-dba.com ICHQ10.1 WCC PDA Dinner Meeting Jan 2012 Your Presenter Partner at NSF-DBA USA Training, Consultancy,
Guidance for Industry Q9 Quality Risk Management U.S. Department of Health and Human Services Food and Drug Administration Center for Drug Evaluation and Research (CDER) Center for Biologics Evaluation
The online version of this procedure is official. Therefore, all printed versions of this document are unofficial copies. St. Louis Public Schools Quality Assurance System Manual 801 North 11 th Street